Algorithmic processing of pressure waveforms to facilitate estimation of cardiac elastance.  
Jump to Full Text  
MedLine Citation:

PMID: 22703604 Owner: NLM Status: MEDLINE 
Abstract/OtherAbstract:

BACKGROUND: Cardiac elastances are highly invasive to measure directly, but are clinically useful due to the amount of information embedded in them. Information about the cardiac elastance, which can be used to estimate it, can be found in the downstream pressure waveforms of the aortic pressure (P(ao)) and the pulmonary artery (P(pa)). However these pressure waveforms are typically noisy and biased, and require processing in order to locate the specific information required for cardiac elastance estimations. This paper presents the method to algorithmically process the pressure waveforms. METHODS: A shear transform is developed in order to help locate information in the pressure waveforms. This transform turns difficult to locate corners into easy to locate maximum or minimum points as well as providing error correction. RESULTS: The method located all points on 87 out of 88 waveforms for Ppa, to within the sampling frequency. For Pao, out of 616 total points, 605 were found within 1%, 5 within 5%, 4 within 10% and 2 within 20%. CONCLUSIONS: The presented method provides a robust, accurate and dysfunctionindependent way to locate points on the aortic and pulmonary artery pressure waveforms, allowing the noninvasive estimation of the left and right cardiac elastance. 
Authors:

David Stevenson; James Revie; J Geoffrey Chase; Christopher E Hann; Geoffrey M Shaw; Bernard Lambermont; Alexandre Ghuysen; Philippe Kolh; Thomas Desaive 
Related Documents
:

12597674  Effects of progressive muscle relaxation on blood pressure and psychosocial status for ... 22382664  The effect of dexmedetomidine on arterialcardiac baroreflex function assessed by spect... 8252694  Effects of different training intensities on 24hour blood pressure in hypertensive sub... 14607584  Lifetable analysis of the success of thermal balloon endometrial ablation in the treat... 12721104  At1 receptor antagonist therapy preferentially ameliorated right ventricular function a... 12853544  Continuous hemodynamic monitoring in pulmonary hypertensive patients treated with inhal... 
Publication Detail:

Type: Journal Article; Research Support, NonU.S. Gov't Date: 20120615 
Journal Detail:

Title: Biomedical engineering online Volume: 11 ISSN: 1475925X ISO Abbreviation: Biomed Eng Online Publication Date: 2012 
Date Detail:

Created Date: 20130101 Completed Date: 20130423 Revised Date: 20130712 
Medline Journal Info:

Nlm Unique ID: 101147518 Medline TA: Biomed Eng Online Country: England 
Other Details:

Languages: eng Pagination: 28 Citation Subset: IM 
Affiliation:

Department of Mechanical Engineering, Centre for Bio Engineering at the University of Canterbury, Christchurch, New Zealand. 
Export Citation:

APA/MLA Format Download EndNote Download BibTex 
MeSH Terms  
Descriptor/Qualifier:

Algorithms* Aorta / physiology Blood Pressure* Heart / physiology* Humans Models, Statistical* Pulmonary Artery / physiology Wavelet Analysis* 
Comments/Corrections 
Full Text  
Journal Information Journal ID (nlmta): Biomed Eng Online Journal ID (isoabbrev): Biomed Eng Online ISSN: 1475925X Publisher: BioMed Central 
Article Information Download PDF Copyright ©2012 Stevenson et al.; licensee BioMed Central Ltd. openaccess: Received Day: 23 Month: 12 Year: 2011 Accepted Day: 28 Month: 5 Year: 2012 collection publication date: Year: 2012 Electronic publication date: Day: 15 Month: 6 Year: 2012 Volume: 11First Page: 28 Last Page: 28 PubMed Id: 22703604 ID: 3533753 Publisher Id: 1475925X1128 DOI: 10.1186/1475925X1128 
Algorithmic processing of pressure waveforms to facilitate estimation of cardiac elastance  
David Stevenson1  Email: david.stevenson@pg.canterbury.ac.nz 
James Revie1  Email: Jamejames.revie@pg.canterbury.ac.nz 
J Geoffrey Chase1  Email: geoff.chase@canterbury.ac.nz 
Christopher E Hann1  Email: chris.hann@canterbury.ac.nz 
Geoffrey M Shaw2  Email: Geoff.Shaw@cdhb.govt.nz 
Bernard Lambermont3  Email: b.lambermont@chu.ulg.ac.be 
Alexandre Ghuysen3  Email: A.Ghuysen@chu.ulg.ac.be 
Philippe Kolh3  Email: Philippe.Kolh@chu.ulg.ac.be 
Thomas Desaive3  Email: tdesaive@ulg.ac.be 
1Department of Mechanical Engineering, Centre for Bio Engineering at the University of Canterbury, Christchurch, New Zealand 

