|Constrained liner in neurologic or cognitively impaired patients undergoing primary THA.|
|PMID: 20376709 Owner: NLM Status: MEDLINE|
|BACKGROUND: THA performed in patients with cognitive deficits or neuromuscular diseases has been associated with a high postoperative dislocation rate. The constrained liner reportedly provides stability in patients with recurrent dislocation. However, achieving stability could be offset by early loosening when used in patients with neurologic diseases.
QUESTIONS/PURPOSES: We therefore asked whether constrained liners had a higher risk of loosening when used in primary THA for patients with neurologic diseases.
METHODS: We retrospectively reviewed a 144 patients (164 hips) with neuromuscular disease who had a constrained polyethylene insert from 1999 to 2004 and compared them to another 120 patients (132 hips) with neuromuscular disease operated on immediately before this period (from 1994 to 1998) who had a conventional polyethylene insert.
RESULTS: Thirty-three (25%) of the 132 hips without a constrained liner were known to have had at least one dislocation and 21 had revision for recurrent dislocation. Ten other hips had revision for loosening of the cup. The survival rate was 82% at 5 years and 77% at 10 years with revision due to recurrent dislocation or loosening of the cup as the endpoint. With a constrained liner, at minimum 5-year followup (mean, 7 years; range, 5-10 years), the incidence of complications, particularly dislocation, was decreased (three dislocations among 164 hips; 2%), with one revision for recurrent dislocation and one revision for loosening.
CONCLUSIONS: This constrained acetabular component provides durable protection against dislocation without substantial increased loosening at midterm followup.
|Philippe Hernigou; Paolo Filippini; Charles-Henri Flouzat-Lachaniette; Sobrinho Uirassu Batista; Alexandre Poignard|
Related Documents :
|10872359 - Charnley low-frictional torque arthroplasty of the hip. 20-to-30 year results.
2783019 - Comparative study of cemented and cementless hip prostheses in the same patient.
16645109 - Percutaneous vertebroplasty for vertebral compression fractures with and without intrav...
19297199 - Survival of the glenoid component in shoulder arthroplasty.
19195829 - Successful performance of the bi-metric uncemented femoral stem at a minimum follow-up ...
20720139 - Influence of obesity on femoral osteolysis five and ten years following total hip arthr...
22151589 - Stn vs. pallidal stimulation in parkinson disease: improvement with experience and bett...
2581099 - A prospective randomized study of hydroxyethyl starch, albumin, and lactated ringer's s...
24403769 - Evaluation of the degree of knee joint osteoarthritis in patients with early gray hair.
|Type: Comparative Study; Journal Article|
|Title: Clinical orthopaedics and related research Volume: 468 ISSN: 1528-1132 ISO Abbreviation: Clin. Orthop. Relat. Res. Publication Date: 2010 Dec|
|Created Date: 2010-11-08 Completed Date: 2010-12-06 Revised Date: 2013-05-29|
Medline Journal Info:
|Nlm Unique ID: 0075674 Medline TA: Clin Orthop Relat Res Country: United States|
|Languages: eng Pagination: 3255-62 Citation Subset: AIM; IM|
|Department of Orthopaedic Surgery, University Paris XII, Hôpital Henri Mondor, 94010, Creteil, France. firstname.lastname@example.org|
|APA/MLA Format Download EndNote Download BibTex|
Aged, 80 and over
Arthroplasty, Replacement, Hip / adverse effects, instrumentation*
Cognition Disorders / complications*
Dislocations / epidemiology, etiology, surgery
Hip Joint / physiopathology, radiography, surgery*
Joint Instability / epidemiology, etiology, surgery
Neuromuscular Diseases / complications*
Proportional Hazards Models
Range of Motion, Articular
From MEDLINE®/PubMed®, a database of the U.S. National Library of Medicine
Previous Document: Fluid Balance, Complications, and Brain Tissue Oxygen Tension Monitoring Following Severe Traumatic ...
Next Document: Conjunctival squamous cell carcinoma due to ocular prostheses: a case report and review of literatur...