|Statistical agreement of left ventricle measurements using cardiac magnetic resonance and 2D echocardiography in ischemic heart failure.|
|Jump to Full Text|
|PMID: 22367134 Owner: NLM Status: MEDLINE|
|BACKGROUND: The aim of this study was to compare cardiac magnetic resonance imaging (CMR) with 2-dimensional echocardiography (2D echo) in the assessment of left ventricle (LV) function parameters and mass in patients with ischemic heart disease and severely depressed LV function. Although 2D echo is commonly used to assess LV indices, CMR is the state-of-the-art technique. Agreement between these 2 methods in these patients has not been well established.
MATERIAL/METHODS: LV indexed end systolic and diastolic volumes (EDVi and ESVi), indexed mass (LVMi) and ejection fraction (EF) were assessed in 67 patients (12 women), using 2D echo and CMR.
RESULTS: According to statistical analysis (Bland-Altman), 2D echo underestimated LV EDV and ESV and overestimated EF and LVMi compared to CMR. The highest correlation between 2D echo and CMR was found for EDVi (R2=0.73, p<0.0001) and ESVi (R2=0.69, p<0.0001) and the lowest for EF (R2=0.21, p=0.001) and LVMi (R2=0.20, p=0.002). The maximal differences between 2D echo and CMR were found for highest mesurements of LV volumes and mass, and for lowest EF values.
CONCLUSIONS: There is moderate to strong correlation between CMR and 2D echo in the assessment of LV function parameters and mass in patients with ischemic heart failure. Between-method agreement depends on the degree of LV dysfunction. The results of assessment of the severely damaged LV obtained by the use of 2D echo should be interpreted with caution.
|Katarzyna Gruszczyńska; Lukasz J Krzych; Krzysztof S Gołba; Jolanta Biernat; Tomasz Roleder; Marek A Deja; Piotr Ulbrych; Marcin Malinowski; Piotr Janusiewicz; Stanisław Woś; Jan Baron|
Related Documents :
|22011774 - The extent of early left ventricular reverse remodelling is related to midterm outcomes...
22564574 - Ventricular assist device application as a bridge to pediatric heart transplantation: a...
22297084 - Three-dimensional mechanical dyssynchrony and myocardial deformation of the left ventri...
22574544 - Utilizing a registry for st segment elevation myocardial infarction from a tertiary car...
9236454 - Relationships between myocardial bioenergetic and left ventricular function in hearts w...
11175204 - Left ventricular regional diastolic dysfunction in patients with first myocardial infar...
|Type: Journal Article; Research Support, Non-U.S. Gov't|
|Title: Medical science monitor : international medical journal of experimental and clinical research Volume: 18 ISSN: 1643-3750 ISO Abbreviation: Med. Sci. Monit. Publication Date: 2012 Mar|
|Created Date: 2012-02-27 Completed Date: 2012-06-19 Revised Date: 2013-06-26|
Medline Journal Info:
|Nlm Unique ID: 9609063 Medline TA: Med Sci Monit Country: Poland|
|Languages: eng Pagination: MT19-25 Citation Subset: IM|
|Department of Radiology and Nuclear Medicine, Medical University of Silesia, Katowice, Poland. firstname.lastname@example.org|
|APA/MLA Format Download EndNote Download BibTex|
Echocardiography / methods*
Heart Ventricles / pathology*, ultrasonography
Magnetic Resonance Imaging / methods*
Myocardial Ischemia / pathology*, ultrasonography
Journal ID (nlm-ta): Med Sci Monit
Journal ID (iso-abbrev): Med. Sci. Monit
Journal ID (publisher-id): Medical Science Monitor
Publisher: International Scientific Literature, Inc.