2Department of Intensive Care, Christchurch Hospital, Christchurch, New Zealand 

3Cardiovascular Research Center, University of Liege, Belgium 
In an Intensive Care Unit (ICU), cardiac disturbances are difficult to diagnose and treat, which can lead to poor management [^{1},^{2}]. Inadequate diagnosis can be common, and plays a significant role in increased length of stay and death [^{3}^{}^{5}], despite access to many different cardiac measurements and metrics. Currently, internal measurements are only possible at the locations where catheters are placed. This limited set of data can severely restrict clinical diagnostic capability, and thus these catheters are not necessarily associated with improved outcomes [^{6}^{}^{8}]. Overall, a lot of data currently available to ICU clinicians, that could have significant clinical value, is under utilised.
Using modelling techniques, this limited set of data can be expanded to estimate a much greater set of clinically relevant data to enable more accurate diagnosis. For example, acute cardiovascular dysfunction, like pulmonary embolism (PE) and septic shock, severely alter \ system (CVS) hemodynamics around the heart. These changes can be seen by catheter measurements as a change in the balance of preload and afterload, resulting in an altered cardiac energetic state [^{9},^{10}]. Detailed cardiac energetics are too invasive to measure in an ICU setting. However, if the relevant energetics could be captured from a nearby catheter, the clinical potential of such measurements could be realised. To date, no such method achieves this aim.
Timevarying cardiac elastance (TVE) is defined [^{11}]:
(1)
e(t)=Pv(t)Vv(t)−Vd 
where, V _{d} is assumed to be equal to V _{0} for simplicity, V _{0} is the intercept of the endsystolic pressurevolume relation (ESPRV) with the volume axis [^{12}], P_{v}(t) is the ventricle pressure and V _{v}(t) is the ventricle volume. It thus provides a measure of heart function and energetics [^{13}^{}^{15}]. The waveform e(t) is typically normalised to a value of 1.0 [^{13}], and can also be used as the input or driver function in lumped parameter CVS models [^{16}^{}^{20}].
There have been several attempts to estimate TVE [^{14},^{21}^{}^{24}]. However, none have estimated it for its own sake. Most studies present a method using the TVE to estimate a specific parameter, most commonly endsystolic elastance (E_{es}) [^{14},^{23},^{24}] and ejection fraction [^{22}]. However, their validation is based on these metrics, not on the resulting TVE waveform.
This research is unique in that the end goal is to produce the TVE function in its own right, validating the TVE waveform on its own accuracy for eventual use as a diagnostic tool. It is unclear to date how much specific information can be obtained from the TVE waveform, other than the highly soughtafter E_{es}[^{25}] (although this cannot be found from the normalised TVE waveform). However, TVE features are highly correlated to clinical parameters [^{26}] and contain similar information to pressurevolume (PV) loops, which are known to contain information on cardiac function [^{27}] including cardiac work [^{28},^{29}], contractility [^{13},^{30}], O_{2} consumption [^{29},^{31}], and all the states of filling, contraction ejection and relaxation [^{1}]. Thus, a TVE waveform reflects cardiac state, cardiac output or blood volume, and net preload and afterload, all of which change with cardiac dysfunction. Hence, the ability to easily and noninvasively obtain TVE waveforms could enable clinically useful diagnostics and metrics.
This paper presents the first step in estimating TVE from already available measurements, namely aortic (P_{ao}) and pulmonary artery (P_{pa}) pressure. This research develops algorithms to process these pressure waveforms to extract specific features and points, which, in turn, allow the estimation of the end goal, the TVE waveform. The P_{ao} and P_{pa} pressure waveforms, as typically measured, are noisy and/or biased, which can significantly effect this process. Hence, a method is presented for automatically processing the pressure waveforms to robustly and accurately locate the points required by the correlations, that enable accurate, not additionally invasive cardiac elastance estimation and construction.
