|Clinical review: Update on neurally adjusted ventilatory assist - report of a round-table conference.|
|Jump to Full Text|
|PMID: 22715815 Owner: NLM Status: Publisher|
|ABSTRACT: Conventional mechanical ventilators rely on pneumatic pressure and flow sensors and controllers to detect breaths. New modes of mechanical ventilation have been developed to better match the assistance delivered by the ventilator to the patient's needs. Among these modes, neurally adjusted ventilatory assist (NAVA) delivers a pressure that is directly proportional to the integral of the electrical activity of the diaphragm recorded continuously through an esophageal probe. In clinical settings, NAVA has been chiefly compared with pressure-support ventilation, one of the most popular modes used during the weaning phase, which delivers a constant pressure from breath to breath. Comparisons with proportional-assist ventilation, which has numerous similarities, are lacking. Because of the constant level of assistance, pressure-support ventilation reduces the natural variability of the breathing pattern and can be associated with asynchrony and/or overinflation. The ability of NAVA to circumvent these limitations has been addressed in clinical studies and is discussed in this report. Although the underlying concept is fascinating, several important questions regarding the clinical applications of NAVA remain unanswered. Among these questions, determining the optimal NAVA settings according to the patient's ventilatory needs and/or acceptable level of work of breathing is a key issue. In this report, based on an investigator-initiated round table, we review the most recent literature on this topic and discuss the theoretical advantages and disadvantages of NAVA compared with other modes, as well as the risks and limitations of NAVA.|
|Nicolas Terzi; Lise Piquilloud; Hadrien Rozé; Alain Mercat; Frédéric Lofaso; Stéphane Delisle; Philippe Jolliet; Thierry Sottiaux; Didier Tassaux; Jean Roesler; Alexandre Demoule; Samir Jaber; Jordi Mancebo; Laurent Brochard; Jean-Christophe Marie Richard|
Related Documents :
|15236085 - Effects of sevoflurane on cardiovascular dynamics, coronary circulation and myocardial ...
7692155 - Effects of d-nebivolol and l-nebivolol on left ventricular systolic and diastolic funct...
10460955 - Regulation of myocardial beta-adrenergic receptor function in adult and neonatal rabbits.
9557805 - Effects of dobutamine on isovolumic and ejection phase indices of cardiac contractility...
2645495 - Rhinomanometry 1988: practice and trends.
15211155 - Hemodynamic responses to epinephrine-containing local anesthetic injection and to emerg...
|Type: JOURNAL ARTICLE Date: 2012-6-20|
|Title: Critical care (London, England) Volume: 16 ISSN: 1466-609X ISO Abbreviation: - Publication Date: 2012 Jun|
|Created Date: 2012-6-21 Completed Date: - Revised Date: -|
Medline Journal Info:
|Nlm Unique ID: 9801902 Medline TA: Crit Care Country: -|
|Languages: ENG Pagination: 225 Citation Subset: -|
|INSERM U1075, Caen F-14000, France. firstname.lastname@example.org.|
|APA/MLA Format Download EndNote Download BibTex|
Journal ID (nlm-ta): Crit Care
Journal ID (iso-abbrev): Crit Care
Publisher: BioMed Central
Copyright ©2012 BioMed Central Ltd
Print publication date: Year: 2012
Electronic publication date: Day: 20 Month: 6 Year: 2012
pmc-release publication date: Day: 20 Month: 6 Year: 2013
Volume: 16 Issue: 3
First Page: 225 Last Page: 225
PubMed Id: 22715815
Publisher Id: cc11297
|Clinical review: Update on neurally adjusted ventilatory assist - report of a round-table conference|
|Nicolas Terzi123||Email: email@example.com|
|Lise Piquilloud4||Email: Lise.Piquilloud@chuv.ch|
|Hadrien Rozé5||Email: firstname.lastname@example.org|
|Alain Mercat67||Email: email@example.com|
|Frédéric Lofaso89||Email: firstname.lastname@example.org|
|Stéphane Delisle10||Email: email@example.com|
|Philippe Jolliet4||Email: Philippe.Jolliet@chuv.ch|
|Thierry Sottiaux11||Email: Thierry.Sottiaux@cndg.be|
|Didier Tassaux4||Email: Didier.Tassaux@hcuge.ch|
|Jean Roesler12||Email: Jean.Roeseler@uclouvain.be|
|Alexandre Demoule13||Email: firstname.lastname@example.org|
|Samir Jaber14||Email: email@example.com|
|Jordi Mancebo15||Email: firstname.lastname@example.org|
|Laurent Brochard916||Email: Laurent.Brochard@hcuge.ch|
|Jean-Christophe Marie Richard161718||Email: Jcm.Richard@hcuge.ch|
1INSERM U1075, Caen F-14000, France
2Université de Caen, Caen F-14000, France
3CHRU Caen, Service de Réanimation Médicale, Caen F-14000, France
4Adult Intensive Care and Burn Unit, Lausanne University Hospital (CHUV), Rue du Bugnon 46, 1011 Lausanne, Switzerland
5Department of Anaesthesiology and Intensive Care 2, Bordeaux University Hospital, F-33000 Bordeaux, France
6LUNAM Université, Angers, France Université Angers, CHU Angers, Medical ICU, 4 rue Larrey, 49933 Angers, Cedex 09 France
7University Angers, CHU Angers, Réanimation Médicale, 49933 Angers, France
8Centre d'Investigation Clinique - Innovations Technologiques, Services de Physiologie - Explorations Fonctionnelles, Hôpital Raymond Poincaré, AP-HP, E.A. 4497, Université de Versailles - Saint Quentin en Yvelines, 92380 Garches, France
9INSERM U955, 94000 Créteil, France
10Hôpital du Sacré-Coeur de Montréal, 5400 boul. Gouin Ouest Montréal, Québec, H4J 1C5, Canada
11Intensive Care Unit, Clinique Notre Dame de Grâce, Gosselies Hospital, 212 Chaussée de Nivelles, B-6041-Gosselies, Belgium
12Intensive Care Unit, University Hospital St-Luc, 10 avenue Hippocrate, 1200 Brussels, Belgium
13Medical Intensive Care Unit and Respiratory Division, Groupe Hospitalier Pitié-Salpêtrière, Université Pierrer et Marie Curie and INSERM974, Paris, France
14Department of Critical Care Medicine and Anesthesiology (DAR B), Saint Eloi University Hospital and Universite of Montpellier, INSERM U1046, 80 Avenue Augustin Fliche, 34295 Montpellier, France
15Intensive Care Unit, Hospital de Sant Pau C. St Quinti 89, 08041 Barcelona, Spain
16Intensive Care Unit, Geneva University Hospital, School of Medicine, University of Geneva, Rue Gabrielle-Perret-Gentil 4, 1205 Geneva, Switzerland
17Medical Intensive Care Unit, University Hospital, Charles Nicolle, 76000 Rouen, France
18UPRES EA3830, Rouen, France
Pressure-support ventilation (PSV) is a widely used mode of assisted mechanical ventilation (MV), notably during the weaning phase [1,2]. Although PSV has been proven valuable in several acute clinical conditions [3,4], predefined ventilator settings - for example, airway pressure (Paw) - that remain unchanged from breath to breath are unlikely to provide optimal assistance all of the time.
To improve the match between the patient's needs and the assistance delivered by the ventilator, manufacturers have developed several new modes of MV [5,6]. Among these new modes we identified proportional-assist ventilation (PAV) and neurally adjusted ventilatory assist (NAVA).
PAV is a mode of support in which the ventilator pressure is proportional to instantaneous flow and volume, and hence to pressure generated by the respiratory muscles . Previous studies have demonstrated that PAV improves the synchrony between patient and ventilator, during several clinical conditions [8-12]. Based on the principles of the equation of motion, software (PAV+; Covidien, Boulder, Colorado, USA) has been developed that automatically adjusts the flow assist and the volume assist so that they always represent constant unloading fractions of the measured values of resistance and elastance loadings of the respiratory system [13-15]. Recent studies demonstrated that PAV+ is a safe and efficient ventilator mode in critically ill intubated patients [16,17]. During PAV the ventilator provides support only during the remaining duration of inspiratory effort, which can cause limitation when dynamic hyperinflation is present and when the inspiratory trigger is delayed due to intrinsic end-expiratory pressure.
The other support mode is NAVA, which will be discussed in this article. There are several similarities between PAV and NAVA, but this first round-table meeting focused on NAVA. A vast literature also exists concerning PAV, but this topic would require a whole chapter and will not be discussed in this current paper; hopefully PAV will be the topic of a different round table.