© Med Sci Monit, 2012
Received Day: 16 Month: 3 Year: 2011
Accepted Day: 26 Month: 10 Year: 2011
collection publication date: Year: 2012
Electronic publication date: Day: 01 Month: 3 Year: 2012
Volume: 18 Issue: 3
First Page: MT19 Last Page: MT25
PubMed Id: 22367134
Publisher Id: 882507
|Statistical agreement of left ventricle measurements using cardiac magnetic resonance and 2D echocardiography in ischemic heart failure|
|Łukasz J. Krzych2CDEG|
|Krzysztof S. Gołba3ABCDEG|
|Marek A. Deja4BG|
1Department of Radiology and Nuclear Medicine, Medical University of Silesia, Katowice, Poland
21st Department of Cardiac Surgery, Medical University of Silesia, Katowice, Poland
3Department of Cardiology, Medical University of Silesia, Katowice, Poland
42nd Department of Cardiac Surgery, Medical University of Silesia, Katowice, Poland
5Department of Diagnostic Imaging, County Specialist Hospital No. 3, Rybnik, Poland
|Correspondence: Katarzyna Gruszczyńska, Department of Radiology and Nuclear Medicine, Medical University of Silesia, 14 Medykow St., 40-752 Katowice, Poland, e-mail: email@example.com
Systolic left ventricle (LV) function is a basic factor influencing prognosis and decision-making in patients with ischemic heart failure [1,2]. Invasive and non-invasive techniques are used to evaluate cardiac function in the course of ischemic disease. Cardiac magnetic resonance imaging (CMR) is the non-invasive, state-of-the-art technique in the assessment of cardiac function, owing to its excellent soft-tissue contrast, good blood-to-myocardium contrast and sufficient spatial/temporal resolution for cardiac imaging [3,4]. However, in routine clinical settings, transthoracic two-dimensional echocardiography (2D echo) is the most widely available technique and the method of choice from the economic point of view . Despite the many comparisons between CMR and 2D echo in the evaluation of LV function existing in the literature, the data on the agreement of both methods in patients with ischemic heart disease are inconclusive [6–9]. Some authors suggest that CMR and 2D echo were not interchangeable in the assessment of cardiac function . According to others, CMR was the preferred technique, owing to its three-dimensional approach and superior image quality . In recent years, many authors have found better agreement between CMR, two-dimensional and, especially, three-dimensional echocardiography in the assessment of cardiac function [11–14]. However, the results in patients with heart failure are still ambiguous [15,16]. It is also unclear if the between-method agreement depends on the degree of LV dysfunction . The aim of our study was to compare CMR with 2D echo in the assessment of LV volumes, LV mass and ejection fraction (EF) in patients with ischemic heart disease and severely depressed LV function.
The study group consisted of consecutive patients admitted to the cardiology or cardiac surgery departments of our hospital with clinically significant congestive heart failure and low EF, as reported by their referring physicians. The cause of heart failure in all the patients was multi-vessel coronary artery disease with a history of myocardial infarction. The patients were referred for clinically indicated CMR and 2D echo as part of their evaluation. The exclusion criteria were arrhythmia and/or contraindications to CMR (cardiac pacemakers, ferromagnetic implants or claustrophobia). Patients with clinically significant aortic valve disease, mitral stenosis and non-ischemic mitral regurgitation were excluded from the analysis. After taking the exclusion criteria into consideration, the study group consisted of 67 patients (55 men and 12 women). The patients’ data are summarized in Table 1. The mean period of time between CMR and 2D echo was 5±2 days. The study was approved by the local ethics committee and written informed consent was obtained from all participating patients.
CMR studies were performed using a 1.0T MR Unit (Magnetom Harmony, Siemens) with the following technical parameters: magnetic gradients amplitude: 40 mT, slew rate 200 mT/m/s, dedicated four-element phase-array receiver coil. Examination protocol consisted of an SE/T1-weighted sequence, ‘dark blood’ images in the axial plane and a Balanced Steady State Free Precession cine ECG gated sequence. MRI scans were performed while the patients were holding their breath. The imaging parameters were: 19 temporal phases per slice, 24 segments, voxel size 2.1×2.1×8 mm, repetition time 3.0–3.6 ms, echo time 1.2–1.4 ms, flip angle 67°, number of averages 1. To cover the entire left ventricle, slices of 8mm with no interslice gap were planned in short-axis view, perpendicular to the long axis on a four-chamber and a two-chamber view.
The parameters of systolic left ventricle function – LV end-diastolic and end-systolic volume indexes (EDVi and ESVi), EF and LV mass index (LVMi) – were assessed by means of a dedicated workstation (Leonardo, Siemens), using cardiac MRI software (Argus, Siemens). Endocardial and epicardial contours were manually drawn on short-axis images by 2 operators (K.G. and P.U.), who were unaware of the 2D echo results. Papillary muscles and endocardial trabeculations were excluded from LV mass. To correct through-plane motion, basal slices with an incomplete muscular ring were partially contoured.
