|Trimming of a Broken Migrated Biliary Metal Stent with the Nd:YAG Laser.|
|Jump to Full Text|
|PMID: 20651959 Owner: NLM Status: Publisher|
|Biliary metal stents are a permanent solution for bile duct stenosis. Complications can arise when the stent migrates, breaks or is overgrown by tumour. The following case demonstrates how a Nd:YAG laser can be used to solve these problems. A 93-year-old man presented with jaundice and fever. Two years earlier a 6-cm metal stent had been implanted into a postinflammatory stenosis of the common bile duct after recurrent cholangitis and repetitive plastic stenting. Duodenoscopy showed that the stent was broken. It had migrated about 3 cm into the duodenum, leading to kinking of the stent and breaking of the wires. The stent was also occluded. It was necessary to purge the common bile duct and to introduce a second stent. However, the only way to reach the papilla was through the broken wires. Placing a second stent this way was impossible. Thus we trimmed the stent with a Nd:YAG laser. The piece that had migrated into the duodenum was retrieved. Now the papilla could be reached. The rest of the metal stent was purged with NaCl 0.9%. A second metal stent was placed. Since an Nd:YAG laser is part of the equipment of most endoscopy units, it can be used to trim a broken or migrated biliary metal stent.|
|I Zuber-Jerger; F Kullmann|
Related Documents :
|15900549 - Emergency stent graft deployment for acute aortic rupture following primary stenting fo...
21267969 - Treatment of saddle pulmonary embolism with streptokinase in an 83 year-old man - a cas...
12069279 - Stent-assisted endoluminal revascularization for the treatment of intracranial atherosc...
15770349 - Treatment of wide-necked aneurysms with balloon-expandable polyurethane-covered stentgr...
16099689 - Mr imaging-guided percutaneous tumor ablation.
15783099 - Confirmation of the correct placement of lower end of dj stent during open surgery: poi...
12391089 - Effects of simulated microgravity on arterial nitric oxide synthase and nitrate and nit...
10411859 - New device for closure of muscular ventricular septal defects in a canine model.
7583729 - Antegrade placement of a ureteric stent by a pull-through technique.
|Type: JOURNAL ARTICLE Date: 2009-3-28|
|Title: Case reports in gastroenterology Volume: 3 ISSN: 1662-0631 ISO Abbreviation: - Publication Date: 2009|
|Created Date: 2010-7-23 Completed Date: - Revised Date: -|
Medline Journal Info:
|Nlm Unique ID: 101474819 Medline TA: Case Rep Gastroenterol Country: -|
|Languages: ENG Pagination: 16-20 Citation Subset: -|
|Department of Internal Medicine I, University of Regensburg, Regensburg, Germany.|
|APA/MLA Format Download EndNote Download BibTex|
Journal ID (nlm-ta): Case Rep Gastroenterol
Journal ID (publisher-id): CRG
Publisher: S. Karger AG, Allschwilerstrasse 10, P.O. Box · Postfach · Case postale, CH–4009, Basel, Switzerland · Schweiz · Suisse, Phone: +41 61 306 11 11, Fax: +41 61 306 12 34, firstname.lastname@example.org
Copyright © 2009 by S. Karger AG, Basel
collection publication date: Year: 2009
Electronic publication date: Day: 28 Month: 3 Year: 2009
pmc-release publication date: Day: 28 Month: 3 Year: 2009
Volume: 3 Issue: 1
First Page: 16 Last Page: 20
PubMed Id: 20651959
Publisher Id: crg0003-0016
|Trimming of a Broken Migrated Biliary Metal Stent with the Nd:YAG Laser|
|Department of Internal Medicine I, University of Regensburg, Regensburg, Germany
|Correspondence: *Ina Zuber-Jerger, MD, Department of Internal Medicine I, University of Regensburg, DE–93042 Regensburg (Germany), Tel. +49 941 944 7010/7020, Fax +49 941 944 7073E-Mail email@example.com
Stenosis of the common bile duct [1, 2, 3], the pancreatic duct  or the oesophagus [5, 6] can be resolved by the implantation of a metal stent. Gastrointestinal metal stents have proven to be a long-lasting solution in situations where surgery is impossible or denied. Complications arising from metal stents are migration, breaking and closure by bolus or tumour ingrowth. The most common complication is tumour ingrowth, which arises in nearly every second patient with oesophagus carcinoma and oesophageal metal stent . This complication can sometimes be avoided by the implantation of a covered metal stent , and if occurring it can usually be solved by tumour debridement with laser and/or placement of a second stent [7, 8]. In the biliary tree the problem of tumour ingrowth into a metal stent is usually solved by the placement of a second covered stent.
Breaking and migration of metal stents in the oesophagus and common bile duct can lead to serious clinical problems, such as as complete closure of the bile duct leading to cholestasis, or of the duodenum or oesophagus leading to either ileus or complete inability of food intake. Gastrointestinal haemorrhage or ulceration or perforation due to duodenal erosion by the stent can arise [9, 10, 11]. The problem of migration and breaking of a stent can sometimes be solved by trimming the stent with laser and taking it out [12, 13]. We present the case of a patient with a broken migrated biliary metal stent leading to cholestasis and incomplete closure of the duodenum that could be solved by trimming the stent with an Nd:YAG laser.