The present article is based on an investigator-initiated round-table meeting. The article aims to review the available knowledge on the physiological rationale and feasibility of the recently introduced NAVA MV modality. Throughout the article, we place emphasis on the most recent findings concerning adjustment of the NAVA settings; on the one hand considering specific issues associated with assisted modes of MV, and on the other considering the expectations placed upon NAVA.
NAVA is an assist mode of MV that delivers a pressure proportional to the integral of the electrical activity of the diaphragm (EAdi) , and therefore proportional to the neural output of the patient's central respiratory command. The level of pressure delivered is thus determined by the patient's respiratory-center neural output. With NAVA, the ventilator is triggered and cycled-off based on the EAdi value, which directly reflects the activity of the neural respiratory command. The inspiratory airway pressure applied by the ventilator is determined by the following equation:
where Paw is the instantaneous airway pressure (cmH2O), EAdi is the instantaneous integral of the diaphragmatic electrical activity signal (μV), and the NAVA level (cmH2O/μV or per arbitrary unit) is a proportionality constant set by the clinician.
In February 2011, several European and Canadian investigators with clinical results about NAVA available in publication or in abstract format organized a round-table discussion at the Geneva University Hospital to describe and discuss recent advances regarding NAVA. A representative of the company that commercializes the NAVA machine (Maquet Critical Care SA, Sölna, Sweden) was invited to attend the meeting in order to answer only technical questions. Maquet Critical Care SA agreed to sign a disclosure form before the meeting specifying that neither the minutes of the meeting nor the content of the report could be modified and/or used for commercial purposes. The main purpose of this meeting was for all of the investigators and participants to expose their standpoint and questions about NAVA, and to share the main results of their studies. Maquet Critical Care SA agreed to provide financial support for organizing the meeting, as detailed at the end of the manuscript, but was not responsible for choosing participants and did not take any part in the writing of this report. We here describe the content of the round-table discussion, focusing on a selection of the most recent studies [19-33] (Table 1).
Assisted modes generally aim at synchronizing the ventilator insufflation to the patient's effort, both to optimize comfort and to minimize the work of breathing. The price to pay for this strategy is a risk of patient-ventilator asynchrony, which can be defined as a mismatch between the patient's neural output and the ventilator's inspiratory and expiratory times [34-37]. Thille and colleagues reported that one-quarter of patients had high rates of asynchrony during assisted ventilation . Frequent asynchrony is associated with a longer duration of MV [34,38].
Compelling evidence accumulated over the last decade also supports the use of tidal volume (VT) values that are lower than those traditionally used. Lower VT values than traditionally used have several main advantages: they diminish the risk of ventilator-induced lung injury [39-41]; they preserve spontaneous breathing by avoiding respiratory alkalosis, thus preventing diaphragmatic disuse atrophy associated with MV [42-48]; they diminish several types of patient-ventilator asynchrony ; and they may improve the efficiency of gas exchange . Assisted modes of ventilation that maintain at least part of the patient's spontaneous breathing activity contribute to preventing these pulmonary and muscular complications.
The new challenge in developing ventilation strategies thus consists of minimizing the risk of lung injury, avoiding disuse atrophy of the diaphragm, and improving the match between the patient's needs and the assistance delivered by the ventilator . New ventilation modes have been designed to meet this challenge , and NAVA is a pressure-assisted mode in which the pressure delivered by the ventilator is proportional to the electrical activity of the diaphragm recorded continuously through an esophageal probe . NAVA theoretically delivers pressure proportional to the neural output of the patient's central respiratory command. During NAVA, however, reliable positioning of the catheter is mandatory in order to obtain a representative EAdi signal from the diaphragm. Barwing and colleagues have evaluated whether a formula based on the measurement from nose to ear lobe to xiphoid process of the sternum (the NEX distance) modified for the EAdi catheter (NEXmod) is adequate for predicting the accurate position of the esophageal probe . They observed in 18 of 25 patients (72%) that at NEXmod the EAdi signal was suitable for running NAVA. The NAVA mode was possible at the optimal position in four patients - the optimal position being defined by checking three criteria: stable EAdi signals, electrical activity highlighted in central leads of the catheter positioning tool, and an absence of the p-wave in the distal lead. The authors thus concluded that positioning the EAdi catheter using NEXmod gives a good approximation in most of the patients.
Moreover, the body position, positive end-expiratory pressure (PEEP) and intra-abdominal pressure are factors known to influence the position of the diaphragm. Barwing and colleagues therefore enrolled 20 patients in order to evaluate the effects of these factors on catheter position . They evaluated six different situations regarding the PEEP, body position and intra-abdominal pressure. Their results demonstrated that these factors may modify the EAdi catheter optimal position, although not compromising a stable signal due to the wide electrode array. One can therefore conclude that the optimal catheter position should be adjusted after major changes in ventilator settings, clinical condition or patient positioning.
The time lag between the neural inspiratory input and the occurrence of a ventilator breath affects all steps of the respiratory cycle (initiation, insufflation, and cycling-off for expiration) . Among the different forms of asynchrony, ineffective triggering (also known as wasted effort) is the most common during invasive MV. During noninvasive ventilation (NIV), leaks at the patient-ventilator interface impair the function of the pneumatic trigger and cycling system , thus promoting specific asynchronies (autotriggering and prolonged insufflation) .
Ineffective efforts are explained both by patients' characteristics and by ventilator settings. The presence of intrinsic PEEP increases the patient effort required to trigger the ventilator, thereby increasing the likelihood that the patient's inspiratory effort will fail to trigger a ventilator breath [36,53,56]. A weak inspiratory effort, which may occur during situations of low respiratory drive such as excessive ventilation, is also a risk factor and is common in patients receiving high assist levels  or sedation . An excessive level of pressure support is also associated with prolonged insufflation, thus promoting hyperinflation and intrinsic PEEP. Reduction of ineffective efforts is often possible through a careful optimization of ventilator settings, at least in short-term studies. Reducing VT during PSV can improve most factors contributing to ineffective efforts . Thille and colleagues showed that wasted efforts could be decreased without increasing the patient's work of breathing, with the main goal of decreasing the pressure-support level to obtain VT values of about 6 ml/kg predicted body weight . Because high pressure-support levels are associated with prolonged insufflation beyond the end of the patient's neural inspiratory time, another useful means of decreasing wasted eff orts consists of adjusting the inspiratory time by increasing the flow threshold of the cycling criterion [49,57].
NAVA involves the transesophageal recording of diaphragmatic electrical activity using specifically designed technology to minimize measurement errors. The EAdi signal reliably monitors and controls the ventilatory assist . During NAVA, the EAdi triggers the assist when the patient initiates an inspiratory effort - even during expiration with intrinsic PEEP - and a decrease in EAdi terminates the assist. NAVA does not therefore depend on measurements of airway pressure or flow and keeps the assist synchronous with the inspiratory efforts (independent of the presence of leaks or intrinsic PEEP) [19,21,22,25,29,59]. NAVA thus has two important features: the delivered pressure is, in theory, synchronous with the diaphragmatic activity, and the VT is completely controlled by the output of the patient's respiratory control center .
A frequent form of minor patient-ventilator asynchrony is a long inspiratory trigger delay (time lag between the onset of neural inspiration, then the detection of a breath initiated by the patient and, finally, the onset of ventilator pressurization). Several factors may increase the inspiratory trigger delay during PSV, including the presence of intrinsic PEEP and suboptimal ventilator performance . The cycling-off delay is the time difference between the end of the neural inspiratory ramp and the end of ventilator pressurization. Piquilloud and colleagues compared these delays and their consequences between NAVA and PSV in a group of 22 patients intubated for acute respiratory failure. The inspiratory trigger delay, the excess inspiratory time, and the frequency of patient-ventilator asynchrony were compared between the two modes . Compared with PSV, NAVA substantially improved patient-ventilator synchrony by reducing the inspiratory trigger delay and the total number of asynchrony events, and by improving expiratory cycling-off.
Increasing the level of ventilatory assist with standard modes may expose the patient to potentially dangerous levels of volume and pressure, and to uncoupling between the patient's neural output and ventilator assistance. In contrast to PSV, there is good evidence that NAVA offers protection against excessive Paw and VT values because there is a downregulation of EAdi in response to increasing assistance levels: the net result is a decrease in the amount of assistance provided [20,21,61-63]. The absence of a VT increase with increasing NAVA levels suggests that the Hering-Breuer reflex is operative , stopping the output from the respiratory control center at the same VT level, irrespective of the NAVA level. Unloading of the respiratory muscles is always partial, as some level of spontaneous activity is maintained, and patient- ventilator synchrony is improved.
