|Negative- versus positive-pressure ventilation in intubated patients with acute respiratory distress syndrome.|
|Jump to Full Text|
|PMID: 22386062 Owner: NLM Status: Publisher|
|ABSTRACT: INTRODUCTION: Recent experimental data suggest that continuous external negative-pressure ventilation (CENPV) results in better oxygenation and less lung injury compared to continuous positive-pressure ventilation (CPPV). The effects of CENPV in patients with acute respiratory distress syndrome (ARDS) remain unknown. METHODS: We compared 2h CENPV with 2h CPPV. The six intubated patients developed ARDS after pulmonary thrombectomy (n=1), aspiration (n=3), sepsis (n=1) or both (n=1). We used a tidal volume of 6ml/kg predicted bodyweight and matched lung volumes at end-expiration. Hemodynamics were assessed with the pulse contour cardiac output (PiCCO) system and, pressure measurements were referenced to atmospheric pressure. RESULTS: CENPV resulted in better oxygenation compared to CPPV (median paO2/FIO2, minimum-maximum, 345mmHg (183-438) vs. 256mmHg (123-419), P<0.05). Tank pressures were -32.5cmH2O (-30- -43) at end-inspiration and -15cmH2O (-15- -19) at end-expiration. Inspiratory transpulmonary pressures decreased (P=0.04) and airway pressures were considerably lower at inspiration (-1.5cmH2O (-3-0) vs. 34.5cmH2O (30-47),P=0.03) and expiration (4.5cmH2O (2-5) vs. 16cmH2O (16-23), P=0.03). During CENPV, intraabdominal pressures decreased from 20.5mmHg (12-30) to 1mmHg (-7-5) (P=0.03). Arterial pressures decreased by approximately 10mmHg and central venous pressures by 18mmHg. Intrathoracic blood volume indices and cardiac indices increased at the initiation of CENPV by 15% and 20% (P<0.05), respectively. Heart rate and extravascular lung water indices remained unchanged. CONCLUSIONS: CENPV with a tankrespirator improved gas-exchange in patients with ARDS at lower transpulmonary, airway and intraabdominal pressures and, at least initially improving hemodynamics. Our observations encourage the consideration of further studies on the physiological effects and the clinical effectiveness of CENPV in patients with ARDS.|
|Konstantinos Raymondos; Ulrich Molitoris; Marcus Capewell; Bjoern Sander; Thorben Dieck; Joerg Ahrens; Christian Weilbach; Wolfgang Knitsch; Antonio Corrado|
|Type: JOURNAL ARTICLE Date: 2012-3-2|
|Title: Critical care (London, England) Volume: 16 ISSN: 1466-609X ISO Abbreviation: - Publication Date: 2012 Mar|
|Created Date: 2012-3-5 Completed Date: - Revised Date: -|
Medline Journal Info:
|Nlm Unique ID: 9801902 Medline TA: Crit Care Country: -|
|Languages: ENG Pagination: R37 Citation Subset: -|
|APA/MLA Format Download EndNote Download BibTex|
Journal ID (nlm-ta): Crit Care
Journal ID (iso-abbrev): Crit Care
Publisher: BioMed Central
Copyright ©2012 Raymondos et al.; licensee BioMed Central Ltd.
Received Day: 23 Month: 8 Year: 2011
Revision Received Day: 27 Month: 12 Year: 2011
Accepted Day: 2 Month: 3 Year: 2012
Print publication date: Year: 2012
Electronic publication date: Day: 2 Month: 3 Year: 2012
Volume: 16 Issue: 2
First Page: R37 Last Page: R37
PubMed Id: 22386062
Publisher Id: cc11216
|Negative- versus positive-pressure ventilation in intubated patients with acute respiratory distress syndrome|
|Konstantinos Raymondos1||Email: Raymondos.Konstantinos@mh-hannover.de|
|Ulrich Molitoris1||Email: email@example.com|
|Marcus Capewell1||Email: Capewell.Marcus@mh-hannover.de|
|Björn Sander1||Email: bjoernX@gmx.de|
|Thorben Dieck1||Email: firstname.lastname@example.org|
|Jörg Ahrens1||Email: email@example.com|
|Christian Weilbach2||Email: firstname.lastname@example.org|
|Wolfgang Knitsch3||Email: Knitsch.email@example.com|
|Antonio Corrado4||Email: firstname.lastname@example.org|
1Anaesthesiology and Intensive Care Medicine, Medical School Hanover, Carl-Neuberg-Strasse 1, D-30625 Hanover, Germany
2Anaesthesiology, St-Josefs-Hospital, Krankenhausstraße 13, D-49661 Cloppenburg, Germany
3General, Visceral and Transplantation Surgery, Medical School Hanover, Hanover, Carl-Neuberg-Strasse 1, D-30625 Hanover, Germany
4Unita' di Terapia Intensiva Pneumologica e, Fisiopatologia Toracica, DAI, Specialità medico-Chirurgiche, Azienda Ospedaliero-Universitaria Careggi, Padiglione San Luca, Via di San Luca 1, I-50136 Florence, Italy
Acute respiratory distress syndrome (ARDS) is usually treated with invasive continuous positive-pressure ventilation (CPPV) , which can aggravate both lung injury and multisystem organ failure . Studies of mechanical ventilation in patients with ARDS have focused on low tidal volume and high positive end-expiratory pressure (PEEP) [2-5]. Less injurious low tidal volume can lead to impaired oxygenation , and even very high PEEP can be insufficient to maintain lung volume in patients with severe ARDS . Other approaches using mechanical ventilation have not been shown to further improve outcome, and mortality in patients with ARDS still reaches 50% .