The specific approach presented defines a representative set of points on the TVE waveforms. These points allow a continuous waveform to be constructed that adequately follows the shape of a true (invasively measured) TVE, capturing the necessary dynamics. Correlations between these representative points and the points or properties on the typically measured waveforms, P_{ao} and P_{pa}, enable construction of continuous beattobeat estimations of TVE with knowledge only of the P_{ao} and P_{pa} waveforms and the predefined correlations. It thus uses typically available data to construct what would otherwise require a highly invasive added test. A high level view of this approach is presented in Figure 1, which shows the formation and use of the correlations to generate and estimated cardiac elastance waveform. This paper is focused on the left half of Figure 1, that of producing the correlations for later use. A brief overview of the whole method is given for clarity:
1. locate points on P_{ao} and P_{pa} (this paper)
2. correlate points of the pressure waveforms to points on the measured cardiac elastance
3. use these correlations to estimate the points on the cardiac elastance
4. create a continuous function e(t) through the estimated points with (2)(4)
5. compare the estimated elastance waveform to the measured elastance waveform
(2)
e(t)=Fα(t)0<t<cα(1−x2)(t−cα)cβ−cα+x2cα<t<cβFβ(t)cβ<t<period 
where:
(3)
Fi=ai·e−bi(t−ci)2 
and the coefficients of (3), also seen in (2), are fitted for a specific waveform, and are defined:
(4)
aα=x2bα=−log(x1/x2)exp(log(−log(x1/x2)·2·(x1/x˙1))·2)cα=−log(x1/x2)·2·x1−x˙1·t1x˙1 
where a_{β}, b_{β} and c_{β }are similarly defined by replacing subscript 1 with 3 and setting x_{2} = 1.
Figure 2 shows an illustrative mapping between points on P_{ao} and TVE. However, this approach is useful if and only if it is possible to automate the detection of the required points, defined in (5), on the P_{ao} and P_{pa} waveforms, shown in Figures 3 and 4.
This paper focuses on the robust capturing of the points on the pressure waveforms and leaves the specific correlations and methodology of creating the TVE waveforms to a paper in review. The details contained in this paper about the formulation of the correlations and there use are illustrative only, and are assumed to be correct for the purposes of demonstrating a potential use for this research.
For describing the methods in this paper, a naming convention is defined:
(5)
e(t)≡time varying cardiac elastancePao≡aortic pressurePpa≡pulmonary artery pressureDMPG≡driver maximum positive gradientMN≡minimum pointMPG≡maximum positive gradientLS≡left shoulderMX≡maximumRS≡right shoulderMNG≡maximum negative gradientDN≡dicrotic notch 
This paper uses a shear transform to extract features from the P_{ao} and P_{pa} waveforms, defined:
(6)
S≡(t,X(t))→t,ϕshear(X(t)) 
where:
(7)
ϕshearX(t)=X(t)+mt+c,t0<t<tend 
(8)
X(t)≡descrete, time valued data 
and the parameters t_{0} and t_{end} are set depending on the region or period of interest. The parameters m and c are chosen such that:
(9)
ϕshear(X(t0))=X(t0)=ϕshear(X(tend)) 
Equation (9) leads to:
(10)
X(t0)+m·t0+c=X(t0)X(tend)+m·tend+c=X(t0) 
Solving (10), for m and c yields:
(11)
m=X(t0)−X(tend)tend−t0c=−m·t0 
To better visualize how this transformation operates, imagine a line from A to C in Figure 5, representing a portion of the waveform in Figure 3, rotated about A so that end points align horizontally, while time remains unchanged. Hence, it is a rotation and contraction that projects the line onto a horizontal axis (time). The effect of this transformation is to transform the difficult to find “shoulder” point B into an easily found peak of a curve, or for the reflection of A → B in x, a minimum or the curve. A “shoulder” is defined as a point at which two near linear lines with different slopes meet, such as the point LS in Figure 3. Thus, the use of this transform makes it far easier to, algorithmically, locate aspects of the waveforms which can be otherwise difficult to find.
The transform, S, is used in two ways. First it is used to locate a point of interest based on the maximum or minimum point of S. This use is demonstrated in Figure 5. Hence, a desired point P is defined:
(12)
P=SmaxorSmin 
Both local maximum, Smax(local), and minimum, Smin(local), are also required. Smax(local) exists and is the maximum point of S, if and only if, there exists a maximum stationary point that does not fall at the temporal boundary of S, and similarly for Smin(local).