Several studies have evaluated the impact of increasing PSV levels versus NAVA levels using similar methods of setting the ventilator [20-22,25]. Inspiratory pressure support was titrated in order to obtain 6 to 8 ml/kg predicted body weight during active inspiration. During PSV, the ventilator function 'NAVA Preview' estimates the NAVA level that would achieve the same peak inspiratory pressure. All studies performed in the ICU consistently showed that NAVA, in contrast to PSV, averted the risk of overassistance when the assist level was increased gradually. NAVA also improved patient-ventilator synchrony, in contrast to PSV, regardless of the underlying diagnosis. Very high levels of NAVA, however, might result in unstable periodic breathing patterns with delivery of high tidal volume followed by periods of apnea and signs of discomfort . To separate the effects of neural triggering and those of proportional assistance, Terzi and colleagues studied a selected population of patients recovering from acute respiratory distress syndrome, using NAVA with two inspiratory triggers: the EAdi signal and the inspiratory flow threshold used previously for PSV  (Figure 1a,b). Not only proportional assistance but also neural triggering improved patient-ventilator synchrony in these patients during the weaning process.
All of the available studies of NAVA in ICU patients have limitations regarding the clinical applicability of the results. Except for two studies [19,25], the patient population was heterogeneous in terms of the cause of respiratory failure. The evaluation time was relatively short in eight studies, but not for two studies [24,27].
Interestingly, and for reasons that are not yet fully understood, NAVA compared with PSV seemed to improve the partial pressure of oxygen in arterial blood in some studies independent of changes in the partial pressure of carbon dioxide in arterial blood (PaCO2) [25,27]. One hypothesis is that the continuous spontaneous inspiratory activity during NAVA improves the matching between ventilation and perfusion. Earlier studies had established that partial ventilatory support allowing some degree of spontaneous breathing activity using modes of ventilation other than NAVA improved the ventilation/perfusion relationship compared with fully controlled MV . In addition, NAVA allows a more natural breathing pattern characterized by greater variability, which may also contribute to improve gas exchange  (see below).
According to the principle of homeostasis, the closed loop that regulates PaCO2 comprises: sensors (or detectors), which are chemoreceptors; a controller (or comparator), which is the central respiratory command; and effectors, which are the respiratory muscles. Each component controls the next component in the loop, and the effectors change their activity (that is, adapt) to keep the PaCO2 value relatively constant. In other words, EAdi and therefore the breathing pattern must adapt to a variety of conditions to maintain PaCO2 within the normal range. Another regulatory mechanism is optimization of the work of breathing. For example, the rate and/or the depth of breathing can be adjusted to minimize the energy expenditure at a given respiratory effort and/or to minimize the stretch on the lungs.
Any strategy based on automated feedback control of ventilatory support should ideally require neural information on the lung volume, rate of lung volume change, and transpulmonary pressure, which are provided by mechanoreceptors in the lungs and chest wall. Finally, the variability and complexity of the breathing pattern are influenced by several factors, including the load-capacity relationship of the respiratory system [68-70], vagal afferent traffic to the brain , and the activity of the central pattern generators .
Ventilatory activity is nonlinear in nature and exhibits chaos-like mathematical complexity [72,73]. Variability is a mathematically complex notion, often expressed using the coefficient of variation, which is the ratio of the standard deviation over the mean. However, the complexity of flow and EAdi variability can also be described using noise titration, the largest Lyapunov exponent, Kolmogorov-Sinai entropy, and three-dimensional phase portraits [74,75]. Schmidt and colleagues used these methods to compare respiratory variability and complexity during PSV and NAVA . Compared with PSV, NAVA increased breathing pattern variability and flow complexity without changing EAdi complexity. Accordingly, when the NAVA level was increased from zero to a high level in healthy individuals, they adapted their inspiratory activity to the NAVA level in order to control VT and to regulate PaCO2 over a broad range of NAVA settings . In contrast, during high-level PSV, VT became almost entirely determined by the ventilator and hypocapnia developed as previously shown in healthy subjects [76,77]. These differences between NAVA and PSV establish that with NAVA, even at a high level of assistance, VT is not imposed by the ventilator but remains under the control of the patient's central respiratory command. NAVA therefore decreases the risk of over-assistance. The extent to which the preserved variability associated with NAVA is beneficial remains to be established. Whether variability restoration could be used to adapt NAVA settings also warrants further studies, as well as the development of specific tools for assessing variability at the bedside.
Patients with respiratory failure probably adjust their breathing activity to achieve the best compromise between the muscular effort needed to breathe and the sensory cost of tolerating elevated PaCO2 levels. NAVA acts as an additional external cost-free muscle controlled by the central respiratory command. NAVA therefore does not seem to alter the closed loop that controls the PaCO2 and respiratory pattern optimization. Accordingly, when introducing NAVA in patients with respiratory failure, progressively increasing the NAVA level allows the PaCO2 (that is, VT) to improve to the optimal value. Further NAVA level increases then lead to respiratory effort adjustments aimed at maintaining this optimal PaCO2 value, but do not change VT .
Moreover, Karagiannidis and colleagues intended recently to evaluate the physiological effect of extracorporeal membrane oxygenation on the pattern of breathing in patients with severe lung failure treated with NAVA . They demonstrated that a downregulation of extracorporeal exchange gas transfer caused an immediate upregulation of ventilation. Eucapnia under NAVA was preserved because the patients adjusted their minute ventilation to their needs. These interesting data highlighted once again that the ventilatory adaptation to maintain normocapnia remains under NAVA.
Determining the optimal NAVA level remains challenging, and several methods have been suggested. Contrary to PSV and as already described, NAVA generates VT levels that can remain constant independent of the assist level once the patient's ventilation needs appear to be satisfied . Consequently, NAVA settings cannot be adjusted based solely on VT (and/or the corresponding PaCO2 target).
Brander and colleagues tried to find the best NAVA level using breathing pattern analysis during a titration procedure . Titration consisted of starting at a minimal assist level of around 3 cmH2O and then increasing the NAVA level every 3 minutes in steps of 1 cmH2O per arbitrary unit (the amount of microvolts recorded from the EAdi signal). The response in terms of VT and Paw was biphasic. During the first phase, VT and Paw increased while the esophageal pressure-time product (that is, inspiratory muscle effort) and EAdi decreased. Further increases in the NAVA level (second phase) did not significantly change Paw or VT but continued to decrease the esophageal pressure-time product and EAdi. The first phase may thus indicate an insufficient NAVA level to supplement the patient's weak breathing effort, while the beginning of the second phase may correspond to the minimal assist level that satisfies the patient's respiratory demand. The optimal (or adequate) NAVA level may thus be indicated by the inflection point of the airway pressure trend graph during a stepwise increase in the NAVA level (Figure 2). In this study the patients were ventilated with these settings for 3 hours without experiencing adverse hemodynamic or respiratory events . Interestingly, the optimal NAVA level occurred at about 75% of the highest EAdi obtained with the minimal NAVA level and PEEP .
As suggested, titration of the NAVA level may be performed by systematically increasing the NAVA level to determine the optimal setting with regard to unloading patient's respiratory muscles [20,61,79]. During a recent observational study, transferring patients to NAVA was uneventful and the NAVA level contributed to adjustments of the preset NAVA level . Interpretation of several interacting physiological parameters might be difficult in cases in which there is no marked decrease in EAdi during NAVA titration . An automated approach enabled faster identification of the best NAVA level with a good accuracy .
Instead of stepwise titration, Rozé and colleagues tried to find the best NAVA level using an EAdi target of 60% of the highest EAdi value recorded during spontaneous breathing . This measurement was reassessed daily using a spontaneous breathing trial with a pressure-support level of 7 cmH2O and no PEEP. This method proved feasible and well tolerated until extubation (Figure 3). The 60% of the highest EAdi value threshold was based on a muscular rehabilitation protocol developed using data on diaphragmatic electromyogram activation during exercise . Whether this approach is also optimal during assisted ventilation needs further evaluation. It is worth noting that EAdi measured during the daily spontaneous breathing trial increased steadily over time in all patients until successful extubation . This improvement probably originated in multiple factors, including discontinuation of sedative agents and gradual restoration of the functional electrophysiologic activity of the diaphragm. Monitoring diaphragmatic activity may be of clinical interest and could be achieved using the NAVA electrode.
