|Complications and mortality of single versus dual chamber implantable cardioverter defibrillators.|
|Jump to Full Text|
|PMID: 16943899 Owner: NLM Status: PubMed-not-MEDLINE|
|BACKGROUND: The implantable cardioverter defibrillators (ICDs) are increasingly being used as a treatment modality for life threatening tachyarrhythmia. The purpose of this study was to compare the frequency of complications and mortality between single-chamber and dual-chamber ICD implantation in Shahid Rajaie cardiovascular center. METHODS AND RESULTS: Between January 2000 and December 2004, 234 patients received ICD by a percutaneous transvenous approach and were followed for 33 +/- 23 months. The cumulative incidence of complications was 9.4% over the follow-up period. There was no significant difference in overall complication rate between single chamber (VR) and dual chamber (DR) ICD groups in the follow-up period (P= 0.11). The risk of complications did not have any statistically significant difference in secondary versus primary prevention groups (P=0.06). The complications were not associated with the severity of left ventricular systolic dysfunction (P=0.16). The frequency of lead-related complications was higher in dual chamber ICDs in comparison with single chamber ICDs (P=0.02). There was no significant difference in mortality between different sex groups (P=0.37), different indications for ICD implantation (P=0.43) or between VR and DR ICD groups (P= 0.55). Predictors of mortality were NYHA class III or more (P<0.001), age >65 years (P=0.011) and LVEF<30% (P<0.001). The mortality in patients with CAD and DCM were significantly higher than those with other structural heart diseases (P=0.001). CONCLUSION: Close monitoring of patients during the first 2 month after ICD implantation is recommended because the majority of complications occur early after the procedure.|
|Ataallah Bagherzadeh; Zahra Emkanjoo; Majid Haghjoo; Maryam Moshkani Farahani; Abolfath Alizadeh; Mohammad Ali Sadr-Ameli|
Related Documents :
|14759419 - A new procedure for ebstein's anomaly.
7300409 - Detachment of the septal tricuspid leaflet during transatrial closure of isolated ventr...
9778329 - Potential benefit from implantable cardioverter-defibrillator therapy in patients with ...
11722059 - Correction of simple congenital heart defects in infants and children through a minitho...
17720399 - Autologous right atrial wall patch for closure of atrial septal defects.
14729789 - Is mobitz type i atrioventricular block benign in adults?
20430849 - Emergent self-expanding stent placement for acute intracranial or extracranial internal...
22417569 - Heated and humidified high-flow oxygen therapy reduces discomfort during hypoxemic resp...
24819199 - Is gait speed improving performance of the euroscore ii for prediction of early mortali...
|Type: Journal Article Date: 2006-04-01|
|Title: Indian pacing and electrophysiology journal Volume: 6 ISSN: 0972-6292 ISO Abbreviation: Indian Pacing Electrophysiol J Publication Date: 2006|
|Created Date: 2006-08-31 Completed Date: 2007-07-11 Revised Date: 2009-11-18|
Medline Journal Info:
|Nlm Unique ID: 101157207 Medline TA: Indian Pacing Electrophysiol J Country: India|
|Languages: eng Pagination: 75-83 Citation Subset: -|
|Department of Pacemaker and Electrophysiology, Rajaie Cardiovascular Medical and Research Center, Iran University of Medical Sciences, Mellat Park, Vali-e-Asr Avenue, Tehran, Iran. firstname.lastname@example.org|
|APA/MLA Format Download EndNote Download BibTex|
Journal ID (nlm-ta): Indian Pacing Electrophysiol J
Journal ID (publisher-id): Indian Pacing Electrophysiol J
Publisher: Indian Pacing and Electrophysiology Group
Copyright: ? 2006 Bagherzadeh et al.
open-access: This is an open-access article distributed under the terms of the Creative Commons Attribution License, which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.