External negative-pressure ventilation with tank respirators is very successful in treating patients with chronic obstructive pulmonary disease , but there are no data regarding patients with hypoxaemic acute respiratory failure. Recent experimental data suggest that continuous external negative-pressure ventilation (CENPV) may distend lungs in a fundamentally different manner from CPPV and may result in better oxygenation and less lung injury at lower transpulmonary pressures . Extrapolation to patients is difficult, and to date only continuous external negative-pressure (CENP) has been applied in three ARDS patients who breathed spontaneously in Emerson tank respirators [8-10]. Furthermore, cuirass  or poncho wrap systems [12,13] have been used for CENP during intermittent positive-pressure ventilation (IPPV), which resulted in improved cardiac output [11-13]. However, both cuirass and poncho wrap systems decrease chest wall compliance when they are affixed to the body [11-13], and effective ventilation in patients with ARDS has not been reported with either these systems or with tank respirators.
We speculated that, similarly to recent experimental data, CENPV with a tank respirator would also result in better oxygenation in intubated patients with ARDS, even when low tidal volumes were used. Therefore, we performed a physiologic study to compare CENPV with CPPV using matched lung volumes at end expiration and matched low tidal volumes. Favourable physiologic effects may help to promote CENPV as an applicable and even noninvasive ventilatory mode for patients with ARDS.
The study was approved by the ethical committee of our institution, and informed consent was obtained from the patients' next-of-kin. We studied six intubated patients between January 2001 and January 2002. Technical personnel and approaches did not change during the study period. Within 12 hours before study entry, all patients met ARDS criteria . Their clinical characteristics, Simplified Acute Physiology Score II scores  and respiratory settings at that time are listed in Table 1. The patients were placed under sedation analgesia and did not breathe spontaneously. We performed a recruitment manoeuvre as described below to standardise the history of lung volume , and, to achieve more comparable conditions, we adjusted PEEP to 16 cmH2O in patients 1 through 5 (and to 23 cmH2O in patient 6). The patients were ventilated with a lung-protective strategy using a tidal volume of 6 ml/kg predicted body weight (as calculated in ).
We compared 2-hour CENPV using a tank respirator with 2-hour CPPV using biphasic positive airway pressure/airway pressure release ventilation with an Evita 1 ventilator (Dräger, Lübeck, Germany). We randomised the sequence of the ventilatory mode to balance the effects of the previous ventilation period. The six patients were randomized to receive 2 hours of CENPV first and then 2 hours of CPPV (n = 3 patients) or 2 hours of CPPV first and then 2 hours of CENPV (n = 3 patients) in an unchanged supine position. Between both ventilatory modes, we returned the ventilation, ventilation is the correct word to baseline, and the whole experiment lasted 6 to 7 hours.
We matched tidal volume, respiratory frequency and the lung volume at end expiration that represented the difference between functional residual capacity (FRC) with and without end-expiratory pressure. The ratio of inspiration to expiration was 1:1. After a recruitment manoeuvre of six deep breaths with an inspiratory pressure of 60 cmH2O, the lung volume at end expiration was measured by a sudden release of the relevant positive or negative pressure at end expiration with a spirometer (Volumeter 3000; Dräger). This step was repeated two more times, and the average value of the three measurements was calculated. Next, lung volumes at end expiration of the second ventilation mode were matched to those measured with the first one. To find out the corresponding pressure at end expiration, we first measured the lung volume at a similar positive or negative pressure. We then increased or decreased pressure according to the achieved volumes and repeated the measurement manoeuvre until the difference between lung volumes at end expiration was smaller than 50 ml. We then performed three measurements again as described above.
