|Time lag between oscillatory pressure and flow affecting accuracy of forced oscillation technique.|
|Jump to Full Text|
|PMID: 21798069 Owner: NLM Status: MEDLINE|
|BACKGROUND: The forced oscillation technique (FOT) is a simple method for assessing the oscillatory mechanics of the respiratory system. The oscillatory properties, respiratory system resistance (Rrs) and reactance (Xrs), are calculated from the oscillatory pressure/flow relationship. Although the FOT has been a well-established technique, some detailed experimental conditions would be different among institutions.
METHODS: We evaluated whether time lags produced by the experimental conditions such as different positions of the sensors can affect the accuracy of the FOT. If the position of the pressure sensor is different from the flow sensor, a time lag may occur in the measurements. The effect of the time lag was studied by numerical analysis.
RESULTS: Rrs was estimated to be increased and Xrs decreased with an increase in the time lag, especially at a high oscillatory frequency of the medium-frequency range (5-35 Hz). At the high-frequency range (10-500 Hz), Rrs and Xrs were strikingly different in the values of the time lag.
CONCLUSION: A time lag between the oscillatory pressure and flow may be involved in the accuracy of the FOT, suggesting that it needs to be minimized or compensated for with signal processing. Researchers should pay attention to such detailed experimental conditions of the FOT apparatus.
|Junichi Ohishi; Hajime Kurosawa|
Related Documents :
|1147389 - A routine for safe, effective endotracheal suctioning.
10552119 - Hemodynamic patterns of survivors and nonsurvivors during high risk elective surgical o...
1112119 - Prolonged rate of decay of arterial po2 following oxygen breathing in chronic airways o...
14530989 - Effects of induced hypothermia after soft-tissue injury.
21974129 - Quantified lysis of cell-like lipid membranes due to nanoparticle-facilitated cavitation.
7678039 - Effects of graded levels of tissue oxygen pressure on dopamine metabolism in the striat...
21285039 - Resolution of headache and papilledema in idiopathic intracranial hypertension associat...
18164579 - Effects of the knee-chest position on cardiac index and propofol requirements during bi...
145319 - Cardiac hypertrophy and antihypertensive therapy.
|Type: Journal Article; Research Support, Non-U.S. Gov't Date: 2011-07-29|
|Title: Biomedical engineering online Volume: 10 ISSN: 1475-925X ISO Abbreviation: Biomed Eng Online Publication Date: 2011|
|Created Date: 2011-09-06 Completed Date: 2011-11-21 Revised Date: 2013-06-28|
Medline Journal Info:
|Nlm Unique ID: 101147518 Medline TA: Biomed Eng Online Country: England|
|Languages: eng Pagination: 65 Citation Subset: IM|
|Department of Occupational Health, Tohoku University Graduate School of Medicine, Sendai 980, Japan.|
|APA/MLA Format Download EndNote Download BibTex|
Diagnostic Techniques, Respiratory System
Oscillometry / methods*
Respiratory Function Tests / instrumentation*, methods*
Respiratory Physiological Phenomena
Journal ID (nlm-ta): Biomed Eng Online
Publisher: BioMed Central
Copyright ©2011 Ohishi and Kurosawa; licensee BioMed Central Ltd.
Received Day: 13 Month: 5 Year: 2011
Accepted Day: 29 Month: 7 Year: 2011
collection publication date: Year: 2011
Electronic publication date: Day: 29 Month: 7 Year: 2011
Volume: 10First Page: 65 Last Page: 65
Publisher Id: 1475-925X-10-65
PubMed Id: 21798069
|Time lag between oscillatory pressure and flow affecting accuracy of forced oscillation technique|
|Junichi Ohishi1||Email: email@example.com|
|Hajime Kurosawa1||Email: firstname.lastname@example.org|
1Department of Occupational Health, Tohoku University Graduate School of Medicine, Sendai, 980, Japan
The forced oscillation technique (FOT) is a simple method for assessing the oscillatory mechanics of the respiratory system, which has provided important findings in respiratory physiology [1,2]. Currently, the FOT using multi-frequency signals such as pseudo random noise or impulse signals is widely used in clinical research [3-6]. Although the FOT has been a well-established technique, some detailed experimental conditions would be different among institutions. To elucidate the effects of the experimental conditions may be important for more accurate FOT measurements. The purpose of this study was to evaluate whether time lags produced by the experimental conditions such as different positions of the sensors can affect the accuracy of the FOT.