collection publication date: Season: Apr?Jun Year: 2006
Electronic publication date: Day: 1 Month: 4 Year: 2006
Volume: 6 Issue: 2
First Page: 75 Last Page: 83
Publisher Id: ipej060075-00
PubMed Id: 16943899
|Complications and Mortality of Single Versus Dual Chamber Implantable Cardioverter Defibrillators|
|Ataallah Bagherzadeh, MD|
|Zahra Emkanjoo, MD|
|Majid Haghjoo, MD|
|Maryam Moshkani Farahani, MD|
|Abolfath Alizadeh, MD|
|Mohammad Ali Sadr-Ameli, MD|
| Department of Pacemaker and Electrophysiology, Rajaie Cardiovascular Medical and Research Center, Iran University of Medical Sciences, Mellat Park, Vali-e-Asr Avenue, Tehran, Iran
|Correspondence: Address for correspondence: Ataallah Bagherzadeh, M.D., Department of Pacemaker and Electrophysiology, Rajaie Cardiovascular Medical and Research Center, Iran University of Medical Sciences, Mellat Park, Vali-e-Asr Avenue, Tehran 1996911151 Iran, P.O.Box: 15745-1341. E-mail: email@example.com
The implantable cardioverter defibrillators (ICDs) are increasingly being used as a treatment modality for high risk survivors of life-threatening tachyarrhythmia [1,2]. Major technological advances like the introduction of transvenous defibrillation leads and "active can" concept have simplified the implantation of ICDs and have reduced implantation-related complications but still there are considerable rate of complications in most of the electrophysiologic centers [3,4].
The purpose of the current study was to identify and characterize prospectively the frequency of lead and implant-related complications and also mortality in patients with single-chamber versus dual-chamber ICD implantation.
Between January, 2000 and December, 2004, 234 patients received ICDs in our center. All of them included in the study and were followed for 33 ?23 months. The primary end point of the study was actuarial survival. The secondary end point was nonfatal complications, including inappropriate ICD therapy, lead-related and implant-related complications. The study was approved by local Ethics Committee and written informed consent was obtained from all the patients.
ICD system manufacturers included Medtronic and St Jude systems in 160 and 75 patients respectively. Atrial leads included 1688 T or 1642T for St Jude and 5076 for Medtronic systems. Ventricular leads included 1570,1572,1571,1580 and 1581 for St Jude and 6944, 6947 and 6948 for Medtronic systems. All of the ventricular leads were bipolar. In implanted devices all the detection and discrimination criteria were activated with the nominal values. In all the devices we defined ventricular fibrillation zone (300ms) plus one VT zone (400 ms). If the patient had an episode of spontaneous or induced sustained monomorphic VT slower than 370 ms we extended the VT zone to VT cycle length plus 40ms. In the VT detection zone the first therapy was three antitachycardia burst pacing. We used the nominal values of the ICDs for the duration and tachyarrhythmia detection criteria. Antibradycardia pacing was programmed in VVI mode with pacing rate of 40 beats per minute in all VR and DR ICDs, unless the patients needed dual chamber pacing, those with long QT syndrome and hypertrophic obstructive cardiomyopathy.
The implantation routes were either subclavian or Axillary veins percutaneously in electrophysiology laboratory by an electrophysiologist.
The patients were followed after 1 month postoperatively and every 3 months thereafter and upon receiving high voltage therapy in our outpatient ICD clinic. ICD evaluations during follow-up visits included routine clinically appropriate measurements including interrogation of the device for tachyarrhythmia episodes, evaluation of sensing and pacing thresholds and lead impedance. Floppy diskettes were used to retrieve all episodes of ICD therapy. All the episodes resulted in ICD therapy, studied independently by two electrophysiologists to define the diagnosis. All the episodes were categorized as appropriate or inappropriate. ICD-related complications were documented, as were mortality and cause of death. Chest radiography (CXR) was obtained before hospital discharge and every 6 months to confirm radiographic lead integrity. Clinical evaluation of patients included history and physical examination of the implantation site. Complications were defined as unexpected adverse events that were felt to be related to the implant procedure or to the ICD lead and generator system.