To achieve comparable conditions, we recruited the lungs again immediately before CPPV or CENPV with 25 deep breaths applied during 1 minute using inspiratory peak pressures of 60 cmH2O. We performed this manoeuvre to standardise the history of lung volume  to improve the comparability between both ventilatory modes. We used a time interval of 1 minute to allow us to take haemodynamic measurements during the recruitment manoeuvre using the PiCCO system (Pulsion Medical Systems, Munich, Germany).
Thanks to an initiative of Prof Ina Pichlmayr the tank respirator and the pump aggregate to generate negative pressures had been manufactured at our institution during the early 1980s . The transparent plastic tank has been used in various clinical settings in patients without endotracheal tubes  with the head placed outside the tank as commonly practiced [6-10]. As we were studying intubated patients, it was not necessary to place the head outside the tank (Figure 1). Covering the whole body including the head avoided several problems such as air leakage at the neck and improved the practicability of using the tank respirator. Nursing care was very limited, and therefore the whole tank was removed when complex nursing care was necessary. Similar to classic tank respirators [6-8], apertures at both sides allowed access to the patients, which enabled the use of nursing procedures such as endotracheal tube suctioning. In case of an emergency such as cardiac arrest, the plastic tank can be removed within a few seconds.
During CENPV, the inspiratory changes of airway flow caused by the tank respirator triggered the conventional mechanical ventilator that delivered flow at a peak inspiratory pressure of 5 cmH2O in the pressure support mode of the Evita 1 ventilator. The conventional ventilator was set to 0 cmH2O at end expiration. The ventilatory circuits, arterial and central venous lines, gastric tube, urine catheter, electrocardiographic leads, pulse oximeters and other connections to the patient were led out of the tank via apertures from its bottom.
Airway pressures were measured via the side hole of a modified Swan-Ganz catheter that was introduced via the endotracheal tube and placed into the trachea 1 cm distal to the tip of the tube. Oesophageal pressures were measured with a conventional balloon catheter system (CP-100 Pulmonary Monitor; BiCore Monitoring Systems, Irvine, CA, USA). The oesophageal balloon catheter was passed to a depth of 60 cm, and placement of the balloon in the stomach was confirmed by a transient increase in pressure during gentle compression of the abdomen. We then withdrew the catheter until oesophageal placement was confirmed by the presence of cardiac artefacts and pressure changes during tidal ventilation . Intraabdominal pressure was obtained by measuring the pressure in the bladder via the urine catheter after filling the empty bladder with 50 ml of saline using the midaxillary level as the reference line . All pressure measurements were referenced to atmospheric pressure outside the tank .
Cardiac index, intrathoracic blood volume index, extravascular lung water index and stroke volume variation were assessed by thermal dilution using the PiCCO system with an arterial catheter inserted into a femoral artery. The Wilcoxon signed-rank test was used to compare values between CENPV and CPPV, and P-values less than 0.05 were considered significant.
At the beginning of CENPV and CPPV, gas exchange was similar (Figure 2). During CENPV, oxygenation improved impressively compared to the corresponding values during CPPV. The mean arterial-to-inspired oxygen pressure ratio (PaO2/FiO2 ratio) increased by 92 mmHg (40%) after 1 hour and by 76 mmHg (30%) after 2 hours. However, the individual responses varied considerably between patients during both CPPV and CENPV (Figure 2). Furthermore, arterial carbon dioxide pressure (PaCO2) decreased and pH increased during CENPV, but statistical significance was reached only at 1 hour (Figure 2).
Individual data of each patient regarding lung volumes and pressures are shown in Tables 2 and 3. Lung volumes were well-matched and did not differ between CPPV and CENPV (Table 2). During CENPV, intraabdominal pressures decreased by 15 to 26 mmHg (Table 2 Figure 3). Endotracheal airway pressures decreased by at least 30 cmH2O at inspiration and by at least 11 cmH2O at expiration (Figure 3), and inspiratory transpulmonary pressures (airway pressure minus oesophageal pressure) were also significantly lower during CENPV (Table 3). In contrast, transrespiratory system pressures (airway pressure minus tank pressure) during CENPV were similar at inspiration and 1 to 4 cmH2O higher at expiration (Table 3) as a result of a short peak of endotracheal airway pressure at the beginning of expiration (Figure 3).