The second way the shear transform of (6) is used relates to the verification of a particular point given an initial guess. For example, the first guess of the point MN is the global minimum of the waveform, after which this point is verified using the shear transformation, resulting in confirmation of the point or a new point to use instead.
This works by locating the maximum of minimum point of the shear transform, P_{2}, over a given range, t, near the first guess, P_{1}, and also defining a threshold time, D. If the point P_{2} lies temporally within tP1±D, then the correct point is the initial guess P_{1}, otherwise the correct point is P_{2}.
The choice of Smax or Smin, the range of time, t, and the threshold time, D, are defined for the type of point under consideration. The specific values, listed in Section Point location method, are chosen empirically, based on what features that appear close to the point of interest and the temporal variation that has been observed in these features.
These two situations are graphically shown in Figure 6 and Figure 7 for positive values of D. However the same applies for negative values, for which the real point lies before the point P_{1}, instead of after it. The complete process is defined:
(13)
P=P1iftP2lies temporally withintP1±DP2otherwise 
where:
(14)
P1≡initial point to be checkedP2≡SmaxorSmin(chosen separately)D≡threshold timet∈{t:t0<t<tend} 
Combined these two methods of use, shown in Figure 5^{}7, create a robust and computationally fast method for locating certain hard to find points on a waveform.
The method for finding the points is described in Figure 8 along with the following two sections (Finding DMPG and Finding DN). Figure 8 gives the full text, reproducible and ordered method except for the points DMPG and DN (which are described in the next two sections) along with a graphical illustration. The graphical illustrations are the outworking of the method for a representative waveform, and are only intended to aid the reader in their understanding of the method, and not to formally describe the method itself. Due to the complexity of the method for the points DMPG and DN, these two have been described in Figure 8 only for the simplest (as well as the and most common) case, with the full method described in separate sections with relevant figures.
There are some cases, where DMPG, see Figure 3, is equivalent to MN. These cases occur when:
(15)
(tMX−tMN)<period·0.25 
When (15) is not true, DMPG is found as the Smax(local) from 34·tMN to MN, see Figure 9. However, there are a few cases, for both sepsis and pulmonary embolism, in which a local maximum of S does not exist except at the boundaries of the region, which is not acceptable if an automated detection method is desired. In this case, a point P_{2}, is defined as the Smin from 34·tMN to MN. If Smax(local) from P_{2} to MN exists, this is taken as DMPG, see Figure 10, otherwise DMPG is defined as Smax(local) from 34·tMN to P_{2}, see Figure 11. If this final local maximum does not exist, DMPG is defined as 34·tMN. These cases occur due to noise, variability and dysfunction and are part of what makes robust algorithmic or automated processing difficult.
This process is defined:
(16)
P=P1if∃P1P3if∃P3P4if∃P434·tMNotherwise 
where:
(17)
P1≡Smax(local),t∈{t:34·tMN<t<tMN}P2≡Smin(local),t∈{t:34·tMN<t<tMN}P3≡Smax(local),t∈{t:tP2<t<tMN}P4≡Smax(local),t∈{t:34·tMN<t<tP2} 
The general approach to find the point DN, see Figure 3, is to find Smin between t_{MX} and t_{end} (or period). However, in a number of cases this approach fails due to oscillations towards the end of the waveform, see Figure 12. Also, using only the first local minimum (as is the case in Figure 12) works only in a few cases and therefore is not a robust solution either. Hence a more specific algorithm is required.
A second maximum point is defined:
(18)
MX2≡max point ofPao(t),t∈t:tMN+period5<t<period 
and two more intermediate points are defined:
(19)
DN1≡lowestSmin(local),t∈{t:tMX2<t<period} 
(20)
DN2≡firstSmin(local),t∈{t:tMX2<t<period} 
From DN_{1} and DN_{2}, the real DN is chosen, defined:
(21)
DN=DN1if(CA∧CB)∨(CC∧CD)DN2otherwise 
where:
(22)
CA≡DN2>MX2CB≡(tDN2−tMX2)<0.15·periodCC≡DN2−DN1Pao(tMX)>0.02CD≡(tDN2−tMX2)<0.11·period 
While using a measured P_{pa} waveform, DN is defined:
(23)
DN=DN1 