Baseline characteristics and complications were summarized as the mean ? SD for continuous variables and otherwise as percentage. SPSS 13 software (SPSS Inc., Chicago, IL, USA) was used for data storage and analysis. Qualitative characteristics were compared using the Pearson?s chi square and Fischer exact tests. Kaplan-Meyer survival curves were calculated. Log rank statistics were used to compare the distribution of time to complications and mortality between different ICD groups. Two-tailed P < 0.05 were considered significant. Regression analysis was performed to evaluate the effect of variables on mortality and complications.
Three patients (1.2%) died during the first month after ICD implantation because of multi-organ failure, massive pulmonary emboli and acute coronary syndrome. The 1-year survival rate was 94.9% and 94.1% for VR and DR ICDs respectively. The overall survival rate during follow-up was 90.6% and 88.9% for VR and DR ICDs respectively. There was no significant difference in mortality between different sex groups (P=0.37), different indications for ICD implantation (P=0.43) or between VR and DR ICD groups (P= 0.55) (Figure 1). Univariate regression analysis showed predictors of mortality as NYHA class III or more (P<0.001), age >65 years (P=0.011) and LVEF<30% (P<0.001). The mortality in patients with CAD and DCM were significantly higher than those with other structural heart diseases (P=0.001). During the follow-up, 59 patients (25.2%) had at least one inappropriate ICD therapy. The mortality of this group was significantly higher than those who had not experienced inappropriate therapy (P=0.03).
Overall, 22 patients (9.4%) suffered from ICD-related complications during follow-up period. There was no significant difference in overall complication rate between VR and DR ICD groups in the follow-up period (P= 0.11) (Figure 2). The risk of complications was not associated with any patient-specific factor including age (P=0.32) and gender (P=0.19) (Table 1). The risk of complications did not have any statistically significant difference in secondary versus primary prevention groups (P=0.06). The complications were not associated with the severity of left ventricular systolic dysfunction (P=0.16). Also, no statistically significant difference was observed between different NYHA classes (P=0.45) and underlying heart diseases (P=0.06). During the follow-up, 59 patients (25.2%) had at least one inappropriate ICD therapy. The risk of complications did not have any statistically significant difference in this group with those who had not experienced inappropriate therapy (P=0.16). The comparison of complications in different models of implanted leads was not possible due to variety of the models and small numbers of each model.
Implant-related complications occurred in 13 patients (5.5%). Hematoma or bleeding was the most frequent complication in this group, found in 7 patients (3%). Pneumothorax and hemothorax were found in 3(1.3%) and 2(0.9%) patients, all of them managed with chest tube insertion. No significant pocket infection was found but only one patient suffered from stitch abscess, treated conservatively with antibiotics.
Lead-related complications occurred in 9 patients (3.8%). Lead dislodgement was the most frequent lead-related complication, found in 5 patients (2.1%). Dislodgement of ventricular and atrial leads was seen in 3 and 2 patients respectively. Lead fracture was found in 2 patients (0.9%). Exit block and insulation break was rare, each of them occurred only in 1 patient (0.4%). 4 of 5 lead dislodgements and all of hematomas occurred in the 2 month of implantation. No cardiac perforation was found. The frequency of lead-related complications was significantly higher in dual chamber ICDs in comparison with single chamber ICDs (P=0.02).
This prospective study characterized the incidence of lead and implant-related complications and mortality of ICD implantations in our center. The baseline demographic data of the patients in this study were comparable to large studies (Table 1) [3-8]. With the downsizing of ICD pulse generators, pectoral implantation has become standard for ICD placement . Older generation pacemakers of comparable size to ICD pulse generators had a significant risk of erosion and other pocket complications . During this study, 234 consecutive ICD implantations over a 33 ?23 months follow-up showed low (9.4%) incidence of complications. The most common implant and lead related complications were hematoma and lead dislodgement which was found in 3% and 2.1% of patients. These findings are consistent with or even rarer than prior report [5-18]. In the majority of previous studies the ICD-related complications did not differ with increasing age which was consistent with this study [15-18]. The infection rate in this case series was very low (0.4%). In large studies, infection rates of 1.3-2.7% were reported. Because most of the infections that were seen in previous reports occurred with abdominal pocket, it seems reasonable to expect a lower infection rate with current generation of ICDs implanted transvenously in the pectoral region [7,20-22]. Intensive use of prophylactic antibiotics before and 48 hours after the procedure and avoidance of cauterization for hemostasis may help us reduce the infection rate.