Using EAdi analysis to titrate NAVA is an interesting approach that could potentially be easier to use than the breathing pattern analysis method of Brander and colleagues (VT change during titration) . The EAdi target of 60% of the highest EAdi value with 7 cmH2O of PSV proposed by Rozé and colleagues should be used cautiously , as Brander and colleagues found that the EAdi at the optimal NAVA level was equal to 75% of the highest EAdi value recorded with minimal NAVA (inspiratory Paw above PEEP = 3 cmH2O) . Further studies are clearly needed to better determine the optimal EAdi target.
NIV is a specific clinical situation during which the occurrence of leaks may greatly affect patient-ventilator interactions, thereby complicating the determination of optimal ventilator settings. In a study by Vignaux and colleagues, more than 40% of patients experienced various types of asynchrony during conventional NIV and the asynchrony rate correlated with the level of leakage . With NAVA, assistance is delivered based on neural triggering, which is not affected by leakage. NAVA may thus, in theory, diminish asynchrony events, thereby improving the tolerance of NIV. New software for NIV has been developed using NAVA technology. With this specifically designed algorithm, NIV assistance is triggered and cycled-off by the neural diaphragmatic activity, which would be expected to improve patient- ventilator synchrony during NIV. This hypothesis has not yet been fully investigated.
A study of NIV-PSV with a helmet interface in healthy volunteers compared asynchrony with a neural trigger and a conventional pneumatic trigger . Increasing PSV levels and respiratory rates applied with neural triggering and cycling-off produced significantly less impairment of synchrony, trigger effort, and breathing comfort, compared with conventional pneumatic triggering and cycling-off .
Cammarotta and colleagues recently compared NAVA and NIV-PSV delivered through a helmet interface in postextubation hypoxemic patients . Ten patients underwent three 20-minute trials of helmet NIV in PSV, NAVA, and PSV again. The authors demonstrated that there was less asynchrony during NAVA than during PSV and no difference in gas exchange, although there were more leaks during NAVA. Moreover it is important to underline that the PSV mode chosen was specifically dedicated to NIV, whereas the NAVA mode dedicated to NIV that is now currently available did not exist at the time of this study.
Recent data obtained in low-birth-weight infants indicate that NAVA can maintain synchrony - both in terms of timing and proportionality - even after extubation in patients with an excessively leaky interface under NIV (all infants in this study were ventilated using a single nasal prong) .
Another consideration for NIV that deserves attention in the near future is the impact on swallowing, phonation, and sleep quality, most notably when NIV is used for several days. Improvements in swallowing performance have been reported in neuromuscular patients receiving MV compared with spontaneous breathing [84,85]. The close relationship between the muscles involved in swallowing and those contributing to inspiration was evidenced by Orlikowski and colleagues using an original method of tongue-strength measurement. The significant tongue weakness observed in 16 weak patients with Guillain-Barré syndrome correlated with the alterations in respiratory parameters . Additional physiological studies are required to document the potential benefits of NAVA on swallowing-breathing interactions during NIV.
Sleep quality during NIV has been shown to be a predictor of success or failure . Sleep quality can also be improved compared with standard NIV settings by careful physiological titration of the ventilator settings . Patient-ventilator asynchrony can cause sleep disruption. Bosma and colleagues demonstrated that PAV, a mode of partial ventilatory support in which the ventilator applies pressure in proportion to the inspiratory load, was more effective than PSV in matching the ventilatory requirements to the level of ventilator assistance, thereby resulting in fewer patient-ventilator asynchronies and better quality of sleep . Delisle and colleagues recently obtained sleep recordings during a crossover study comparing NAVA and PSV in 14 mechanically ventilated patients . Each condition was studied for 4 hours, and recordings were obtained over 19 consecutive hours in all. Patient-ventilator asynchrony varied significantly across sleep stages, and no asynchrony occurred with NAVA. Overassistance occurred only with PSV, which probably explained the improvements in physiological indices of sleep quality observed with NAVA.
MV in children and in low-birth-weight infants is more difficult to apply than in adults and has several speciicities. First infants take a very small tidal volume, have a rapid respiratory rate, have a limited chest wall musculature, and have variable and fluctuating lung compliance. Second, most neonatal units use uncuffed tracheal tubes for fears of pressure necrosis and air leak is always present, making reliable measurements and triggering problematic. Third, ventilators that are efficient in adults are not systematically efficient in children and infants, mainly because the inspiratory triggers are not sufficiently sensitive for early detection of infants'/children's inspiratory effort .
Whether or not the respiratory drive of the preterm infant is suitable to control MV is unknown. Beck and colleagues first evaluated patient-ventilator interaction with NAVA in seven very-low-birth-weight infants . As suggested by previous animal studies , they demonstrated that NAVA could be implemented for a short-term period, both invasively and noninvasively, in infants with body weight as low as 640 g up to 3 years old. During invasive ventilation with NAVA, EAdi and ventilator pressure were correlated and patient-ventilator synchrony was improved compared with the other mode. Moreover, this synchrony persisted after extubation while ventilating the patient with an excessively leaky interface. After this first physiological demonstration, Bengtsson and Edberg demonstrated the clinical feasibility and safety with use of NAVA in pediatric patients . Similarly, Breatnach and colleagues compared NAVA (with a neural trigger) and PSV (with a pneumatic trigger) in 16 ventilated infants . This prospective crossover comparison demonstrated that ventilation with NAVA improved patient-ventilator synchrony.
Furthermore, Alander and colleagues recently compared NAVA with pressure-controlled ventilation for newborns and with pressure-regulated controlled ventilation for children older than 3 months (with conventional trigger modes: pressure and flow trigger) . In this prospective cross-over study, 18 patients requiring MV were randomized for 10 minutes with the different modes. During NAVA, the peak airway pressure was lower, the respiratory rate was 10 breaths/minute higher than in the pressure group, and patient-ventilator synchronization was improved. However, there were no differences in tidal volume and in oxygen saturation.
To evaluate the effects of the neural trigger on trigger delay, ventilator response time, or work of breathing, Clement and colleagues conducted a study in 23 pediatric patients aged 0 to 24 months with a diagnosis of bronchiolitis presenting respiratory failure requiring MV . The authors compared the neural trigger and the pneumatic trigger using similar NAVA assistance, and observed that the trigger delay, the ventilator response time, and the work of breathing were reduced by the neural trigger.
Finally, all of these studies seem to demonstrate the feasibility of and a potential advantage for NAVA in children compared with the other assisted ventilatory modes. Because patient-ventilator synchrony is improved with NAVA, the children may require lower doses of sedation with this mode of MV , which could reduce the time of MV.
Clinical studies obtained in critically ill patients confirm many of the expected short-term physiological benefits associated with NAVA, as discussed above.
Particularly, NAVA seems to markedly improve the problems of nonsynchronization between the patient and the ventilator and the problems of risk of overventilation - including the risk of ineffective or missed inspiratory efforts due to intrinsic PEEP observed in chronic obstructive pulmonary disease patients, or to a rapid breathing frequency with a very small tidal volume observed in pediatric patients.
In addition, NAVA minimizes the risk of overinflation because the duration and level of pressurization remain under respiratory-center control, and minimizes the risk of diaphragmatic inactivity because the presence of pressure assistance requires the presence of this inspiratory activity.
A preserved respiratory muscle function is pivotal for weaning from MV . By using NAVA, which outperforms the previous modes of MV for adequately assisting the patient's inspiratory effort without inducing patient/ventilator dyssynchrony, a reduction in the duration of MV could be expected. Studies are needed to evaluate the best time to begin the weaning process with NAVA.
The NAVA setting is an important question not yet fully resolved. If clinicians are accustomed to set a PSV level, this is not the case for NAVA. Furthermore, because the breathing pattern is less modified by the NAVA setting than during PSV, it is much less informative for NAVA adjustment. As described above, the literature suggests that the adjustment should consider the electromyographic activity of the diaphragm, but this method is not simple. As recently proposed, a direct evaluation of patient comfort and sense of dyspnea for the NAVA setting should be evaluated .
Finally, the next research step will be to evaluate NAVA over longer periods, in order to know whether this mode can replace the modes usually used during MV and the weaning period, like PSV. Appreciating the safety, the feasibility and the constraints of this technology will be useful. It is therefore necessary to test, during the total weaning period, the effectiveness of the esophageal probe and to know whether it is regularly necessary to adjust the probe position. One of the most difficult questions to address, however, is in which situations it is not desirable to let the respiratory centers drive the ventilation. Situations of severe metabolic acidosis, of high respiratory drive and of high catecholamine levels may induce situations of extreme hyperventilation, which may be dangerous for the lungs. When sedation and/or paralysis become necessary is therefore an important question to address before widespread use of this mode .