The method presented was developed on a set of five pigs (51 waveforms) that were induced with pulmonary embolism [^{32},^{33}], and then independently tested on a further five pigs (37 waveforms) induced with septic shock, and treated with haemofiltration [^{34},^{35}].
The points for all waveforms (see Figures 3^{}4) were identified or checked individually by eye. The two gradients (MPG and MNG) were first located through simple computation, and then individually checked by eye and corrected were necessary. Due to the nature and location of these points (maximum gradient of a sigmoidal function), they are the two easiest and most reliable to find algorithmically, and in fact the algorithmic approach is more accurate than hand selection. The two shoulders (LS and RS) were first located through the algorithm developed prior to that which is described in this paper, after which each point was individually checked and corrected. Because there is no formal definition for the location of these “shoulder” points, it was left to an algorithmic definition. For a validation test this definition is self fulfilling. However, as LS (RS is only used to aid in finding MNG , and is hence not included in the validation results) is found as an intermediate step to the estimation of the cardiac elastance, its full and more formal validation would be the results of the cardiac estimation which is not in the scope of this paper. All the remaining points were hand selected.
The automated method was applied to the waveforms and the identified points assessed against the known points for accuracy in time. The use of separate data with different cardiac dysfunction to design and test the method ensures the robustness of the validated method.
For the points (MN and DN) required when using P_{pa} (Figure 4), the method located both points in 87 of the 88 waveforms to within the sample frequency of 200Hz (0.005 sec), missing DN, from one waveform. This missed point is in a waveform at the start of the third pig of the sepsis cohort and is unique to the data set, both in the measured TVE and P_{pa}, as shown in Figure 13, compared to the more typical P_{pa} waveform in Figure 4. The failure is due to the unusual second peak of P_{pa}, and the early decay of the TVE.
For each P_{ao} waveform, the method locates eight points, MX, MN, DMPG, LS, MPG, DN, RS, MNG. However, RS and MN are only used to aid the location of other points. These two points were both located sufficiently to enable the method to progress in all 88 waveforms. Results for the other six points are shown in Table 1. Of 616 total points, 605 were found within 1%, 5 within 5%, 4 within 10% and 2 within 20%.
The automated, algorithmic method presented enables the mapping between aortic pressure (P_{ao}), pulmonary artery pressure (P_{pa}), and the ventricle TVE (e_{rv}(t) and e_{lv}(t)), by accurately processing the P_{ao} and P_{pa} waveforms to identify specific points. Once combined, they enable a very useful tool, for clinicians to obtain very accurate TVE without further invasive or risky sensors or procedures.
There are other ways to locate the points on the pressure waveforms, most notably a derivative and second derivative method. However, this becomes problematic in practice due the noise inherent in the waveforms. The method that has been developed in this paper, was designed to work with the level of noise that is typically seen on these measurements, and is therefore more involved than a simple derivative method.
The method presented was robust to the typical and significant variation and noise in P_{ao} and P_{pa} waveforms. The method was developed on five pigs induced with pulmonary embolism, and then tested independently with data from another set of five pigs induced with septic shock. The results give confidence that this method will generalize to a wider set of disease states and to human data.
However, while the results were very good, this research needs further validation on a wider cohort of pigs and types of dysfunction to further quantify the limits and accuracy of this approach. Direct validation on humans is the ultimate goal. However, the results appear robust, and justify and enable a wide range of further more indepth validation studies of both the method and its potential uses when reconstructing TVE for monitoring and diagnosis.