A 93-year-old man presented to the emergency room with fever and jaundice. Two years earlier an uncovered Zilver metal stent (length 6 cm, diameter 10 mm) had been implanted into a postinflammatory stenosis of the common bile duct because of recurrent cholangitis. Plastic stents had been occluded every second week and had thus not been able to avoid cholangitis.
At presentation the patient complained of chills and abdominal pain. Clinical examination revealed jaundice and painful palpation of the epigastrium. Fever rose up to 39°C. Laboratory tests showed elevated bilirubin (2.6 mg/dl) and gamma-GT (1,340 U/l), neutrophilia (82.5%) and an elevated CRP (66.48 mg/dl). Duodenoscopy revealed that the stent was totally occluded with detritus and had migrated into the duodenum about 3 cm. This migration had led to a kinking of the stent and the wires of the stent were broken at the curve (fig. 1). Thus, the broken biliary metal stent occluded the common bile duct completely and the duodenum partly. The papilla was inaccessible due to the stent. Ulceration of the duodenal wall with gastrointestinal bleeding or perforation had to be expected. Retrieving the stent seemed unavoidable, but we feared complications such as bleeding due to mucosa maceration and perhaps even perforation of the bile duct when forcefully retrieving a metal stent that was securely impacted in the bile duct after two years in place.
We thus tried to trim the stent at the papilla with an Nd:YAG laser with 308 J and 50 W. We managed to cut the wires exactly at the papilla. The piece of stent cut off by the laser rested in the duodenum after relaxation of the small intestine with butylscopolamine (fig. 2) and was taken out with a snare. The common bile duct was purged through the stent with NaCl 0.9% and a second metal stent (Ultraflex uncovered, length 8 cm, diameter 10 mm) was placed (fig. 3). Injection of contrast dye through a standard catheter showed that the biliary tree was free of detritus and stones and that the stream of bile was adequate. The patient was discharged symptom-free. No biliary complications have arisen in the last two years.
A Nd:YAG laser is part of the equipment of most endoscopy units. It is used predominantly for malignant oesophageal stenosis [7, 8]. It can also be used to trim broken or migrated metal stents [12, 13], avoiding clinical complications such as bleeding and perforation and allowing, like in our case, further stenting of the common bile duct to resolve cholestasis.
|1.||Srikureya W,Chang KJ. Endoscopic palliation of pancreatic adenocarcinomaCurr Opin GastroenterolYear: 20052160160516093777|
|2.||Brugge WR. Endoscopic techniques to diagnose and manage biliary tumoursJ Clin OncolYear: 2005234561456516002848|
|3.||Siriwardana HP,Siriwardana AK. Systematic appraisal of the role of metallic endobiliary stents in the treatment of benign bile duct structureAnn SurgYear: 2005242101915973096|
|4.||Eisendrath P,Deviere J. Expandable metal stents for benign pancreatic duct obstructionAm J GastroenterolYear: 19999547554|
|5.||Kozarek RA. Endoscopic palliation of esophageal malignancyEndoscopyYear: 200535S9S1312929047|
|6.||Von Rahden BH,Stein HJ. Staging and treatment of advanced esophageal cancerCurr Opin GastroenterolYear: 20052147247715930991|
|7.||Sargeant IR,Loizou LA,Tulloch M,Thorpe S,Brown SG. Recanalization of tube overgrowth: a useful new indication for laser in palliation of malignant dysphagiaGastrointest EndoscYear: 1992381651691373700|
|8.||McManus K,Khan I,McGuigan J. Self-expanding oesophageal stents: strategies for re-interventionEndoscopyYear: 20013360160411473332|
|9.||Isayama H,Komatsu Y,Tsujino T,Sasahira N,Hirano K,Toda N,Nakai Y,Yamamoto N,Tada M,Yoshida H,Shiratori Y,Kawabe T,Omata M. A prospective randomized study of ‘covered’ versus ‘uncovered’ diamond stents for the management of distal malignant biliary obstructionGutYear: 20045372973415082593|
|10.||Marano BJ,Bonanno CA. Metallic biliary endoprosthesis causing duodenal perforation and acute upper gastrointestinal bleedingGastrointest EndoscYear: 1994402572588013842|
|11.||Ee H,Laurence BH. Haemorrhage due to erosion of a metal biliary stent through the duodenal wallEndoscopyYear: 1992244314321505493|
|12.||Shaver CP,Brady P,Pinkas H. You don't have to retrieve to relieve: How and when to trim a self-expanding metal biliary stentEndoscopyYear: 20043683315326584|
|13.||Vanbierbliet G,Piche T,Caroli-Bose FX,Dumas R,Peten EP,Huet PM,Tran A,Demarquay JF. Endoscopic argon plasma trimming of biliary and gastrointestinal metallic stentsEndoscopyYear: 20053743443815844021|
Keywords: Key Words Broken migrated biliary metal stent, Nd:YAG laser, Trimming.
Previous Document: Successful Hemostasis with Recombinant Activated Factor VII in a Patient with Massive Hepatic Subcap...
Next Document: Metastatic Signet-Ring Cell Gastric Carcinoma Masquerading as Breast Primary.