Central venous pressures decreased more than arterial pressures during CENPV (Figure 4). Simultaneously, the intrathoracic blood volume index increased by 15% and the cardiac index increased by 20%, whereas no differences were found after 1 and 2 hours (Figure 4). After 2 hours of CENPV, the heart rate ranged between 57 and 126 beats/minute (median = 89 beats/minute), and the extravascular lung water index varied between 6 and 14 ml/kg (median = 9.5 ml/kg), and these parameters also did not differ between CPPV and CENPV. During the recruitment manoeuvre, no relevant impairments were observed and all changes returned to baseline immediately.
CENPV improved gas exchange considerably compared to CPPV, which was achieved at matched tidal and end-expiratory lung volumes with lower airway, intraabdominal and transpulmonary pressures, and at least initially improving haemodynamics. Values for inspiratory airway pressures during CPPV were similar compared with tank pressures during CENPV, although negative inspiratory tank pressures were reached only at end inspiration. The matching of end-expiratory lung volumes was obtained with an end-expiratory negative pressure of -15 cmH2O, which corresponded to a PEEP value of 16 cmH2O in five of six patients. This constancy was surprising and probably indicates very similar degrees of lung injury. Concordantly, the patients had very similar PaO2/FiO2 values at a PEEP of 16 cmH2O immediately before the measurement period. With less efficient cuirass or poncho wrap systems, higher pressure values were necessary to achieve the same end-expiratory lung volumes compared to positive pressures [11,12]. Concordantly, when these systems were used to apply continuous negative pressure during IPPV in patients with lung injury, gas exchange either did not improve [11,12] or even deteriorated when end-expiratory lung volumes were not matched . We matched lung volumes, and the PaO2/FiO2 ratio increased impressively during CENPV, which may indicate alveolar recruitment with a decrease in pulmonary shunting.
On the contrary, even relatively high PEEP values used in the present study were apparently still insufficient to maintain recruited lung volumes during CPPV. In surfactant-depleted rabbits, Grasso et al. also matched end-expiratory lung volumes and observed better oxygenation during CENPV than during CPPV . This effect was associated both with more aerated lung tissue and less lung injury after 2.5 hours with the use of high tidal volumes of 12 ml/kg . In our pilot study, we measured neither aerated lung tissue nor FRC and we did not assess markers of lung injury. However, we speculate that, similarly to the observations of Grasso et al. , the improved oxygenation during CENPV observed in the present study could also be associated with alveolar recruitment and an increase in FRC. Increased FRC during CENPV would reduce alveolar strain (ratio between tidal volume inflated and FRC) , and improved lung recruitability during CENPV might be associated with less injurious intratidal alveolar opening and closing of lung tissue . Grasso et al. assumed that CENPV might be more effective and less injurious because of more homogeneous distension of the lung as negative pressure is distributed across a broad surface of the chest wall and abdomen .
Concordantly, at given levels of transpulmonary pressure, Grasso et al. observed greater end-expiratory volumes, and transpulmonary pressures were lower when corresponding positive and negative pressure values were compared . Our data seem to confirm their observations, although we did not measure transpulmonary pressures under static conditions. Therefore, and because we did not measure changes in transpulmonary pressure at 0 cmH2O at expiration as described by Chiumello et al. , we did not assess the global average lung stress in the present pilot study. Similarly to other experimental data , however, transpulmonary pressures were lower during CENPV at end inspiration; therefore, we speculate that lung stress could be lower as well in comparison to CPPV. Concordantly with the observations made by Grasso et al. , our data also suggest that the development of transpulmonary distending pressures may substantially differ during CENPV compared to CPPV. This may be associated with different regional pleural pressure gradients throughout the lungs that are poorly represented on the basis of just one value of transpulmonary pressure .
During CENPV, transrespiratory system pressures (TRP) (airway pressure minus tank pressure) were about 3 cmH2O lower at inspiration and about 3 cmH2O higher at expiration as endotracheal airway pressures became positive during CENPV, probably due to the expiratory resistance of the endotracheal tube. These slightly higher TRP values at expiration may be sufficient to explain the improved oxygenation observed during CENPV.
The TRP differences between CENPV and CPPV, which varied considerably at inspiration and expiration, might be cause by different distributions of positive and negative pressures, depending on individual differences in pulmonary mechanics. Despite our study design, in which we used matched tidal volumes and randomization of ventilatory modes, we found the same TRP value of 31 cmH2O in three patients, which may also reflect very similar degrees of lung injury.
