|Using average correction factors to improve the estimated sound pressure level near the tympanic membrane.|
|PMID: 23072965 Owner: NLM Status: MEDLINE|
|BACKGROUND: Sound pressure-based real ear measurements are considered best practice for ensuring audibility among individuals fitting hearing aids. The accuracy of current methods is generally considered clinically acceptable for frequencies up to about 4 kHz. Recent interest in the potential benefits of higher frequencies has brought about a need for an improved, and clinically feasible, method of ensuring audibility for higher frequencies.
PURPOSE: To determine whether (and the extent to which) average correction factors could be used to improve the estimated high-frequency sound pressure level (SPL) near the tympanic membrane (TM).
RESEARCH DESIGN: For each participant, real ear measurements were made along the ear canal, at 2-16 mm from the TM, in 2-mm increments. Custom in-ear monitors were used to present a stimulus with frequency components up to 16 kHz.
STUDY SAMPLE: Twenty adults with normal middle-ear function participated in this study.
INTERVENTION: Two methods of creating and implementing correction factors were tested.
DATA COLLECTION AND ANALYSIS: For Method 1, correction factors were generated by normalizing all of the measured responses along the ear canal to the 2-mm response. From each normalized response, the frequency of the pressure minimum was determined. This frequency was used to estimate the distance to the TM, based on the ¼ wavelength of that frequency. All of the normalized responses with similar estimated distances to the TM were grouped and averaged. The inverse of these responses served as correction factors. To apply the correction factors, the only required information was the frequency of the pressure minimum. Method 2 attempted to, at least partially, account for individual differences in TM impedance, by taking into consideration the frequency and the width of the pressure minimum. Because of the strong correlation between a pressure minimum's width and depth, this method effectively resulted in a group of average normalized responses with different pressure-minimum depths. The inverse of these responses served as correction factors. To apply the correction factors, it was necessary to know both the frequency and the width of the pressure minimum. For both methods, the correction factors were generated using measurements from one group of ten individuals and verified using measurements from a second group of ten individuals.
RESULTS: Applying the correction factors resulted in significant improvements in the estimated SPL near the TM for both methods. Method 2 had the best accuracy. For frequencies up to 10 kHz, 95% of measurements had <8 dB of error, which is comparable to the accuracy of real ear measurement methods that are currently used clinically below 4 kHz.
CONCLUSIONS: Average correction factors can be successfully applied to measurements made along the ear canals of otologically healthy adults, to improve the accuracy of the estimated SPL near the TM in the high frequencies. Further testing is necessary to determine whether these correction factors are appropriate for pediatrics or individuals with conductive hearing losses.
|Karrie LaRae Recker; Tao Zhang; Weili Lin|
Related Documents :
|15184355 - Orthostatic tolerance and blood volumes in andean high altitude dwellers.
11883155 - Autonomic functions in buerger's disease.
702045 - Regulation of arterial blood pressure in australian tiger snakes.
9068805 - The capnography-tilt test for the diagnosis of hyperventilation syncope.
19041045 - Tricuspid regurgitation contributes to renal dysfunction in patients with heart failure.
7562655 - An animal model and computer-controlled surface pressure delivery system for the produc...
|Type: Clinical Trial; Journal Article; Validation Studies|
|Title: Journal of the American Academy of Audiology Volume: 23 ISSN: 1050-0545 ISO Abbreviation: J Am Acad Audiol Publication Date: 2012 Oct|
|Created Date: 2012-10-17 Completed Date: 2012-12-11 Revised Date: 2014-07-31|
Medline Journal Info:
|Nlm Unique ID: 9114646 Medline TA: J Am Acad Audiol Country: Canada|
|Languages: eng Pagination: 733-50 Citation Subset: IM|
|American Academy of Audiology.|
|APA/MLA Format Download EndNote Download BibTex|
Acoustic Impedance Tests
Aged, 80 and over
Auditory Threshold / physiology
Calibration / standards
Ear Canal / physiology*
Ear, Middle / physiology
Hearing Aids / standards*
Pitch Perception / physiology
Reproducibility of Results
Tympanic Membrane / physiology*
From MEDLINE®/PubMed®, a database of the U.S. National Library of Medicine
Previous Document: Cochlear implant combined with a linear frequency transposing hearing aid.
Next Document: Controlled nanoscale diffusion-limited chemical etching for releasing polystyrene nanocones from rec...