Clinically, it must be noted that for this method to work a SwanGanz catheter is assumed. If radial artery pressure was measured instead, there would be more oscillations in the waveform, potentially requiring modifications. However, SwanGanz catheters are still commonly used, and this application would add value to their use, which is otherwise sometimes contested [^{6}^{}^{8}].
The method developed in this paper shows promise for gaining clinical insight and improving diagnosis. It can enable clinicians to get more information about the current patient state, without the use of more invasive measurements, as well as beattobeat tracking of this information. This level of detail is far more than currently available and could potentially lead to better and earlier diagnosis of dysfunction, as well as better knowledge of response to treatment, noninvasively, as it needs no further procedures or sensors required.
This paper has presented a robust, potentially dysfunctionindependent method to find the waveform points necessary to use proven methods to noninvasively and automatically estimate the otherwise unavailable left and right ventricle TVEs with accuracy well within measurement error. This capability is enabled using standard measurements that are already commonly used in an intensive care setting, thus involving no additional risk to the patient. The results thus justify prospective validation of these conclusions.
The authors declare that they have no competing interests.
DS drafted the manuscript and developed the algorithm. JR and JC participated in the algorithm development with added input from TD. CH participated in the initial mathematical formulation. GS provided physiological understanding and clinical input at all stages. BL, AG, PK and TD provided the porcine data and further clinical input and relevance. JC and TD edited and aided the writing of the manuscript and revisions with DS. All authors read and approved the final manuscript.
This work was supported by the Foundation for Research Science and Technology (FRST) of New Zealand, the Fonds de la Recherche Scientifique (FNRS, Belgium) and the French Community of Belgium (Actions de Recherches Concertées Académie WallonieEurope)
References
Guyton A,Hall J,Textbook of Medical PhysiologyYear: 2000Philadelphia: W.B. Saunders Company  
Grenvik A,Ayres SM,Holbrook PR,Textbook of Critical CareYear: 1989Philadelphia: W.B. Saunders Company  
Angus DC,LindeZwirble WT,Lidicker J,Clermont G,Carcillo J,Pinsky MR,Epidemiology of severe sepsis in the United States: analysis of incidence, outcome, and associated costs of careCrit Care MedYear: 200129713031010.1097/000032462001070000000211445675  
Kearon C,Diagnosis of pulmonary embolismCMAJYear: 200316821839412538548  
Pineda LA,Hathwar VS,Grant BJB,Clinical Suspicion of Fatal Pulmonary EmbolismChestYear: 200112037915 [ http://chestjournal.chestpubs.org/content/120/3/791.full]. 10.1378/chest.120.3.79111555511  
Frazier S,Skinner GJ,Pulmonary artery catheters: state of the controversyJournal of Cardiovascular NursingYear: 2008231132118382253  
Chatterjee K,The SwanGanz Catheters: Past, Present, and FutureCirculationYear: 200911914715210.1161/CIRCULATIONAHA.108.81114119124674  
Cooper A,Doig WJ,Sibbald GS,Pulmonary artery catheters in the critically illCrit Care MedYear: 19961277794  
Weber KT,Janicki JS,The heart as a musclepump system and the concept of heartfailureAm Heart JYear: 19799833718410.1016/00028703(79)900516382811  
Ross JJ,Afterload mismatch and preload reserve: a conceptual framework for the analysis of ventricular functionProg Cardiovasc DisYear: 1976182556410.1016/00330620(76)900219128034  
Suga H,Sagawa K,Instantaneous pressurevolume relationships and their ratio in the excised, supported canine left ventricleCirc ResYear: 1974351172610.1161/01.RES.35.1.1174841253  
Sagawa K,The endsystolic pressurevolume relation of the ventricle: definition, modifications and clinical useCirculationYear: 19816361223710.1161/01.CIR.63.6.12237014027  
Suga H,Sagawa K,Shoukas AA,Load independence of the instantaneous pressurevolume ratio of the canine left ventricle and effects of epinephrine and heart rate on the ratioCirc ResYear: 19733233142210.1161/01.RES.32.3.3144691336  
Senzaki H,Chen CH,Kass DA,Singlebeat estimation of endsystolic pressurevolume relation in humans. A new method with the potential for noninvasive applicationCirculationYear: 19969410249750610.1161/01.CIR.94.10.24978921794  