The high intraabdominal pressures  decreased by 20 mmHg during CENPV. This may counteract the effects of high intraabdominal pressures such as cranial shifts of the diaphragm with consequent lung volume reduction, reduced lymphatic flow and lung oedema formation . Intraabdominal perfusion pressure improved as the mean arterial pressure decreased only by 10 mmHg, which could be beneficial, especially when visceral blood flow is impaired.
In comparison to arterial pressures, the quite elevated central venous pressures decreased more extensively during CENPV, indicating the relatively greater impact on the venous circulation than on the arterial circulation, where the vessel tone is stronger. When central venous pressures decreased during CENPV, the high intrathoracic blood volume indices further increased by 15%, reflecting improved venous return, and the cardiac index improved considerably by 20%. Compared to CPPV, this improved venous return may result in less alteration of mixed venous oxygen content and therefore may contribute to maintaining higher levels of arterial oxygen content during CENPV.
In the present study, the heart rate also remained unchanged, indicating that the greater transpulmonary blood flow during CENPV apparently resulted from higher stroke volumes. Borrelli et al. observed very similar increases in cardiac output at lower intrathoracic blood volumes when a poncho was used to apply continuous negative pressure during IPPV . The greater preload in the present study outweighed the effects on afterload, which increased when intrathoracic pressures decreased. During CENPV, both central venous and intraabdominal pressures decreased, which could result in similar pressure gradients for venous return compared to CPPV. In agreement with this finding, Grasso et al. did not observe changes in cardiac output when a whole-body device was used, but they did when negative pressure was applied to the chest only .
CENPV has been suspected to increase extravascular lung water compared to CPPV as a result of more negative pleural and interstitial pressures and because of higher left ventricular filling after enhanced venous return [24,25]. In our present study, extravascular lung water indices did not differ during CENPV compared to CPPV, which has also been observed in experimental studies [7,24,25]. Lung water can increase with PEEP by decreasing lung lymph flow, which has been attributed to compressed pulmonary lymphatic vessels . These compressions do not occur under CENPV and may outweigh other effects, resulting in lung water values similar to those associated with CPPV.
The very small number of patients in this study represents its main limitation and primary source of errors. In four of the six patients, pulmonary aspiration of gastric content was either the main or one contributory predisposing factor in the development of ARDS. Extrapolating this observation to patients with other predisposing factors must be done with caution. In any case, the considerable variation in PaO2/FiO2 responses to CENPV was apparently independent of the underlying disease, the efficiency of the prior lung recruitment manoeuvre or the severity of lung injury. Interestingly, increased oxygenation in response to placement in the prone position also was not related to lung recruitability in response to positive pressures . Finally, at this stage the reasons for the different individual responses to CENPV remain unclear.
As a possibly less injurious and more effective mode of ventilation, CENPV appears especially attractive when the potential to eliminate endotracheal intubation is taken into consideration. Tank respirators were used decades ago to apply continuous external negative pressure in three patients with severe pneumonia who were not intubated [8-10]. This improved oxygenation and enabled maintenance of spontaneous breathing in severe lung injury.
Our results demonstrate for the first time that CENPV is applicable and effective, even in severely critically ill patients in a modern intensive care setting. The present study confirms recent experimental data and encourages consideration of further studies of the physiological effects and clinical effectiveness of CENPV in patients with ARDS.
• CENPV differs substantially from CPPV and improves oxygenation under more physiologic conditions in patients with ARDS.
ARDS: acute respiratory distress syndrome; CENPV: continuous external negative-pressure ventilation; CPPV: continuous positive-pressure ventilation; FiO2: fraction of inspired oxygen; FRC: functional residual capacity; ECG: electrocardiogram; IPPV: intermittent positive-pressure ventilation; paO2: arterial oxygen tension; paCO2: carbon dioxide tension; PEEP: positive end-expiratory pressure; PiCCO: Pulse Contour Cardiac Output; SAPS: Simplified Acute Physiology Score; TRP: trans-respiratory system pressure.
The authors declare that they have no competing interests.
KR, UM, MC, WK and AC participated in the design of the study; KR, UM and MC participated in the collection and the assembly of the data; KR and TD performed the statistical analysis; BS, JA, CW and AC gave technical or logistical support; KR, UM, TD, JA, CW and AC helped to draft the manuscript and, all authors read and approved the final manuscript.