Some authors encourage the usage of cephalic vein cut downs as the preferred approach for ICD lead venous approach for prevention of vascular complications of blind, percutaneous subclavian venipuncture such as pneumothorax and hemothorax [23,24]. The time course of specific complications reinforces the need for continued lead surveillance in ICD patients. Although bleeding, pneumothorax, lead dislodgement and infection tends to occur soon after implant, lead-related problems such as lead fracture and insulation breaks tend to occur throughout follow-up [7-19,25,26].
The present study must be interpreted in light of certain methodological limitations. The ICDs were implanted with multiple operators of variable training and experience. Therefore, the effects of differences in experience or surgical technique on complication rates were not controlled. However, the inclusion of the large number of cases and relatively long follow-up periods makes it more likely that these results are reliable. The second limitation was lack of cooperation of some patients for regular follow-up and routine CXRs. Therefore, the incidence of lead fractures and insulation breaks may have been underestimated. Also, small hematomas usually managed by sand bag and under- reported.
This study showed that the pectoral implantation of ICDs by electrophysiologists in our center has a low rate of complications, comparable with most of the qualified centers. Since most of implant-related and some of lead-related complications occur soon after the implant, close early monitoring of patients during first 2 month after implantation is recommended. The continued occurence of lead fractures and the need for premature system revision over time supports the practice of close, routine ICD surveillance. Vascular complications of blind percutaneous subclavian venipuncture could be avoided if cephalic vein cut down is used for ICD lead venous access.
|Zipes DP,Roberts DP. The Pacemaker Cardioverter Defibrillator Investigators. Results of the international study of the implantable pacemaker cardioverter defibrillator: a comparison of epicardial and endocardial lead systemsCirculation 1995;92:59–65. [pmid: 7788918]|
|Stambler BS,Wood MA,Damiano RJ. Sensing/pacing lead complications with a newer generation implantable cardioverter defibrillator: World wide experience the Guardian ATP 4210 clinical trialJ Am Coll Cardiol 1994;23:123–132. [pmid: 8277070]|
|Bardy GH,Johnson G,Poole JE,et al. A simplified, single lead unipolar transvenous cardioversion-defibrillation systemCirculation 1993;88:543–547. [pmid: 8339416]|
|Bardy GH,Yee R,Jung W. for the active can investigators. Multicenter experience with a pectoral unipolar implantable cardioverter defibrillatorJ Am Coll Cardiol 1996;28:400–410. [pmid: 8800117]|
|The Antiarrhythmic Versus Implantable Defibrillator (AVID) investigators. A comparison of anti arrhythmic drug therapy with implantable defibrillators in patients resuscitated from near-fatal ventricular arrhythmiasN Eng J Med 1997;337:1576–1583.|
|Gradatus R,Block M,Brachmann C,et al. Mortality, Morbidity and complications in 3334 patients with implantable cardioverter-defibrillatorsPACE 2003;26:1511–1518. [pmid: 12914630]|
|Kron J,Herre J,Renfroe EG,Takahashi A,et al. Lead and device-related complications in the Antiarrhythmics Versus Implantable Defibrillators TrialAm Heart J 2001;141:92–98. [pmid: 11136492]|
|Connolly SJ,Gent M,Roberts RS,et al. for the CIDS investigators. Canadian Implantable Defibrillator Study (CIDS): A randomized trial of implantable cardioverter defibrillator against AmiodaroneCirculation 2000;101:1297–1302. [pmid: 10725290]|