To compare first and second measurements of 1 observer (intra-observer variability) and to compare the first and the second measurements of 2 independent investigators (inter-observer variability), interclass correlation coefficients were calculated. The results showed almost complete between-measurement correlation with substantial intra-observer agreement (ICC=0.984÷0.998), and strong between-measurement correlation with good inter-observer agreement (ICC=0.828÷0.995).
2D echo exams were performed by 1 operator (J.B.), using a Sonos 7500 scanner coupled with a 2.5 MHz sector transducer (Philips Medical System). EDVi, ESVi and EF were obtained by the biplane Simpson’s method. Calculation of the LV mass was performed using the area-length method . To obtain a wall motion score index, the LV was divided into 16 segments as recommended by the American Society of Echocardiography, and a score was assigned to each segment (1 = normal; 2 = hypokinesis; 3 = severe hypokinesis; 4 = akinesis; 5 = dyskinesis) (18). All the segment scores were then added and divided by the number of segments analyzed.
The values are presented as means and standard deviation for quantitative variables and percentages for qualitative variables. The normality of the quantitative data was confirmed with the Shapiro-Wilk W test. The logarithmic transformation of all non-normally distributed data enabled the standard approach to be used. The between-group comparison was evaluated using Student’s t-test. The correlation between the continuous measures investigated in the 2D echo and CMR was evaluated using Pearson’s correlation coefficient and the coefficient of determinantion (R2). The relationship between variables was then assessed with a linear regression analysis and concordance correlation coefficients (CCC) were used to evaluate agreement . The between-method comparison included a Bland-Altman analysis where the difference CMRvalue − 2D echovalue was plotted against the mean (CMRvalue + 2D echovalue) / 2 . In the between-method agreement assessment, a linear regression equation was used to evaluate whether the differences were dependent on the magnitude of measurements. The results of the statistical tests were considered as statistically significant, with p<0.05. The statistical analysis was performed by means of MedCalc 126.96.36.199 (MedCalc, Belgium).
The comparison of left venticular measures with the use of 2D echo and CMR revealed that ESVi and EDVi were statistically significantly higher, but EF and LVMi were statistically significantly lower by CMR compared to 2D echo (p<0.001 for all). The linear regression analysis revealed a moderate to strong correlation between the 2D echo and CMR measurements. These results are summarized in Table 2. The highest values of the coefficients of determination were found in relation to EDVi (R2=0.73) and ESVi (R2=0.69), and the lowest for EF (R2=0.21) and LVMi (R2=0.20). The coefficients of concordance were also the highest for ESVi (CCC=0.46) and EDVi (CCC=0.41) and the lowest for LVMi (CCC=0.12). The linear regression analysis for the 2D echo and CMR measurements of LV parameters is shown in Figure 1. The Bland-Altman analysis revealed that both LV volumes were statistically significantly underestimated in the 2D echo as compared to the CMR. The effect was seen for all values of EDVi and ESVi. Figure 2A shows the Bland-Altman analysis for EDVi. The mean difference between the CMR and 2D echo was +43.7 ml/m2 (95% CI +37.0; +50.5). The lower limit of agreement was −10.8 ml/m2 (95% CI −22.4; +0.9) and the upper limit was +98.2 ml/m2 (95% CI+88.6; +109.9). Figure 2B shows the Bland-Altman plot for ESVi. The mean difference between the CMR and 2D echo was +41.4 ml/m2 (95% CI +35.2; +42.7), with a lower limit of agreement of −8.9 ml/m2 (95% CI −19.6; +1.9) and an upper limit of +91.7 ml/m2 (95% CI +81.0; +102.5). The between-method agreement for EF is shown in Figure 2C. The mean difference between CMR and 2D echo was −5.7% (95% CI −7.1; −4.2), with the lower limit of agreement equal to −17.4% (95% CI −19.9; −14.9) and an upper limit of +6.0% (95% CI +3.6; +8.5). For EF average values less than 35%, which corresponds to the actual low EF, the 2D echo overestimated the measure, as compared to the CMR. The Bland-Altman analysis also revealed that for the total range of values, LVMi was statistically significantly overstimated in the 2D echo as compared to the CMR (Figure 2D). The mean difference between methods was −79.9 g/m2 (95% CI −94.1; −65.8), the lower limit was −173.4 g/m2 (95% CI −197.7; −149.0) and the upper limit was +13.5 g/m2 (95% CI −10.9; +37.8).