Support for the publication fee was granted by the German Research Foundation (DFG). The authors thank Dr Horst Rückoldt, Prof Dr Jörn Heine, Dr Ljiljana Verner and Prof Dr Siegfried Piepenbrock for their administrative support.
|Villar J,Blanco J,Añón JM,Santos-Bouza A,Blanch L,Ambrós A,Gandía F,Carriedo D,Mosteiro F,Basaldúa S,Fernández RL,Kacmarek RM,ALIEN NetworkThe ALIEN study: incidence and outcome of acute respiratory distress syndrome in the era of lung protective ventilationIntensive Care MedYear: 2011161932194110.1007/s00134-011-2380-421997128|
|Ranieri VM,Giunta F,Suter PM,Slutsky AS,Mechanical ventilation as a mediator of multisystem organ failure in acute respiratory distress syndromeJAMAYear: 200016434410.1001/jama.284.1.4310872010|
|Amato MB,Barbas CS,Medeiros DM,Magaldi RB,Schettino GP,Lorenzi-Filho G,Kairalla RA,Deheinzelin D,Munoz C,Oliveira R,Takagaki TY,Carvalho CR,Effect of a protective-ventilation strategy on mortality in the acute respiratory distress syndromeN Engl J MedYear: 19981634735410.1056/NEJM1998020533806029449727|
|The Acute Respiratory Distress Syndrome NetworkVentilation with lower tidal volumes as compared with traditional tidal volumes for acute lung injury and the acute respiratory distress syndromeN Engl J MedYear: 2000161301130810793162|
|Brower RG,Lanken PN,MacIntyre N,Matthay MA,Morris A,Ancukiewicz M,Schoenfeld D,Thompson BT,National Heart, Lung, and Blood Institute ARDS Clinical Trials NetworkHigher versus lower positive end-expiratory pressures in patients with the acute respiratory distress syndromeN Engl J MedYear: 20041632733615269312|
|Corrado A,Gorini M,Melej R,Baglioni S,Mollica C,Villella G,Consigli GF,Dottorini M,Bigioni D,Toschi M,Eslami A,Iron lung versus mask ventilation in acute exacerbation of COPD: a randomised crossover studyIntensive Care MedYear: 20091664865510.1007/s00134-008-1352-919020859|
|Grasso F,Engelberts D,Helm E,Frndova H,Jarvis S,Talakoub O,McKerlie C,Babyn P,Post M,Kavanagh BP,Negative-pressure ventilation: better oxygenation and less lung injuryAm J Respir Crit Care MedYear: 20081641241818079496|
|Shyamal K,Sanyal SK,Bernal R,Hughes WT,Feldman S,Continuous negative chest-wall pressure: successful use for severe respiratory distress in an adultJAMAYear: 1976161727172810.1001/jama.1976.032701600490311067453|
|Sanyal SK,Turner S,Ossi M,Coburn TP,Continuous negative chest wall pressure therapy in management of severe hypoxemia due to aspiration pneumonitis: a case reportRespir CareYear: 19791610221025|
|Morris AH,Elliott CG,Adult respiratory distress syndrome: successful support with continuous negative extrathoracic pressureCrit Care MedYear: 19851698999010.1097/00003246-198511000-000434053650|
|Scholz SE,Knothe C,Thiel A,Hempelmann G,Improved oxygen delivery by positive pressure ventilation with continuous negative external chest pressureLancetYear: 1997161295129610.1016/S0140-6736(05)62507-X9142069|
|Borelli M,Benini A,Denkewitz T,Acciaro C,Foti G,Pesenti A,Effects of continuous negative extrathoracic pressure versus positive end-expiratory pressure in acute lung injury patientsCrit Care MedYear: 1998161025103110.1097/00003246-199806000-000219635650|
|Torelli L,Zoccali G,Casarin M,Dalla Zuanna F,Lieta E,Conti G,Comparative evaluation of the haemodynamic effects of continuous negative external pressure (CNEP) and positive end-expiratory pressure (PEEP) in mechanically ventilated trauma patientsIntensive Care MedYear: 199516677010.1007/BF024251577560477|
|Bernard GR,Artigas A,Brigham KL,Carlet J,Falke K,Hudson L,Lamy M,Legall JR,Morris A,Spragg R,The American-European Consensus Conference on ARDS: definitions, mechanisms, relevant outcomes, and clinical trial coordinationAm J Respir Crit Care MedYear: 1994168188247509706|
|Le Gall JR,Lemeshow S,Saulnier F,A new Simplified Acute Physiology Score (SAPS II) based on a European/North American multicenter studyJAMAYear: 1993162957296310.1001/jama.1993.035102400690358254858|
|Talmor D,Sarge T,Malhotra A,O'Donnell CR,Ritz R,Lisbon A,Novack V,Loring SH,Mechanical ventilation guided by esophageal pressure in acute lung injuryN Engl J MedYear: 2008162095210410.1056/NEJMoa070863819001507|