NAVA, which is based on an original physiological concept, adds new knowledge on patient-ventilator interactions during spontaneous breathing, thus helping to unravel the complex mechanisms involved in breathing control during MV. There is compelling evidence that NAVA, as well as the PAV+ software, improves patient-ventilator interactions and increases respiratory variability in comparison with PSV. This advantage holds potential for many applications. The short-term and long-term experience with NAVA, however, remains scant. Further clinical studies are needed to assess the feasibility and safety of NAVA. A key challenge is how to determine the best NAVA settings according to the patient's ventilatory needs and the acceptable level of work of breathing.
EAdi: electrical activity of the diaphragm; MV: mechanical ventilation; NAVA: neurally adjusted ventilatory assist; NEXmod: nose to ear lobe to xiphoid process of the sternum distance modified for the EAdi catheter; NIV: noninvasive ventilation; PaCO2: partial pressure of carbon dioxide in arterial blood; PAV: proportional-assist ventilation; Paw: airway pressure; PEEP: positive end-expiratory pressure; PSV: pressure-support ventilation; VT: tidal volume.
SJ received a grant from Maquet Critical Care SA in 2009 and 2010 for clinical research. PJ's research laboratory received grants from Maquet Critical Care SA Drager, Res-Med, Hamilton, for clinical research over the last 5 years. LB's research laboratory received a grant from Maquet Critical Care SA in 2009 and 2010 for clinical research. LB's research laboratory has also received research grants from Drager, Covidien, General Electric and Fisher Paykel over the last 5 years. J-CMR and AM received grants from Drager, Covidien, and General Electric over the last 5 years for their research group. AD's research laboratory received a grant from Maquet Critical Care SA in 2008 and 2009 for clinical research. AD coordinates a multicenter clinical study on NAVA. The remaining authors declare that they have no competing interests.
NT, J-CMR and LB initiated and wrote the manuscript; all authors contributed to the revision of this. All authors read and approved the final manuscript.
The round-table meeting was initiated and organized solely by the authors. Maquet Critical Care SA provided financial support for travel and accommodation costs, thus allowing the authors to meet for a half-day conference, but had no influence on the content of the manuscript.
|Esteban A,Anzueto A,Alía I,Gordo F,Apezteguía C,Pálizas F,Cide D,Goldwaser R,Soto L,Bugedo G,Rodrigo C,Pimentel J,Raimondi G,Tobin MJ,How is mechanical ventilation employed in the intensive care unit? An international utilization reviewAm J Respir Crit Care MedYear: 20001611450145810806138|
|Esteban A,Ferguson ND,Meade MO,Frutos-Vivar F,Apezteguia C,Brochard L,Raymondos K,Nin N,Hurtado J,Tomicic V,González M,Elizalde J,Nightingale P,Abroug F,Pelosi P,Arabi Y,Moreno R,Jibaja M,D'Empaire G,Sandi F,Matamis D,Montañez AM,Anzueto A; VENTILA Group,Evolution of mechanical ventilation in response to clinical researchAm J Respir Crit Care MedYear: 200817717017717962636|
|Cereda M,Foti G,Marcora B,Gili M,Giacomini M,Sparacino ME,Pesenti A,Pressure support ventilation in patients with acute lung injuryCrit Care MedYear: 2000281269127510.1097/00003246-200005000-0000210834664|
|Nava S,Bruschi C,Fracchia C,Braschi A,Rubini F,Patient-ventilator interaction and inspiratory effort during pressure support ventilation in patients with different pathologiesEur Respir JYear: 19971017718310.1183/09031936.97.100101779032512|
|Navalesi P,Costa R,New modes of mechanical ventilation: proportional assist ventilation, neurally adjusted ventilatory assist, and fractal ventilationCurr Opin Crit CareYear: 20039515810.1097/00075198-200302000-0001012548030|
|Moerer O,Effort-adapted modes of assisted breathingCurr Opin Crit CareYear: 201218616922222841|
|Younes M,Puddy A,Roberts D,Light RB,Quesada A,Taylor K,Oppenheimer L,Cramp H,Proportional assist ventilation. Results of an initial clinical trialAm Rev Respir DisYear: 199214512112910.1164/ajrccm/145.1.1211731574|
|Giannouli E,Webster K,Roberts D,Younes M,Response of ventilator-dependent patients to different levels of pressure support and proportional assistAm J Respir Crit Care MedYear: 19991591716172510351909|
|Alexopoulou C,Kondili E,Vakouti E,Klimathianaki M,Prinianakis G,Georgopoulos D,Sleep during proportional-assist ventilation with load-adjustable gain factors in critically ill patientsIntensive Care MedYear: 2007331139114710.1007/s00134-007-0630-217458541|
|Ranieri VM,Giuliani R,Mascia L,Grasso S,Petruzzelli V,Puntillo N,Perchiazzi G,Fiore T,Brienza A,Patient-ventilator interaction during acute hypercapnia: pressure-support vs. proportional-assist ventilationJ Appl PhysiolYear: 1996814264368828695|
|Bosma K,Ferreyra G,Ambrogio C,Pasero D,Mirabella L,Braghiroli A,Appendini L,Mascia L,Ranieri VM,Patient-ventilator interaction and sleep in mechanically ventilated patients: pressure support versus proportional assist ventilationCrit Care MedYear: 2007351048105410.1097/01.CCM.0000260055.64235.7C17334259|
|Grasso S,Puntillo F,Mascia L,Ancona G,Fiore T,Bruno F,Slutsky AS,Ranieri VM,Compensation for increase in respiratory workload during mechanical ventilation. Pressure-support versus proportional-assist ventilationAm J Respir Crit Care MedYear: 20001613 Pt 181982610712328|
|Younes M,Webster K,Kun J,Roberts D,Masiowski B,A method for measuring passive elastance during proportional assist ventilationAm J Respir Crit Care MedYear: 2001164506011435238|
|Younes M,Kun J,Masiowski B,Webster K,Roberts D,A method for noninvasive determination of inspiratory resistance during proportional assist ventilationAm J Respir Crit Care MedYear: 200116382983911282753|
|Kondili E,Prinianakis G,Alexopoulou C,Vakouti E,Klimathianaki M,Georgopoulos D,Respiratory load compensation during mechanical ventilation - proportional assist ventilation with load-adjustable gain factors versus pressure supportIntensive Care MedYear: 20063269269910.1007/s00134-006-0110-016523329|
|Xirouchaki N,Kondili E,Vaporidi K,Xirouchakis G,Klimathianaki M,Gavriilidis G,Alexandopoulou E,Plataki M,Alexopoulou C,Georgopoulos D,Proportional assist ventilation with load-adjustable gain factors in critically ill patients: comparison with pressure supportIntensive Care MedYear: 2008342026203410.1007/s00134-008-1209-218607562|
|Costa R,Spinazzola G,Cipriani F,Ferrone G,Festa O,Arcangeli A,Antonelli M,Proietti R,Conti G,A physiologic comparison of proportional assist ventilation with load-adjustable gain factors (PAV+) versus pressure support ventilation (PSV)Intensive Care MedYear: 2011371494150010.1007/s00134-011-2297-y21720909|
|Sinderby C,Navalesi P,Beck J,Skrobik Y,Comtois N,Friberg S,Gottfried SB,Lindstrom L,Neural control of mechanical ventilation in respiratory failureNat MedYear: 199951433143610.1038/7101210581089|
|Wu XY,Huang YZ,Yang Y,Liu SQ,Liu HG,Qiu HB,[Effects of neurally adjusted ventilatory assist on patient-ventilator synchrony in patients with acute respiratory distress syndrome]Zhonghua Jie He He Hu Xi Za ZhiYear: 20093250851219954004|
|Brander L,Leong-Poi H,Beck J,Brunet F,Hutchison SJ,Slutsky AS,Sinderby C,Titration and implementation of neurally adjusted ventilatory assist in critically ill patientsChestYear: 200913569570310.1378/chest.08-174719017889|