The oscillatory properties, Rrs as the resistive component and Xrs as the reactive one, are calculated by a spectral relationship between the oscillatory pressure and flow. Generally, the oscillatory pressure and flow are measured simultaneously by sensors such as a pressure transducer or pneumotachograph near the subject's mouth, which are analyzed in the signal processing by Fourier transform . In some previous papers, the sensors in the diagram were at different distances from the subject's mouth [2-4]. This difference would cause a time lag in the oscillatory pressure and flow in the signal processing. To evaluate whether the time lag could affect the measurement of the FOT, this effect was examined by numerical analysis as follows.
The respiratory impedance of a subject (Zsub) is often represented by a simple linear model of the resistance (R), inertance (L), and compliance (C). That is,
where ω = 2πf (f = oscillatory frequency), . If the flow signals (f) are recorded τ seconds earlier than the pressure signals (p) due to differences in the positions of the sensors, the respiratory impedance (Zrs) is calculated as follows,
The resonant and cutoff frequencies (f0 and fc respectively) are the properties of Xrs, the former of which is used to evaluate the respiratory system. Fc is only used to determine the Xrs-curve versus oscillatory frequency, which is not actually used in the FOT. Those parameters are defined by
Then, Rrs and Xrs are given by
Here, if R = 1 (standard value without units), f0 = 8 Hz, fc = 3 Hz, and τ varies from 0 to 600 μ seconds (every 200 μ seconds), Rrs and Xrs versus the medium oscillatory frequency (from 5 to 35 Hz) are as shown in figure 1. Rrs was estimated to be increased and Xrs decreased with an increase in τ, especially at the high oscillatory frequency of this range. Figure 2 shows those properties versus a high oscillatory frequency (from 10 to 500 Hz). Rrs and Xrs were strikingly different in the τ values in this frequency range. These results indicate that the time lag between the oscillatory pressure and flow can affect the oscillatory properties, suggesting that errors in the FOT measurements would occur. If the oscillatory pressure is regarded as the transmission of air, a time lag of 200 μ seconds would correspond to a distance of approximately 6.86 cm between the two points (under the condition in which the speed of sound is 343 m/s). Such a distance between the sensors may cause a time lag of several hundred microseconds, which can affect the accuracy of the FOT.
In conclusion, the time lag between the oscillatory pressure and flow may affect the accuracy of the FOT. This result suggests that the time lag needs to be minimized or compensated for in the signal processing for the optimal FOT measurement. Researchers should pay attention to such detailed experimental conditions of the FOT apparatus.
The authors declare that they have no competing interests.
OJ performed the analysis and drafted the manuscript. KH conceived of the study and participated in its design. All authors read and approved the final manuscript.
This study was supported by Grant-in-Aid for Japanese Society for the Promotion of Science Fellows.
|DuBois AB,Brody AW,Lewis DH,Burgess BF Jr,Oscillation mechanics of lung and chest in manJ Appl PhysiolYear: 1956858759413331841|
|Oostveen E,MacLeod D,Lorino H,Farre R,Hantos Z,Desager K,Marchal F,ERS Task Force on Respiratory Impedance MeasurementsThe forced oscillation technique in clinical practice: methodology, recommendations and future developmentsEur Respir JYear: 2003221026104110.1183/09031936.03.0008940314680096|
|Frey U,Forced oscillation technique in infants and young childrenPaediatr Respir RevYear: 2005624625410.1016/j.prrv.2005.09.01016298307|
|Beydon N,Davis SD,Lombardi E,Allen JL,Arets HG,Aurora P,Bisgaard H,Davis GM,Ducharme FM,Eigen H,Gappa M,Gaultier C,Gustafsson PM,Hall GL,Hantos Z,Healy MJ,Jones MH,Klug B,Lødrup Carlsen KC,McKenzie SA,Marchal F,Mayer OH,Merkus PJ,Morris MG,Oostveen E,Pillow JJ,Seddon PC,Silverman M,Sly PD,Stocks J,Tepper RS,Vilozni D,Wilson NM,American Thoracic Society/European Respiratory Society Working Group on Infant and Young Children Pulmonary Function TestingAn official American Thoracic Society/European Respiratory Society statement: pulmonary function testing in preschool childrenAm J Respir Crit Care MedYear: 20071751304134510.1164/rccm.200605-642ST17545458|
|Faria AC,Lopes AJ,Jansen JM,Melo PL,Evaluating the forced oscillation technique in the detection of early smoking-induced respiratory changesBiomed Eng OnlineYear: 200982210.1186/1475-925X-8-22|
|Marchal F,Hall GL,Forced oscillation techniqueEur Respir MonYear: 201047121136|
|Michaelson ED,Grassman ED,Peters WR,Pulmonary mechanics by spectral analysis of forced random noiseJ Clin InvestYear: 1975561210123010.1172/JCI1081981184746|
Previous Document: Australian medical students' perceptions of professionalism and ethics in medical television program...
Next Document: Rigorous assessment and integration of the sequence and structure based features to predict hot spot...