The results of the Bland-Altman regression analysis are summarized in Table 3 and presented in Figure 2. There were noticable proportional errors; in relation to EDVi, ESVi and LVMi the highest between-method differences were found for the highest measurements and in relation to EF for the lowest values. The results confirmed that the agreement between the 2D echo and CMR were highly dependent on the magnitude of the measurements and that stronger decline in LV function results in higher differences between methods.
The aim of the study was to compare CMR and 2D echo in the evaluation of left ventricle function, volumes and mass in patients with severely depressed LV function caused by ischemic heart disease. There are only a few papers comparing these 2 diagnostic modalities in this specific population [10,13,14]. Our results suggest a moderate to strong correlation between 2D echo and CMR measurements. We found that this correlation was highly dependent on the degree of LV dysfunction for all of the parameters analyzed.
We report a strong correlation between 2D echo and CMR in the evaluation of LV volumes. Our results of the Bland-Altman analysis revealed that LV volumes were significantly underestimated by the 2D echo, as compared to the CMR. These results are similar to those obtained in other studies. According to Bellenger et al. the 2D echo underestimated LV volumes (mean difference: EDV: 95 ml; ESV: 133 ml) and the limit of agreement was wider than in our study . Gardner et al. reported that the 2D echo underestimated EDVi by 69 ml/m2 and ESVi by 35 ml/m2. The source of this discrepancy is the different techniques used by these 2 cardiac imaging methods. 2D ECHO is a real-time imaging modality, with better temporal resolution than CMR, which was an advantage in our population with ischemic wall motion abnormalities . On the other hand, CMR has better spatial and contrast resolution, which is especially important in cases with thin LV wall due to ischemia, as in our study . The 2D technique of the evaluation of LV function in ECHO is the source of error from foreshortened views and inaccurate geometric modeling . However, multislice CMR technique, based on the analysis of the stack of short-axis images, is also difficult in patients with aneurysmal LV dilatation, due to poor endocardial definition near the apex, as the result of partial-volume effect . The agreement between 3DECHO and CMR in the literature seems to be better. In a single-centre study carried out by Sugeng et al, the 3DECHO only slightly underestimated LV volumes (5–6 ml, 95% limit of agreement) . However, in a multi-centre validation paper by Mor-Avi et al, the 2D echo-derived volumes, as compared to the CMR, showed significant negative biases – (−29% for EDV and −27% for ESV) .
In our study, the correlation between the CMR and 2D echo in EF (R2=0.21) was weaker than for LV volumes, but was still statistically significant. The mean difference between the CMR and the 2D echo was −5.7%. These values are consistent with those described in the literature. The 2D echo overestimates EF by 10%, according to Nowosielski  and by 4% according to Gardner . Sugeng et al. reported an agreement between CMR and 3DECHO at a level of 8% . EF evaluation in CMR and 2D echo depends on the accuracy of LV ESV and EDV measurements. Even small differences in volume measurements may result in increased error of LV EF estimates . In our study, LV function and mass in the 2D echo were calculated using the Simpson rule . According to Bellenger, this method is the most accurate for volumetric LV quantification as compared to the CMR . However, this method is less reliable when the geometrical model does not correspond to the true ventricular cavity anatomy in patients with LV ischemic remodeling, as in our study.
The CMR software we used did not allow for the correction of the LV contours on the long axis, which was a source of error. To correct through-plane motion, basal slices with an incomplete muscular ring of less then 75% were only partially contoured. According to Kirschbaum et al, the addition of the long-axis to the short-axis contours during the CMR post-processing limits the extent of the volume at the base of the heart, and reduces inter-study variability . We found that tracing LV contours in CMR in the regions of LV apex in patients with LV aneurysm was difficult due to the low contrast resolution between blood pool and muscle. We assume that regional wall-motion and wall thickness abnormalities due to ischemia impact the accuracy of endocardial contouring in both CMR and 2D ECHO, which could be a source of the discrepancy between the 2 methods. As we examined the population with severe LV dysfunction, the differences between the 2 methods became larger compared to other studies.