|Colombo D,Cammarota G,Bergamaschi V,De Lucia M,Corte FD,Navalesi P,Physiologic response to varying levels of pressure support and neurally adjusted ventilatory assist in patients with acute respiratory failureIntensive Care MedYear: 2008342010201810.1007/s00134-008-1208-318629471|
|Spahija J,de Marchie M,Albert M,Bellemare P,Delisle S,Beck J,Sinderby C,Patient-ventilator interaction during pressure support ventilation and neurally adjusted ventilatory assistCrit Care MedYear: 20103851852610.1097/CCM.0b013e3181cb0d7b20083921|
|Schmidt M,Demoule A,Cracco C,Gharbi A,Fiamma MN,Straus C,Duguet A,Gottfried SB,Similowski T,Neurally adjusted ventilatory assist increases respiratory variability and complexity in acute respiratory failureAnesthesiologyYear: 201011267068110.1097/ALN.0b013e3181cea37520179505|
|Rozé H,Lafrikh A,Perrier V,Germain A,Dewitte A,Gomez F,Janvier G,Ouattara A,Daily titration of neurally adjusted ventilatory assist using the diaphragm electrical activityIntensive Care MedYear: 2011371087109410.1007/s00134-011-2209-121424827|
|Terzi N,Pelieu I,Guittet L,Ramakers M,Seguin A,Daubin C,Charbonneau P,du Cheyron D,Lofaso F,Neurally adjusted ventilatory assist in patients recovering spontaneous breathing after acute respiratory distress syndrome: physiological evaluationCrit Care MedYear: 2010381830183710.1097/CCM.0b013e3181eb3c5120639752|
|Piquilloud L,Vignaux L,Bialais E,Roeseler J,Sottiaux T,Laterre PF,Jolliet P,Tassaux D,Neurally adjusted ventilatory assist improves patient-ventilator interactionIntensive Care MedYear: 20113726327110.1007/s00134-010-2052-920871978|
|Coisel Y,Chanques G,Jung B,Constantin JM,Capdevila X,Matecki S,Grasso S,Jaber S,Neurally adjusted ventilatory assist in critically ill postoperative patients: a crossover randomized studyAnesthesiologyYear: 201011392593510.1097/ALN.0b013e3181ee2ef120823760|
|Passath C,Takala J,Tuchscherer D,Jakob SM,Sinderby C,Brander L,Physiologic response to changing positive end-expiratory pressure during neurally adjusted ventilatory assist in sedated, critically ill adultsChestYear: 201013857858710.1378/chest.10-028620435654|
|Beck J,Reilly M,Grasselli G,Mirabella L,Slutsky AS,Dunn MS,Sinderby C,Patient-ventilator interaction during neurally adjusted ventilatory assist in low birth weight infantsPediatr ResYear: 20096566366810.1203/PDR.0b013e31819e72ab19218884|
|Bengtsson JA,Edberg KE,Neurally adjusted ventilatory assist in children: an observational studyPediatr Crit Care MedYear: 20101125325710.1097/PCC.0b013e3181b0655e19593241|
|Breatnach C,Conlon NP,Stack M,Healy M,O'Hare BP,A prospective crossover comparison of neurally adjusted ventilatory assist and pressure-support ventilation in a pediatric and neonatal intensive care unit populationPediatr Crit Care MedYear: 20101171110.1097/PCC.0b013e3181b0630f19593246|
|Cammarota G,Olivieri C,Costa R,Vaschetto R,Colombo D,Turucz E,Longhini F,Della Corte F,Conti G,Navalesi P,Noninvasive ventilation through a helmet in post-extubation hypoxemic patients: physiologic comparison between neurally adjusted ventilatory assist and pressure support ventilationIntensive Care MedYear: 2011371943195010.1007/s00134-011-2382-222005826|
|Clement KC,Thurman TL,Holt SJ,Heulitt MJ,Neurally triggered breaths reduce trigger delay and improve ventilator response times in ventilated infants with bronchiolitisIntensive Care MedYear: 2011371826183210.1007/s00134-011-2352-821946913|
|Thille AW,Rodriguez P,Cabello B,Lellouche F,Brochard L,Patient-ventilator asynchrony during assisted mechanical ventilationIntensive Care MedYear: 2006321515152210.1007/s00134-006-0301-816896854|
|Tobin MJ,Jubran A,Laghi F,Patient-ventilator interactionAm J Respir Crit Care MedYear: 20011631059106311316635|
|Parthasarathy S,Jubran A,Tobin MJ,Cycling of inspiratory and expiratory muscle groups with the ventilator in airflow limitationAm J Respir Crit Care MedYear: 19981585 Pt 1147114789817695|
|Beck J,Tucci M,Emeriaud G,Lacroix J,Sinderby C,Prolonged neural expiratory time induced by mechanical ventilation in infantsPediatr ResYear: 20045574775410.1203/01.PDR.0000119368.21770.3314739354|
|de Wit M,Pedram S,Best AM,Epstein SK,Observational study of patient-ventilator asynchrony and relationship to sedation levelJ Crit CareYear: 200924748010.1016/j.jcrc.2008.08.01119272542|
|Dreyfuss D,Saumon G,Ventilator-induced lung injury: lessons from experimental studiesAm J Respir Crit Care MedYear: 19981572943239445314|
|Ricard JD,Dreyfuss D,Saumon G,Ventilator-induced lung injuryEur Respir J SupplYear: 2003422s9s12945994|
|The Acute Respiratory Distress Syndrome NetworkVentilation with lower tidal volumes as compared with traditional tidal volumes for acute lung injury and the acute respiratory distress syndromeN Engl J MedYear: 20003421301130810793162|
|Levine S,Nguyen T,Taylor N,Friscia ME,Budak MT,Rothenberg P,Zhu J,Sachdeva R,Sonnad S,Kaiser LR,Rubinstein NA,Powers SK,Shrager JB,Rapid disuse atrophy of diaphragm fibers in mechanically ventilated humansN Engl J MedYear: 20083581327133510.1056/NEJMoa07044718367735|
|Sieck GC,Mantilla CB,Effect of mechanical ventilation on the diaphragmN Engl J MedYear: 20083581392139410.1056/NEJMe080122618367743|
|Jaber S,Petrof BJ,Jung B,Chanques G,Berthet JP,Rabuel C,Bouyabrine H,Courouble P,Koechlin-Ramonatxo C,Sebbane M,Similowski T,Scheuermann V,Mebazaa A,Capdevila X,Mornet D,Mercier J,Lacampagne A,Philips A,Matecki S,Rapidly progressive diaphragmatic weakness and injury during mechanical ventilation in humansAm J Respir Crit Care MedYear: 201118336437110.1164/rccm.201004-0670OC20813887|
|Anzueto A,Peters JI,Tobin MJ,de los Santos R,Seidenfeld JJ,Moore G,Cox WJ,Coalson JJ,Effects of prolonged controlled mechanical ventilation on diaphragmatic function in healthy adult baboonsCrit Care MedYear: 1997251187119010.1097/00003246-199707000-000219233746|
|Sassoon CS,Ventilator-associated diaphragmatic dysfunctionAm J Respir Crit Care MedYear: 20021661017101810.1164/rccm.220700812379541|
|Sassoon CS,Caiozzo VJ,Manka A,Sieck GC,Altered diaphragm contractile properties with controlled mechanical ventilationJ Appl PhysiolYear: 2002922585259512015377|
|Sassoon CS,Zhu E,Caiozzo VJ,Assist-control mechanical ventilation attenuates ventilator-induced diaphragmatic dysfunctionAm J Respir Crit Care MedYear: 200417062663210.1164/rccm.200401-042OC15201132|
|Thille AW,Cabello B,Galia F,Lyazidi A,Brochard L,Reduction of patient-ventilator asynchrony by reducing tidal volume during pressure-support ventilationIntensive Care MedYear: 2008341477148610.1007/s00134-008-1121-918437356|
|Pinto Da Costa N,Di Marco F,Lyazidi A,Carteaux G,Sarni M,Brochard L,Effect of pressure support on end-expiratory lung volume and lung diffusion for carbon monoxideCrit Care MedYear: 2011392283228910.1097/CCM.0b013e3182227fce21666442|
|Barwing J,Ambold M,Linden N,Quintel M,Moerer O,Evaluation of the catheter positioning for neurally adjusted ventilatory assistIntensive Care MedYear: 2009351809181410.1007/s00134-009-1587-019652950|
|Barwing J,Pedroni C,Quintel M,Moerer O,Influence of body position, PEEP and intra-abdominal pressure on the catheter positioning for neurally adjusted ventilatory assistIntensive Care MedYear: 2011372041204510.1007/s00134-011-2373-321997127|
|Parthasarathy S,Jubran A,Tobin MJ,Assessment of neural inspiratory time in ventilator-supported patientsAm J Respir Crit Care MedYear: 20001622 Pt 154655210934085|
|Calderini E,Confalonieri M,Puccio PG,Francavilla N,Stella L,Gregoretti C,Patient-ventilator asynchrony during noninvasive ventilation: the role of expiratory triggerIntensive Care MedYear: 19992566266710.1007/s00134005092710470568|