Sunagawa K,Sagawa K,Maughan WL,Ventricular interaction with the loading systemAnn Biomed EngYear: 19841221638910.1007/BF025842296507965  
Smith BW,Chase JG,Nokes RI,Shaw GM,Wake G,Minimal haemodynamic system model including ventricular interaction and valve dynamicsMed Eng PhysYear: 2004262131910.1016/j.medengphy.2003.10.00115036180  
Starfinger C,Chase JG,Hann CE,Shaw GM,Lambermont B,Ghuysen A,Kolh P,Dauby PC,Desaive T,Modelbased identification and diagnosis of a porcine model of induced endotoxic shock with hemofiltrationMath BiosciYear: 20082162132910.1016/j.mbs.2008.08.01418817788  
Chung DC,Niranjan SC,Clark J,Bidani A,Johnston WE,Zwischenberger JB,Traber DL,A dynamic model of ventricular interaction and pericardial influenceAm J PhysiolYear: 19972726 Pt 2H2942—629227574  
Revie J,Stevenson D,Chase G,Hann CE,Lambermont B,Ghuysen A,Kolh P,Morimont P,Shaw GM,Desaive T,Clinical detection and monitoring of acute pulmonary embolism: proof of concept of a computerbased methodAnnals of Intensive CareYear: 2011133 [ http://www.annalsofintensivecare.com/content/1/1/33].  
Revie J,Stevenson D,Chase JG,Hann CE,Lambermont B,Ghuysen A,Kolh P,Shaw GM,Heldmann S,Desaive T,Validation of subjectspecific cardiovascular system models from porcine measurementsComput Methods Programs BiomedYear: 2011 inpress. [ http://www.sciencedirect.com/science/article/pii/S0169260711003014].  
Guarini M,Urzúa J,Cipriano A,González W,Estimation of caridac function from computer analysis of the arterial pressure waveformIEEE Transactions on Biomedical EngineeringYear: 199845121420810.1109/10.7304369835191  
Swamy G,Kuiper J,Gudur MSR,Bari Oliver N,Mukkamala R,Continuous Left Ventricular Ejection Fraction Monitoring by Aortic Pressure Waveform AnalysisAnn Biomed EngYear: 200937610556810.1007/s104390099675419308732  
Shishido T,Hayashi K,Shigemi K,Sato T,Sugimachi M,Sunagawa K,Singlebeat estimation of endsystolic elastance using bilinearly approximated timevarying elastance curveCirculationYear: 2000102161983910.1161/01.CIR.102.16.198311034949  
ten Brinke EA,Klautz RJ,Verwey HF,van der Wall EE,Dion RA,Steendijk P,Singlebeat estimation of the left ventricular endsystolic pressurevolume relationship in patients with heart failureActa Physiol (Oxf)Year: 2010198374610.1111/j.17481716.2009.02040.x19735484  
Burkhoff D,Chasing the Elusive PressureVolume RelationshipJ Am Coll Cardiol ImgYear: 2009212824  
Starfinger C,Hann CE,Chase JG,Desaive T,Ghuysen A,Shaw GM,Modelbased cardiac diagnosis of pulmonary embolismComput Methods Programs BiomedYear: 200787466010.1016/j.cmpb.2007.03.01017478006  
Sagawa K,The ventricular pressurevolume diagram revisitedCirc ResYear: 19784356778710.1161/01.RES.43.5.677361275  
Suga H,Ventricular energeticsPhysiol RevYear: 1990702247772181496  
Burkhoff D,Sagawa K,Ventricular efficiency predicted by an analytical modelRegu PhysiolYear: 19862506R1021—7  
Broscheit JA,Weidemann F,Strotmann J,Steendijk P,Karle H,Roewer N,Greim CA,Timevarying elastance concept applied to the relation of carotid arterial flow velocity and ventricular areaJ Cardiothorac Vasc AnesthYear: 2006203340610.1053/j.jvca.2006.01.01416750733  
Suga H,Total mechanical energy of a ventricle model and cardiac oxygen consumptionHeartYear: 19792363H4985  
Desaive T,Dutron S,Lambermont B,Kolh P,Hann CE,Chase JG,Dauby PC,Ghuysen A,Closeloop model of the cardiovascular system including ventricular interaction and valve dynamics: application to pulmonary embolismYear: 2005Singapore  
Ghuysen A,Lambermont B,Kolh P,TchanaSato V,Magis D,Gerard P,Mommens V,Janssen N,Desaive T,D’Orio V,Alteration of right ventricularpulmonary vascular coupling in a porcine model of progressive pressure overloadingShockYear: 200829219720417693928  
Lambermont B,Ghuysen A,Kolh P,TchanaSato V,Segers P,Gerard P,Morimont P,Magis D,Dogne JM,Masereel B,D’Orio V,Effects of endotoxic shock on right ventricular systolic function and mechanical efficiencyCardiovasc ResYear: 2003592412810.1016/S00086363(03)00368712909324  
Lambermont B,Delanaye P,Dogne JM,Ghuysen A,Janssen N,Dubois B,Desaive T,Kolh P,D’Orio V,Krzesinski JM,Largepore membrane hemofiltration increases cytokine clearance and improves right ventricularvascular coupling during endotoxic shock in pigsArtif OrgansYear: 2006307560410.1111/j.15251594.2006.00260.x16836739 
Figures
[Figure ID: F1] 
Figure 1