In our study, the correlation between the CMR and 2D echo for cardiac mass measurements was moderate. LVMi was significantly overestimated in 2D echo as compared to CMR. We have described a significant discrepancy between these 2 methods in calculating LV mass (R2=0.20). The well-recognized formula used in 2D ECHO to calculate LV mass relies on a geometric assumption of uniform chamber size and shape . LV mass cannot be measured properly in heavily dilated hearts by means of 2D echo because they do not fit the pyramidal scan volume. Previous authors have highlighted the fact that the accuracy of LV mass measurements in 2D echo is limited due to difficulties in obtaining anatomically corrected apical views [24–26]. This may result in the foreshortening of the LV long axis, especially in patients with a dilated LV. In a study carried out on 83 patients (33 of them with wall motion abnormalities), Pouleur et al. found a reasonably good correlation between echocardiography and CMR. However, only 14% of the patients in that study had an EF of 36% . In a study covering 25 patients, Chuang et al. reported a bias of 16g between the 2D echo and the CMR . Caiani et al. found that 2D echo underestimated LV mass comparing to CMR . He postulated the need for further investigations of patients with LV dilatation and aneurysms.
The agreement between the 2D echo and the CMR relied on the number of imaging planes. Taking into account slice thickness and interslice distance, LV mass can be obtained by multiplication of volume with the specific density of the myocardium . However, we examined a population of patients after myocardial infarction, with a large area of myocardial scar, which has a different specific density than healthy myocardium. This could affect the accuracy of LV mass calculation in our study.
To decrease the measurement error, endocardial contouring of CMR short axis images in our study was performed by a single operator. Papillary muscles and LV myocardial trabeculae were included into the LV volume, which increases measurement reproducibility; however, this can overestimate LV volumes and decrease LV mass [27–30].
A medium-field CMR unit was used in this study (1.0 T). In the literature, we were unable to find any comparison between this type of scanner and the commonly used 1.5 T CMR systems in the assessment of LV function. The decreased field strength, which could decrease the signal-to-noise ratio, may be a study limitation. Furthermore, the 8mm slice thickness could result in partial volume effects.
We found that the agreement between CMR and 2D echo in the evaluation of the LV volumes was highly dependent on the magnitude of the measurements. The highest between-method difference was found for the highest volumes. We find this result to be clinically important, taking into account the fact that the 2D echo underestimates LV volumes in comparison to the CMR. This means that, in patients with severe ischemic LV dilatation and aneurysm, LV volume as measured by 2D echo should be analyzed with special caution, as the difference between the 2 methods could be at its highest. Similarly, if the severely damaged LV was examined using 2D echo, the latter overestimates EF proportionately with greater degree of LV damage. Again, this result is clinically important. In patients with the most severe ischemic LV dysfunction, the EF may be even lower than a value obtained using the 2D echo .
A similar relationship was found for LV mass. There were noticeable proportional errors between both methods, depending on the LVMi. The increase in the mass of LV correlated well with the increase of LV dysfunction and we found that the agreement between the 2D echo and CMR was highly dependent on the degree of LV dysfunction – the highest differences existed in patients with the highest LV myocardial mass.
This problem is not really recognized in the literature. We were able to find only a few studies analyzing the correlation between CMR and 2D echo on the basis of the degree of LV dysfunction. Gardner et al. found that the differences between the 2 methods in the measurement of LV volumes increased with lower EF values, with a threshold of 46%. However, inter-modality differences were not statistically dependent on the EF subgroup [14,32]. In an article comparing CMR and 2D echo in patients with acute MI, Nowosielski suggested that slight myocardial impairment (EF >55%) might be difficult to detect by 2D echo; thus, correlation between transesophageal echocardiography and CMR might be more significant in a more severely injured myocardium . Contrary to this, in a multi-centre study validating the 3D echo vs. the CMR, Mor-Avi et al. found a good agreement between the CMR and the 3D echo in the measurement of EF and LV volumes in patients with an EF more than 35% . With the exception of works by Bellenger (the mean EF by CMR was 30%) and Mor-Avi (a wide range of EF with the lowest values starting from 20%), LV function was only moderately impaired in the other papers published (mean EF from 45% to 55%) [9,16]. Moreover, the population described with ischemic heart failure was also smaller than ours.
Our study has several limitations. The size of evaluated sample is limited, and it might require more data to confirm our results. For a comprehensive study, an experimental invasive validation would be important, and the lack of such validation is a limitation of our results.