|Vignaux L,Tassaux D,Carteaux G,Roeseler J,Piquilloud L,Brochard L,Jolliet P,Performance of noninvasive ventilation algorithms on ICU ventilators during pressure support: a clinical studyIntensive Care MedYear: 2010362053205910.1007/s00134-010-1994-220689921|
|Leung P,Jubran A,Tobin MJ,Comparison of assisted ventilator modes on triggering, patient effort, and dyspneaAm J Respir Crit Care MedYear: 1997155194019489196100|
|Tassaux D,Gainnier M,Battisti A,Jolliet P,Impact of expiratory trigger setting on delayed cycling and inspiratory muscle workloadAm J Respir Crit Care MedYear: 20051721283128910.1164/rccm.200407-880OC16109983|
|American Thoracic Society/European Respiratory SocietyATS/ERS Statement on respiratory muscle testingAm J Respir Crit Care MedYear: 200216651862412186831|
|Moerer O,Beck J,Brander L,Costa R,Quintel M,Slutsky AS,Brunet F,Sinderby C,Subject-ventilator synchrony during neural versus pneumatically triggered noninvasive helmet ventilationIntensive Care MedYear: 2008341615162310.1007/s00134-008-1163-z18512045|
|Thille AW,Lyazidi A,Richard JC,Galia F,Brochard L,A bench study of intensive-care-unit ventilators: new versus old and turbine-based versus compressed gas-based ventilatorsIntensive Care MedYear: 2009351368137610.1007/s00134-009-1467-719352622|
|Allo JC,Beck JC,Brander L,Brunet F,Slutsky AS,Sinderby CA,Influence of neurally adjusted ventilatory assist and positive end-expiratory pressure on breathing pattern in rabbits with acute lung injuryCrit Care MedYear: 2006342997300416957635|
|Brander L,Sinderby C,Lecomte F,Leong-Poi H,Bell D,Beck J,Tsoporis JN,Vaschetto R,Schultz MJ,Parker TG,Villar J,Zhang H,Slutsky AS,Neurally adjusted ventilatory assist decreases ventilator-induced lung injury and non-pulmonary organ dysfunction in rabbits with acute lung injuryIntensive Care MedYear: 2009351979198910.1007/s00134-009-1626-x19760209|
|Sinderby C,Beck J,Spahija J,de Marchie M,Lacroix J,Navalesi P,Slutsky AS,Inspiratory muscle unloading by neurally adjusted ventilatory assist during maximal inspiratory efforts in healthy subjectsChestYear: 200713171171710.1378/chest.06-190917356084|
|Guz A,Hering and Breuer revisited in humans: an invasive study before the days of ethics committeesAm J Respir Crit Care MedYear: 20011641110111111673193|
|Patroniti N,Bellani G,Saccavino E,Zanella A,Grasselli G,Isgro S,Milan M,Foti G,Pesenti A,Respiratory pattern during neurally adjusted ventilatory assist in acute respiratory failure patientsIntensive Care MedYear: 20123823023910.1007/s00134-011-2433-822127483|
|Putensen C,Rasanen J,Lopez FA,Downs JB,Effect of interfacing between spontaneous breathing and mechanical cycles on the ventilation-perfusion distribution in canine lung injuryAnesthesiologyYear: 19948192193010.1097/00000542-199410000-000187943842|
|Gama de Abreu M,Spieth PM,Pelosi P,Carvalho AR,Walter C,Schreiber-Ferstl A,Aikele P,Neykova B,Hubler M,Koch T,Noisy pressure support ventilation: a pilot study on a new assisted ventilation mode in experimental lung injuryCrit Care MedYear: 20083681882710.1097/01.CCM.0000299736.55039.3A18431269|
|Brack T,Jubran A,Tobin MJ,Effect of elastic loading on variational activity of breathingAm J Respir Crit Care MedYear: 1997155134113489105077|
|Brack T,Jubran A,Tobin MJ,Effect of resistive loading on variational activity of breathingAm J Respir Crit Care MedYear: 19981576 Pt 1175617639620902|
|Brack T,Jubran A,Tobin MJ,Dyspnea and decreased variability of breathing in patients with restrictive lung diseaseAm J Respir Crit Care MedYear: 20021651260126410.1164/rccm.220101811991875|
|Sammon MP,Bruce EN,Vagal afferent activity increases dynamical dimension of respiration in ratsJ Appl PhysiolYear: 199170174817622055854|
|Fiamma MN,Straus C,Thibault S,Wysocki M,Baconnier P,Similowski T,Effects of hypercapnia and hypocapnia on ventilatory variability and the chaotic dynamics of ventilatory flow in humansAm J Physiol Regul Integr Comp PhysiolYear: 2007292R1985R199310.1152/ajpregu.00792.200617218438|
|Wysocki M,Fiamma MN,Straus C,Poon CS,Similowski T,Chaotic dynamics of resting ventilatory flow in humans assessed through noise titrationRespir Physiol NeurobiolYear: 2006153546510.1016/j.resp.2005.09.00816303337|
|Schurmann T,Grassberger P,Entropy estimation of symbol sequencesChaosYear: 1996641442710.1063/1.16619112780271|
|Briggs K,An improved method for estimating Liapunov exponents of chaotic time seriesPhys Lett AYear: 1990151273210.1016/0375-9601(90)90841-B|
|Lofaso F,Isabey D,Lorino H,Harf A,Scheid P,Respiratory response to positive and negative inspiratory pressure in humansRespir PhysiolYear: 199289758810.1016/0034-5687(92)90072-51518988|
|Scheid P,Lofaso F,Isabey D,Harf A,Respiratory response to inhaled CO2 during positive inspiratory pressure in humansJ Appl PhysiolYear: 1994778768828002542|
|Karagiannidis C,Lubnow M,Philipp A,Riegger GA,Schmid C,Pfeifer M,Mueller T,Autoregulation of ventilation with neurally adjusted ventilatory assist on extracorporeal lung supportIntensive Care MedYear: 2010362038204410.1007/s00134-010-1982-620689930|
|Lecomte F,Brander L,Jalde F,Beck J,Qui H,Elie C,Slutsky AS,Brunet F,Sinderby C,Physiological response to increasing levels of neurally adjusted ventilatory assist (NAVA)Respir Physiol NeurobiolYear: 200916611712410.1016/j.resp.2009.02.01519429528|
|Barwing J,Linden N,Ambold M,Quintel M,Moerer O,Neurally adjusted ventilatory assist vs. pressure support ventilation in critically ill patients: an observational studyActa Anaesthesiol ScandYear: 2011551261127110.1111/j.1399-6576.2011.02522.x22092132|
|Ververidis D,Van Gils M,Passath C,Takala J,Brander L,Identification of adequate neurally adjusted ventilatory assist (NAVA) during systematic increases in the NAVA levelIEEE Trans Biomed EngYear: 2011582598260621690003|
|Sinderby C,Spahija J,Beck J,Kaminski D,Yan S,Comtois N,Sliwinski P,Diaphragm activation during exercise in chronic obstructive pulmonary diseaseAm J Respir Crit Care MedYear: 20011631637164111401887|
|Vignaux L,Vargas F,Roeseler J,Tassaux D,Thille AW,Kossowsky MP,Brochard L,Jolliet P,Patient-ventilator asynchrony during non-invasive ventilation for acute respiratory failure: a multicenter studyIntensive Care MedYear: 20093584084610.1007/s00134-009-1416-519183949|
|Terzi N,Orlikowski D,Aegerter P,Lejaille M,Ruquet M,Zalcman G,Fermanian C,Raphael JC,Lofaso F,Breathing-swallowing interaction in neuromuscular patients: a physiological evaluationAm J Respir Crit Care MedYear: 200717526927617110642|
|Terzi N,Prigent H,Lejaille M,Falaize L,Annane D,Orlikowski D,Lofaso F,Impact of tracheostomy on swallowing performance in Duchenne muscular dystrophyNeuromuscul DisordYear: 20102049349810.1016/j.nmd.2010.05.00920558065|
|Orlikowski D,Terzi N,Blumen M,Sharshar T,Raphael JC,Annane D,Lofaso F,Tongue weakness is associated with respiratory failure in patients with severe Guillain-Barre syndromeActa Neurol ScandYear: 200911936437010.1111/j.1600-0404.2008.01107.x18976323|
|Roche Campo F,Drouot X,Thille AW,Galia F,Cabello B,d'Ortho MP,Brochard L,Poor sleep quality is associated with late noninvasive ventilation failure in patients with acute hypercapnic respiratory failureCrit Care MedYear: 20103847748510.1097/CCM.0b013e3181bc824319789439|
|Fanfulla F,Delmastro M,Berardinelli A,Lupo ND,Nava S,Effects of different ventilator settings on sleep and inspiratory effort in patients with neuromuscular diseaseAm J Respir Crit Care MedYear: 200517261962410.1164/rccm.200406-694OC15961699|