|Pichlmayr I,Lehmkuhl P,Pichlmayr R,[Postoperative ventilation using the iron lung principle. A renaissance of this form of artificial respiration?] [in German]Anasth Intensivther NotfallmedYear: 198416141810.1055/s-2007-10034056585150|
|Knitsch W,Pohl S,Schultz A,Fabel H,Pichlmayr I,[Atelectasis treatment by ventilatory support using an iron lung] [in German]Dtsch Med WochenschrYear: 1989161441144410.1055/s-2008-10667802507268|
|De Keulenaer BL,De Waele JJ,Powell B,Malbrain ML,What is normal intra-abdominal pressure and how is it affected by positioning, body mass and positive end-expiratory pressure?Intensive Care MedYear: 20091696997610.1007/s00134-009-1445-019242675|
|Shoptaugh M,Cvetnic WG,Hallman M,Waffarn F,Stevens J,Blansfield T,Barsotti M,Pulmonary mechanics generated by positive end-expiratory and continuous negative pressureJ PerinatolYear: 1993163413488263617|
|Chiumello D,Carlesso E,Cadringher P,Caironi P,Valenza F,Polli F,Tallarini F,Cozzi P,Cressoni M,Colombo A,Marini JJ,Gattinoni L,Lung stress and strain during mechanical ventilation for acute respiratory distress syndromeAm J Respir Crit Care MedYear: 20081634635510.1164/rccm.200710-1589OC18451319|
|Caironi P,Cressoni M,Chiumello D,Ranieri M,Quintel M,Russo SG,Cornejo R,Bugedo G,Carlesso E,Russo R,Caspani L,Gattinoni L,Lung opening and closing during ventilation of acute respiratory distress syndromeAm J Respir Crit Care MedYear: 20101657858610.1164/rccm.200905-0787OC19910610|
|Quintel M,Pelosi P,Caironi P,Meinhardt JP,Luecke T,Herrmann P,Taccone P,Rylander C,Valenza F,Carlesso E,Gattinoni L,An increase of abdominal pressure increases pulmonary edema in oleic acid-induced lung injuryAm J Respir Crit Care MedYear: 20041653454114670801|
|Skaburskis M,Michel RP,Gatensby A,Zidulka A,Effect of negative-pressure ventilation on lung water in permeability pulmonary edemaJ Appl PhysiolYear: 1989162223223010.1063/1.3442752501278|
|Kudoh I,Andoh T,Doi H,Kaneko K,Okutsu Y,Okumura F,Continuous negative extrathoracic pressure ventilation, lung water volume, and central blood volume: studies in dogs with pulmonary edema induced by oleic acidChestYear: 19921653053310.1378/chest.101.2.5301735284|
|Van der Zee H,Cooper JA,Hakim TS,Malik AB,Alternations in pulmonary fluid balance induced by positive end-expiratory pressureRespir PhysiolYear: 19861612513310.1016/0034-5687(86)90036-83518003|
|Protti A,Chiumello D,Cressoni M,Carlesso E,Mietto C,Berto V,Lazzerini M,Quintel M,Gattinoni L,Relationship between gas exchange response to prone position and lung recruitability during acute respiratory failureIntensive Care MedYear: 2009161011101710.1007/s00134-009-1411-x19189081|
Clinical characteristics and respiratory variables of the patients within 12 hours before study entrya
|Cause of lung injury||Demographics||Respiratory variables|
|Patient||Disorders predisposing to ARDS||Underlying disease||SAPS II||Body mass index||PaO2/FiO2 ratio||Plateau pressure (cmH2O)||PEEP (cmH2O)||FiO2||PaCO2 (mmHg)||pH||Days on ventilator||Outcomes|
|2||Severe pulmonary thromboembolism and thrombectomy||Parkinson's disease||33||25.7||190||32||10||0.4||46||7.40||3||Survived|
|3||Sepsis, liver failure after valproate administration||Endometritis, epilepsy||35||31.1||153||30||14||0.5||45||7.43||5||Survived|
|5||Aspiration||Gastric ulcer perforation||34||29.9||153||36||13||0.6||49||7.32||2||Survived|
|6||Sepsis, aspiration||Colon diverticulitis||42||30.4||118||46||22||0.8||65||7.30||47||Deceased|
|Mean ± SD||35 ± 3||29 ± 2||159 ± 27||33 ± 7||13 ± 5||0.5 ± 02||48 ± 9||7.39 ± 0.07||18 ± 11|
aPatient 2 developed ARDS after embolectomy and three days of mechanical ventilation at pulmonary artery wedge pressures below 18 mmHg. Patient 6 developed ARDS already at admission due to a combination of sepsis after bowel suture insufficiency and aspiration and was treated with lung-protective ventilation already for 47 days at study entry. ARDS, acute respiratory distress syndrome; FiO2, fraction of inspired oxygen pressure; PaCO2, arterial carbon dioxide pressure; PaO2/FiO2, arterial oxygen-to-fraction of inspired oxygen pressure ratio; PEEP, positive end-expiratory pressure; SAPS II, Simplified Acute Physiology Score II .