Overview. The figure shows a conceptualised overview of the process described in this paper and further implications. From the many measured left cardiac elastance (e(t)) waveforms, alone with many aortic pressure (P_{ao}) waveforms, correlations are derived (the information flow is shown through the large grey arrow). Once these correlations are known, they can be used along with the aortic pressure waveform (from a patient), to arrive at an estimation of their cardiac elastance waveform. The equivalent for the right cardiac elastance is also shown, with the pulmonary artery pressure (P_{pa}). 
[Figure ID: F2] 
Figure 2
Illustrative elastance estimation. An example of what can be done with the identified points on the aortic pressure, and an example of the formation of the estimated cardiac elastance is shown here, while the terms are defined in (5). This figure is not part of the method of this paper, rather as a illustration of what the method as a whole leads to. 
[Figure ID: F3] 
Figure 3
Aortic pressure waveform and relevant points. A representative aortic pressure waveform over one heart beat with relevant points (defined in (5)) marked on it. The two dashed circles, MN and RS are used only in locating other points. 
[Figure ID: F4] 
Figure 4
Pulmonary artery pressure waveform and relevant points. A representative pulmonary artery pressure waveform over one heart beat with relevant points (defined in (5)) marked on it. The dashed circle, MX is only used to help find other points. 
[Figure ID: F5] 
Figure 5
Shear transform. An illustration of the shear transformation of (6), turning a hard to locate “shoulder” ( B¯) points into an easily found maximum point ( B¯). 
[Figure ID: F6] 
Figure 6
Use of the shear transform, A. The desired point for MN is P. However, in this example the global minimum of the waveform is P_{1}, which is the initial guess for MN. A shear transform of the pressure waveform between P_{1 }and MX reveals a minimum (P_{2}) outside the range of D, and hence the time of P_{2 }is taken as the time of MN. 
[Figure ID: F7] 
Figure 7
Use of the shear transform, B. This example is the other situation in the process of finding MN to Figure 6, i.e. the initial guess of the global minimum for MN is correct. Here, the minimum of the shear transform from P_{1 }to MX falls within the range of D and hence the P_{1 }is taken as MN. 
[Figure ID: F8] 
Figure 8
The method. The step by step method for finding the points on P_{ao }and P_{pa}, as labelled on the right. The graphics beside each step are for illustration only and are not meant to be part of the definition of the method, rather to see the method in operation on a representative P_{ao} waveform. Note that the methods described here for DMPG and DN are note complete as these require a more complex method, refer to Sections Finding DMPG and Finding DN for the complete method for these two points. 
[Figure ID: F9] 
Figure 9
Finding DMPG, A. A straight forward case for finding DMPG, where P_{1 }of (17) exists, hence DMPG ≡ P_{1}. 
[Figure ID: F10] 
Figure 10
Finding DMPG, B. A less common case for finding DMPG, where P_{1 }of (17) does not exist, but P_{3 }does, hence DMPG ≡ P_{3}. 
[Figure ID: F11] 
Figure 11
Finding DMPG, C. A less common case for finding DMPG, where P_{1}, and P_{3 }of (17) do not exist, but P_{4 }does, hence DMPG ≡ P_{4}. 
[Figure ID: F12] 
Figure 12
Finding DN. An example of where the first local minimum of the shear transform is the correct time for the point DN. 
[Figure ID: F13] 
Figure 13
Where the method fails.P_{pa} alongside the matching TVE. The automatic or algorithmic method failed to capture the correct DN point (circle), the real DN and associated MX are marked by squares. 
Tables
The error data for each location, grouped by number of points found per error band for each location
<1%  1−5%  5−10%  10−20%  

DMPG

84

1

3

0

MPG

88

0

0

0

LS^{∗}

86

1

1

0

MX

88

0

0

0

MNG

87

0

0

1

DN

86

2

0

0

TOTAL  605  5  4  2 
Accuracy of the method: number of points grouped by absolute error (of 88 total points). ^{*} note that the validation LS here is partially self fulfilling and only included here for completeness.
Article Categories:

Previous Document: Incidence and prevalence of idiopathic inflammatory myopathies among commercially insured, Medicare ...
Next Document: Subclinical impairment of ovarian reserve in juvenile systemic lupus erythematosus after cyclophosph...