|Delisle S,Ouellet P,Bellemare P,Tetrault JP,Arsenault P,Sleep quality in mechanically ventilated patients: comparison between NAVA and PSV modesAnn Intensive CareYear: 201114210.1186/2110-5820-1-4221955588|
|Fauroux B,Leroux K,Desmarais G,Isabey D,Clement A,Lofaso F,Louis B,Performance of ventilators for noninvasive positive-pressure ventilation in childrenEur Respir JYear: 2008311300130710.1183/09031936.0014480718321932|
|Beck J,Campoccia F,Allo JC,Brander L,Brunet F,Slutsky AS,Sinderby C,Improved synchrony and respiratory unloading by neurally adjusted ventilatory assist (NAVA) in lung-injured rabbitsPediatr ResYear: 20076128929410.1203/01.pdr.0000257324.22406.9317314685|
|Alander M,Peltoniemi O,Pokka T,Kontiokari T,Comparison of pressure-, flow-, and NAVA-triggering in pediatric and neonatal ventilatory carePediatr PulmonolYear: 201247768310.1002/ppul.2151921830318|
|Rhoney DH,Murry KR,National survey of the use of sedating drugs, neuromuscular blocking agents, and reversal agents in the intensive care unitJ Intensive Care MedYear: 20031813914510.1177/088506660325120014984632|
|Papazian L,Forel JM,Gacouin A,Penot-Ragon C,Perrin G,Loundou A,Jaber S,Arnal JM,Perez D,Seghboyan JM,Constantin JM,Courant P,Lefrant JY,Guérin C,Prat G,Morange S,Roch A; ACURASYS Study Investigators,Neuromuscular blockers in early acute respiratory distress syndromeN Engl J MedYear: 20103631107111610.1056/NEJMoa100537220843245|
Main available clinical studies
|Study||Diagnosis (number of patients)||Design||Duration||Major findings|
|Colombo and colleagues, 2008 ||Acute respiratory failure (unselected) (n = 14)||Crossover - PSV set to obtain VT 6 to 8 ml/kg predicted body weight; NAVA vs. PSV increased or decreased by 50%. Effects of modification of assist level||20 minutes ×3||NAVA averted the risk of overassistance and improved synchrony|
|Wu and colleagues, 2009 ||ARDS (n = 18)||PSV vs. NAVA - randomized study. Incremental PSV and NAVA run randomly in four steps. The PSV level was gradually increased by 5 cmH2O every 5 minutes from 5 to 20 cmH2O. Incremental NAVA was individually set in steps of 0.2 to 1.0 cmH2O/μV every 5 minutes to determine the NAVA level providing an airway pressure in each step equivalent to that with PSV. Evaluation of patient-ventilator synchrony (trigger delay, ineffective effort); effect of assist level||5 minutes ×4||Improved synchrony with NAVA|
|Brander and colleagues, 2009 ||Acute respiratory failure (unselected) (n = 15)||NAVA level increased progressively. Method for titrating the NAVA level||30 minutes + 3 hours||Progressive implementation of NAVA may be a method for determining the adequate level - downregulation of EAdi confirmed|
|Schmidt and colleagues, 2010 ||Mainly acute lung injury (n = 12)||Longitudinal observational study - PSV set to obtain VT 6 to 8 ml/kg predicted body weight; NAVA vs. PSV with increasing assist. Breath-by-breath variability of flow and EAdi-related variables quantified by the coefficient of variation and autocorrelation analysis||10 minutes ×4 = 40 minutes||NAVA increases breathing pattern variability|
|Coisel and colleagues, 2010 ||Postoperative patients (n = 15)||Crossover randomized - PSV set to obtain VT 6 to 8 ml/kg predicted body weight; NAVA vs. PSV. Effects on breathing pattern, gas exchange, and variability of respiratory cycles (evaluated by coefficient of variation)||24 hours||Variability of, tidal volume and minute ventilation were significantly higher with NAVA than with PSV. Variability of electrical diaphragmatic activity was significantly lower with NAVA than with PSV. Oxygenation increased during NAVA|
|Terzi and colleagues, 2010 ||ARDS (n = 11)||Crossover randomized - PSV set to obtain VT 6 to 8 ml/kg predicted body weight; NAVA vs. PSV. Effect of neural trigger vs. flow trigger. Assist level was randomized to be 20%, 40%, or 60% over the basal level. Assessment of the physiological response to varying PSV and NAVA levels in selected ARDS patients and the effect of neural triggering||5 minutes ×4 × 3 = 60 minutes||Compared with PSV, NAVA limited the risk of overassistance, prevented patient-ventilator asynchrony, and improved overall patient-ventilator interactions. Compared with the pneumatic trigger, the neural trigger (from NAVA) considerably decreased patient-ventilator asynchrony|
|Spahija and colleagues, 2010 ||COPD (14)||Prospective, comparative crossover - PSV set to obtain VT 6 to 8 ml/kg predicted body weight; NAVA vs. PSV. Patients were ventilated for 10-minute periods, using two PSV levels (lowest tolerable and 7 cmH2O higher) and two NAVA levels (same peak pressures and external PEEP as with PSV), delivered in random order||10 minutes ×2||NAVA improved patient-ventilator synchrony by reducing the triggering and cycling delays, especially at higher levels of assist, while preserving breathing and maintaining blood gas exchange|
|Passath and colleagues, 2010 ||Unselected patients (n = 20)||Longitudinal observational study. Evaluation of effects of PEP on breathing pattern and neuroventilatory efficiency during NAVA. Adequate NAVA level was determined as the NAVA level early after the transition from an initial steep increase in Paw and VT to a less steep increase or even plateau of Paw and VT, as described by Brander and colleagues. PEEP was set at 20 cmH2O then decreased to 1 cmH2O. VT/EAdi was evaluated as an indicator of neuroventilatory efficiency||20 minutes ×3||During adequate-assist NAVA, increasing PEEP reduces respiratory drive. Patients adapt their neuroventilatory efficiency such that the individual ventilatory pattern is preserved over a wide range of PEEP levels. Monitoring VT/EAdi during PEEP changes allows identification of a PEEP level at which tidal breathing occurs at minimal EAdi cost|
|Piquilloud and colleagues, 2011 ||Unselected patients (n = 22; COPD n = 8/22)||Prospective interventional study - three consecutive periods of ventilation: PSV-NAVA-PSV. Airway pressure, flow, and transesophageal diaphragmatic electromyography were recorded continuously. To determine whether, compared with PSV, NAVA reduced trigger delay, inspiratory time excess, and the number of patient-ventilator asynchrony events||20 minutes ×3||NAVA reduces trigger delay, improves expiratory synchrony (inspiratory time excess was reduced) and reduces total asynchrony events|
|Rozé and colleagues, 2011 ||Unselected patients (n = 15)||To determine the feasibility of daily titration of the NAVA level in relation to the maximal diaphragmatic electrical activity (EAdimaxSBT) measured during a SBT during PSV. EAdimaxSBT was determined daily during a SBT using PSV with 7 cmH2O inspiratory pressure and no PEEP. If the SBT was unsuccessful, NAVA was used and the level was then adjusted to obtain an EAdi of 60% of the EAdimaxSBT. Arterial blood gas analyses were performed 20 minutes after each change in NAVA level||Until extubation||Daily titration of NAVA level with an electrical goal of 60% EAdimaxSBT is feasible and well tolerated|
ARDS, acute respiratory distress syndrome; COPD, chronic obstructive pulmonary disease; EAdi, electrical activity of the diaphragm; NAVA, neurally adjusted ventilatory assist; Paw, airway pressure; PEEP, positive end-expiratory pressure; PEP, end-expiratory pressure; PSV, pressure-support ventilation; SBT, spontaneous breathing trial; VT, tidal volume.
Previous Document: Children's memory distortions following social contact with a co-witness: Disentangling social and c...
Next Document: DISSIPATION OF FUNGICIDES IN A VINEYARD SOIL AMENDED WITH DIFFERENT SPENT MUSHROOM SUBSTRATES.