There is moderate to strong correlation between CMR and 2D echo in the assessment of LV function parameters and mass in patients with severe ischemic left ventricular dysfunction. Unfortunately, between-method agreement strongly depends on the degree of LV dysfunction. Differences in LV indices are higher in patients with more severe LV dysfunction. The results of assessment of the severely damaged LV obtained by the use of 2D echo should be interpreted with caution.
fn8-medscimonit-18-3-mt19Source of support: This work was supported by the Polish Ministry of Science and Higher Education [Medical University of Silesia grant number KNW-1-036/08]
|1.||Hobbs FD,Roalfe AK,Davis RC,et al. Prognosis of all-cause heart failure and borderline left ventricular systolic dysfunction: 5 year mortality follow-up of the Echocardiographic Heart of England Screening Study (ECHOES)Eur Heart JYear: 20072811283417459902|
|2.||Kuch M,Janiszewski M,Mamcarz A,et al. Major adverse cardiac event predictors in survivors of myocardial infarction with asymptomatic left ventricular dysfunction or chronic heart failureMed Sci MonitYear: 2009156PH404819478711|
|3.||Hundley WG,Bluemke DA,Finn JP,et al. ACCF/ACR/AHA/NASCI/SCMR 2010 Expert Consensus Document on Cardiovascular Magnetic Resonance. A Report of the American College of Cardiology Foundation Task Force on Expert Consensus DocumentsJ Am Coll CardiolYear: 20105526146220513610|
|4.||Pennell DJ,Sechtem UP,Higgins CB,et al. Clinical indications for cardiovascular magnetic resonance (CMR): Consensus Panel reportJ Cardiovasc Magn ResonYear: 200467276515646878|
|5.||Dickstein K,Cohen-Solal A,Filippatos G,et al. ESC Guidelines for the diagnosis and treatment of acute and chronic heart failure 2008: the Task Force for the Diagnosis and Treatment of Acute and Chronic Heart Failure 2008 of the European Society of Cardiology. Developed in collaboration with the Heart Failure Association of the ESC (HFA) and endorsed by the European Society of Intensive Care Medicine (ESICM)Eur Heart JYear: 200829238844218799522|
|6.||Bellenger NG,Burgess MI,Ray SG,et al. Comparison of left ventricular ejection fraction and volumes in heart failure by echocardiography, radionuclide ventriculography and cardiovascular magnetic resonance; are they interchangeable?Eur Heart JYear: 20002113879610952828|
|7.||Lyne JC,Pennell DJ. Cardiovascular magnetic resonance in the quantitative assessment of left ventricular mass, volumes and contractile functionCoron Artery DisYear: 2005163374316118538|
|8.||Darasz KH,Underwood SR,Bayliss J,et al. Measurement of left ventricular volume after anterior myocardial infarction: comparison of magnetic resonance imaging, echocardiography, and radionuclide ventriculographyInt J Cardiovasc ImagingYear: 2002181354212108909|
|9.||Bellenger NG,Marcus NJ,Rajappan K,et al. Comparison of techniques for the measurement of left ventricular function following cardiac transplantationJ Cardiovasc Magn ResonYear: 200242556312074140|
|10.||Wassmuth R,Schulz-Menger J. Cardiac magnetic resonance in acute myocarditis and dilated cardiomiopathyMed Sci MonitYear: 2009155RA1111519396051|
|11.||Soliman OI,Kirschbaum SW,van Dalen BM,et al. Accuracy and reproducibility of quantitation of left ventricular function by real-time three-dimensional echocardiography versus cardiac magnetic resonanceAm J CardiolYear: 20081027788318774006|
|12.||Pouleur AC,le Polain de Waroux JB,Pasquet A,et al. Assessment of left ventricular mass and volumes by three-dimensional echocardiography in patients with or without wall motion abnormalities: comparison against cine magnetic resonance imagingHeartYear: 20089410505717974699|
|13.||Sugeng L,Mor-Avi V,Weinert L,et al. Quantitative assessment of left ventricular size and function: side-by-side comparison of real-time three-dimensional echocardiography and computed tomography with magnetic resonance referenceCirculationYear: 20061146546116894035|