Lung volume and intraabdominal pressure during continuous positive-pressure ventilation and continuous external negative-pressure ventilationa
|Tidal volume (ml)||Minute volume
at end expiration (ml)
|Patient||CPPV||CENPV||CPPV||CENPV||Respiratory rate (breaths/minute)||CPPV||CENPV||CPPV||CENPV|
|Mean ± SD||506 ± 70||509 ± 72||10.1 ± 2.6||10.2 ± 2.6||19.8 ± 3.1||682 ± 137||677 ± 138||21 ± 8||0 ± 6*|
aCENPV, continuous external negative-pressure ventilation; CPPV, continuous positive-pressure ventilation. Patient 6 had an open abdomen. *P = 0.03 compared to CPPV.
Ventilatory pressures during continuous positive-pressure ventilation and continuous external negative-pressure ventilation at inspiration and expirationa
|Airway pressure (cmH2O)||Tank pressure (cmH2O)||Transrespiratory
|Transpulmonary pressure (cmH2O)|
|Inspiration||33||-3 (27)||-30||27||20||-9 (21)||13||6|
|Expiration||16||5 (20)||-15||20||15||4 (19)||1||1|
|Inspiration||36||-2 (31)||-33||31||22||-9 (24)||14||7|
|Expiration||16||5 (20)||-15||20||15||5 (20)||1||0|
|Inspiration||30||0 (31)||-31||31||19||-8 (23)||11||8|
|Expiration||16||2 (17)||-15||17||15||2 (17)||1||0|
|Inspiration||33||-1 (31)||-32||31||25||-9 (23)||8||8|
|Expiration||16||4 (19)||-15||19||16||4 (19)||0||0|
|Inspiration||39||-2 (38)||-40||38||23||-10 (30)||16||8|
|Expiration||16||4 (19)||-15||19||16||3 (18)||0||1|
|Inspiration||47||0 (43)||-43||43||32||-10 (33)||15||10|
|Expiration||23||5 (24)||-19||24||22||5 (24)||1||0|
|Mean ± SD|
|Inspiration||36 ± 6||-1.3 ± 1.2*
(33.5 ± 5.9)
|-35 ± 5||34 ± 6||24 ± 5||-9 ± 0.8*
(25.7 ± 4.7)
|13 ± 3||8 ± 1.3†|
|Expiration||17 ± 3||4.2 ± 1.2*
(19.5 ± 2.3)
|-16 ± 1.6||20 ± 2*||17 ± 3||4 ± 1.2*
(19.5 ± 2.4)
|1 ± 0.5||0 ± 0.5|
aIn parentheses, both airway and oesophageal pressures are shown in reference to tank pressure to enable calculation of transpulmonary pressures (airway pressure - oesophageal pressure) in reference to both atmospheric and body surface pressure inside the tank. Transrespiratory system pressures during CENPV (airway pressure - tank pressure) were compared to airway pressures during CPPV. CENPV, continuous external negative-pressure ventilation; CPPV, continuous positive-pressure ventilation. *P = 0.03 and †P = 0.04 compared to the corresponding inspiratory or expiratory value during CPPV.
Keywords: iron lung, tank respirator, external negative-pressure ventilation, acute lung injury.
Previous Document: A systematic review of cross-cultural comparison studies of child, parent, and health professional o...
Next Document: Reversible cerebral vasoconstriction syndrome associated with brain parenchymal hemorrhage.