|Gold MR,Peters RW,Johnson JW,et al. Complications associated with pectoral implantation of Cardioverter DefibrillatorsPACE 1997;20:208–211. [pmid: 9121991]|
|Witte J,Bondke HJ. The incidence of pacemaker pocket complications depending on pulse generator size and shape- conclusion for defibrillator therapyPACE 1995;18:1793.|
|Grimm W,Menz N,Hoffmann J. Complications of third generation implantable cardioverter defibrillator therapyPACE 1999;22:206–211. [pmid: 9990632]|
|Rosenqvist M,Beyer T,Block M. European 7219 Jewel ICD investigators. Adverse events with transvenous implantable cardioverter- defibrillators: a prospective multicenter studyCirculation 1998;98:663–670. [pmid: 9715859]|
|Nunain SO,Roelke M,Trouton T,et al. Limitations and late complications of third generation automatic cardioverter- defibrillatorsCirculation 1995;19:2204–2213. [pmid: 7697850]|
|Pacifico A,Wheelan KR,Nasir N,et al. Long term follow-up of cardioverter defibrillator implanted under conscious sedation in prepectoral subfacial positionCirculation 1997;95:946–950. [pmid: 9054755]|
|Brooks R,Garan H,Torchiana D,et al. Three-year outcome of a non-thoracotomy approach to cardioverter defibrillator implantation in 189 consecutive patientsAm J Cardiol 1994;74:1011–1015. [pmid: 7977038]|
|Strickberger SA,Hummel JD,Daoud E,et al. Implantation by electrophysiologists of 100 consecutive cardioverter Defibrillators with nonthoracotomy lead systemsCirculation 1994;90:868–872. [pmid: 8044958]|
|Schwartzman D,Nallamothu N,Callans DJ,et al. Postoperative lead-related complications in patients with nonthoracotomy defibrillation systemsAm J Cardiol 1995;26:776–786.|
|Kuch KH,Cappata R,Siebels J. Randomized comparison of antiarrhythmic drug therapy with implantable defibrillators in patients resuscitated from cardiac arrest: The Cardiac Arrest Study Hamburg (CASH)Circulation 2000;102:748–754. [pmid: 10942742]|
|Noseworthy PA,Lashevsky I,Dorian P,et al. Feasibility of implantatable Cardioverter Defibrillator use in elderly patientsPACE 2004;24:373–378. [pmid: 15009867]|
|Gupta S,Prevel CD,Shaheen K,et al. Wound complications and treatment of the infected implantable Cardioverter Defibrillator generatorJ Card Surg 1993;8:671–677. [pmid: 8286873]|
|Wunderly RG,Maloney J,Edel T,et al. Infections in implantable Cardioverter Defibrillator patientsPACE 1990;13:1360–1364. [pmid: 1701886]|
|Hauser RG,Kurschinski DT,Mcveigh K,et al. Clinical results with nonthoracotomy ICD systemsPACE 1993;16:141–148. [pmid: 7681561]|
|Byrd CL. Clinical experience with the extrathoracic introducer insertion techniquePACE 1993;16:1781–1784. [pmid: 7692408]|
|Parsonnet V,Roelke M. The cephalic vein cut down versus subclavian puncture for pacemaker/ICD lead implantationPACE 1999;22:695–697. [pmid: 10353126]|
|Kettering K,Mewis C,Dornberger V,et al. Long term experience with subclavian ICD leads: A comparison among three different types of subclavian leadsPACE 2004;27:1355–1361. [pmid: 15511244]|
|Glikson M,Lipchenca I,Viskin S,et al. Long term outcome of patients who received implantable cardioverter-defibrillators for stable ventricular tachycardiaJ Cardiovasc Electrophysiol 2004;15:658–664. [pmid: 15175060]|
Keywords: implantable cardioverter-defibrillator, complication, mortality.
Previous Document: Pacemaker prevention therapy in drug-refractory paroxysmal atrial fibrillation: reliability of diagn...
Next Document: The Postural Tachycardia Syndrome (POTS): pathophysiology, diagnosis & management.