|14.||Gardner BI,Bingham SE,Allen MR,et al. Cardiac magnetic resonance versus transthoracic echocardiography for the assessment of cardiac volumes and regional function after myocardial infarction: an intrasubject comparison using simultaneous intrasubject recordingsCardiovasc UltrasoundYear: 200973819689809|
|15.||Brzozowska-Czarnek A,Urbanik A,Karcz P,et al. Usefulness of Cardiac Magnetic Resonance Delayed Enhancement technique in the assessment of the postinfarction scar size and the function of the left ventricle in patients after myocardial infarction, treated with Primary Percutaneous Coronary InterventionPol J RadiolYear: 2009744754|
|16.||Mor-Avi V,Jenkins C,Kuhl HP,et al. Real-time 3-dimensional echocardiographic quantification of left ventricular volumes: multicenter study for validation with magnetic resonance imaging and investigation of sources of errorJACC Cardiovasc ImagingYear: 200814132319356461|
|17.||Lang RM,Bierig M,Devereux RB,et al. Recommendations for chamber quantificationEur J EchocardiogrYear: 200677910816458610|
|18.||Schiller NB,Shah PM,Crawford M,et al. Recommendations for quantitation of the left ventricle by two-dimensional echocardiography. American Society of Echocardiography Committee on Standards, Subcommittee on Quantitation of Two-Dimensional EchocardiographyJ Am Soc EchocardiogrYear: 19892358672698218|
|19.||Lin LI. A concordance correlation coefficient to evaluate reproducibilityBiometricsYear: 198945255682720055|
|20.||Bland JM,Altman DG. Statistical methods for assessing agreement between two methods of clinical measurementLancetYear: 1986327307102868172|
|21.||Devereux RB,Alonso DR,Lutas EM,et al. Echocardiographic assessment of left ventricular hypertrophy: comparison to necropsy findingsAm J CardiolYear: 198657450582936235|
|22.||Nowosielski M,Schocke M,Mayr A,et al. Comparison of wall thickening and ejection fraction by cardiovascular magnetic resonance and echocardiography in acute myocardial infarctionJ Cardiovasc Magn ResonYear: 2009112219589148|
|23.||Kirschbaum SW,Baks T,Gronenschild EH,et al. Addition of the long-axis information to short-axis contours reduces interstudy variability of left-ventricular analysis in cardiac magnetic resonance studiesInvest RadiolYear: 2008431618097271|
|24.||de Simone G,Gottdiener JS,Chinali M,Maurer MS. Left ventricular mass predicts heart failure not related to previous myocardial infarction: the Cardiovascular Health StudyEur Heart JYear: 2008297414718204091|
|25.||Chuang ML,Salton CJ,Hibberd MG,et al. Relation between number of component views and accuracy of left ventricular mass determined by three-dimensional echocardiographyAm J CardiolYear: 20079913212417478166|
|26.||Caiani EG,Corsi C,Sugeng L,et al. Improved quantification of left ventricular mass based on endocardial and epicardial surface detection with real time three dimensional echocardiographyHeartYear: 2006922131915890763|
|27.||Zagrodzka M,Szmit S,Brzeziński J,Opolski G. The role of MR imaging and cardio-pulmonary exercise test (CPET) in oncology patient – a case reportPol J RadiolYear: 2009746065|
|28.||Gruszczyńska K. Three arteries – no problemPol J RadiolYear: 201075Suppl 18990|
|29.||Wiliński J,Czarnecka D,Wojciechowska W,et al. Baseline tissue Doppler imaging-derived echocardiographic parameters and left ventricle reverse remodelling following cardiac resynchronization therapy introductionArch Med SciYear: 2011758132222291826|
|30.||Kirschbaum S,Aben JP,Baks T,et al. Accurate automatic papillary muscle identification for quantitative left ventricle mass measurements in cardiac magnetic resonance imagingAcad RadiolYear: 20081512273318790393|
|31.||Sekhri V,Sanal S,DeLorenzo LJ,et al. Cardiac sarcoidosis: a comprehensive reviewArch Med SciYear: 2011745465422291785|
|32.||Kurek P. Diagnostic imaging of heart from a clinicists standpointPol J RadiolYear: 201075Suppl 164|
Keywords: cardiac magnetic resonance imaging, two-dimensional echocardiography, end-stage ischemic heart failure.
Previous Document: Takotsubo syndrome in a patient after renal transplantation.
Next Document: The relationship between family socioeconomic condition and childhood injury frequency in selected l...